-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 13:17:52 2023 UTC
-- File 20220624_095156_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.405922 to 37.206954
-each separately from 5s to 37s and from 206665us to 986516us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.580016 in 1 packets: 18
      18.220149 in 1 packets: 36
MIU packet time from 2022-06-24T09:52:04.405.922.000 to 2022-06-24T09:52:36.206.954.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.580016 in 1 packets: 18
      18.220149 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.1408615 to 39.1407427
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999517 in 1 packets: 18
      18.9999295 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_095156_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.5467848 to 7.5470738
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5799670  -- OK for SL --
   MIU time from 2022-06-24T09:52:04.405.922.000 to 2022-06-24T09:52:04.406.211.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.580016

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1270408 to 21.1273298
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2200790  -- OK for SL --
   MIU time from 2022-06-24T09:52:17.986.227.000 to 2022-06-24T09:52:17.986.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.220149

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3474088 to 39.3476978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T09:52:36.206.665.000 to 2022-06-24T09:52:36.206.954.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.546784800 corrected by 40bit 10MHz counter of 7.140861500 obtained at 1Hz counter of 5. Result: 5.405924802
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T09:52:04.405.781.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T09:52:36.206.813.725
   Min 1st sample XO time among all SL packets: 5.405784526
   Max 1st sample XO time among all SL packets: 37.206815590
   TM values from -253 to 211 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T09:52:04.405.781.725 (XO: 5.405784526 - 5.406090528, t=37.2 degC)  Rank: 00083
   TM values: Max =   211  Min =  -235  Ave =    -4.49  Std =    21.78
   Min TM value:-235 at an offset of 115.988 usec -> 2022-06-24T09:52:04.405.897.712 (XO: 5.405900514)
   Max TM value:211 at an offset of 113.513 usec -> 2022-06-24T09:52:04.405.895.237 (XO: 5.405898039)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 116.150 usec -> 2022-06-24T09:52:04.405.897.875 (XO: 5.405900677)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 113.638 usec -> 2022-06-24T09:52:04.405.895.362 (XO: 5.405898164)

 - Block number: 2 starting at 2022-06-24T09:52:17.986.086.725 (XO: 18.986090974 - 18.986396975, t=37.2 degC)  Rank: 00087
   TM values: Max =   205  Min =  -246  Ave =    -4.48  Std =    24.89
   Min TM value:-246 at an offset of 112.275 usec -> 2022-06-24T09:52:17.986.199.000 (XO: 18.986203249)
   Max TM value:205 at an offset of 109.800 usec -> 2022-06-24T09:52:17.986.196.525 (XO: 18.986200774)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 112.413 usec -> 2022-06-24T09:52:17.986.199.137 (XO: 18.986203386)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 109.925 usec -> 2022-06-24T09:52:17.986.196.650 (XO: 18.986200899)

 - Block number: 3 starting at 2022-06-24T09:52:36.206.524.725 (XO: 37.206526589 - 37.206832590, t=37.2 degC)  Rank: 00080
   TM values: Max =   190  Min =  -253  Ave =    -4.40  Std =    20.05
   Min TM value:-253 at an offset of 112.288 usec -> 2022-06-24T09:52:36.206.637.012 (XO: 37.206638877)
   Max TM value:190 at an offset of 109.788 usec -> 2022-06-24T09:52:36.206.634.512 (XO: 37.206636377)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 112.413 usec -> 2022-06-24T09:52:36.206.637.137 (XO: 37.206639002)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 109.888 usec -> 2022-06-24T09:52:36.206.634.612 (XO: 37.206636477)
