-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 12:51:41 2023 UTC
-- File 20220624_092938_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.137467 to 44.378800
-each separately from 4s to 44s and from 137467us to 378800us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.240125 in 1 packets: 18
      19.000341 in 1 packets: 36
MIU packet time from 2022-06-24T09:29:45.137.467.000 to 2022-06-24T09:30:25.378.800.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.240125 in 1 packets: 18
      19.000341 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.6433436 to 45.6431965
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999229 in 1 packets: 18
      18.9999300 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_092938_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.7808118 to 5.7811008
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2400470  -- OK for SL --
   MIU time from 2022-06-24T09:29:45.137.467.000 to 2022-06-24T09:29:45.137.756.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.240125

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0211478 to 27.0214368
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0002710  -- OK for SL --
   MIU time from 2022-06-24T09:30:06.377.881.000 to 2022-06-24T09:30:06.378.170.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.000341

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.0217078 to 46.0219968
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T09:30:25.378.511.000 to 2022-06-24T09:30:25.378.800.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 5.780811800 corrected by 40bit 10MHz counter of 5.643343600 obtained at 1Hz counter of 4. Result: 4.137468709
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T09:29:45.137.326.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T09:30:25.378.659.725
   Min 1st sample XO time among all SL packets: 4.137328433
   Max 1st sample XO time among all SL packets: 44.378661426
   TM values from -259 to 211 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T09:29:45.137.326.725 (XO: 4.137328433 - 4.137634434, t=37.2 degC)  Rank: 00078
   TM values: Max =   198  Min =  -259  Ave =    -4.58  Std =    21.58
   Min TM value:-259 at an offset of 115.875 usec -> 2022-06-24T09:29:45.137.442.600 (XO: 4.137444309)
   Max TM value:198 at an offset of 113.888 usec -> 2022-06-24T09:29:45.137.440.612 (XO: 4.137442321)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 118.225 usec -> 2022-06-24T09:29:45.137.444.950 (XO: 4.137446659)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 115.713 usec -> 2022-06-24T09:29:45.137.442.437 (XO: 4.137444146)

 - Block number: 2 starting at 2022-06-24T09:30:06.377.740.725 (XO: 25.377742423 - 25.378048424, t=37.2 degC)  Rank: 00078
   TM values: Max =   188  Min =  -257  Ave =    -4.56  Std =    20.60
   Min TM value:-257 at an offset of 133.300 usec -> 2022-06-24T09:30:06.377.874.025 (XO: 25.377875723)
   Max TM value:188 at an offset of 130.825 usec -> 2022-06-24T09:30:06.377.871.550 (XO: 25.377873248)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 133.413 usec -> 2022-06-24T09:30:06.377.874.137 (XO: 25.377875836)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 130.900 usec -> 2022-06-24T09:30:06.377.871.625 (XO: 25.377873323)

 - Block number: 3 starting at 2022-06-24T09:30:25.378.370.725 (XO: 44.378372425 - 44.378678426, t=37.2 degC)  Rank: 00079
   TM values: Max =   211  Min =  -254  Ave =    -4.54  Std =    27.94
   Min TM value:-254 at an offset of 130.288 usec -> 2022-06-24T09:30:25.378.501.012 (XO: 44.378502713)
   Max TM value:211 at an offset of 127.788 usec -> 2022-06-24T09:30:25.378.498.512 (XO: 44.378500213)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 130.400 usec -> 2022-06-24T09:30:25.378.501.125 (XO: 44.378502825)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 127.888 usec -> 2022-06-24T09:30:25.378.498.612 (XO: 44.378500313)
