-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 13:26:35 2023 UTC
-- File 20220624_101451_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.322070 to 19.392449
-each separately from 8s to 19s and from 322070us to 392449us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.069801 in 1 packets: 18
MIU packet time from 2022-06-24T10:15:02.322.070.000 to 2022-06-24T10:15:13.392.449.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.069801 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.6394102 to 20.6393682
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999580 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_101451_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.9614808 to 9.9617698
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0697590  -- OK for SL --
   MIU time from 2022-06-24T10:15:02.322.070.000 to 2022-06-24T10:15:02.322.359.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.069801

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0315288 to 21.0318178
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T10:15:13.392.160.000 to 2022-06-24T10:15:13.392.449.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.961480800 corrected by 40bit 10MHz counter of 9.639410200 obtained at 1Hz counter of 8. Result: 8.322071824
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996200
   Min 1st sample MIU time among all SL packets: 2022-06-24T10:15:02.321.929.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T10:15:13.392.308.725
   Min 1st sample XO time among all SL packets: 8.321931548
   Max 1st sample XO time among all SL packets: 19.392310816
   TM values from -248 to 188 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T10:15:02.321.929.725 (XO: 8.321931548 - 8.322237550, t=37.2 degC)  Rank: 00079
   TM values: Max =   188  Min =  -248  Ave =    -4.38  Std =    21.86
   Min TM value:-248 at an offset of 128.838 usec -> 2022-06-24T10:15:02.322.058.562 (XO: 8.322060386)
   Max TM value:188 at an offset of 126.288 usec -> 2022-06-24T10:15:02.322.056.012 (XO: 8.322057836)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 128.938 usec -> 2022-06-24T10:15:02.322.058.662 (XO: 8.322060486)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 126.400 usec -> 2022-06-24T10:15:02.322.056.125 (XO: 8.322057949)

 - Block number: 2 starting at 2022-06-24T10:15:13.392.019.725 (XO: 19.392021815 - 19.392327816, t=37.2 degC)  Rank: 00082
   TM values: Max =   185  Min =  -238  Ave =    -4.51  Std =    23.61
   Min TM value:-238 at an offset of 130.625 usec -> 2022-06-24T10:15:13.392.150.350 (XO: 19.392152440)
   Max TM value:185 at an offset of 126.288 usec -> 2022-06-24T10:15:13.392.146.012 (XO: 19.392148103)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 126.163 usec -> 2022-06-24T10:15:13.392.145.887 (XO: 19.392147978)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 128.263 usec -> 2022-06-24T10:15:13.392.147.987 (XO: 19.392150078)
