-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 16:59:50 2023 UTC
-- File 20220624_005206_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.182846 to 100.441661
-each separately from 10s to 100s and from 1541us to 442857us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.119648 in 1 packets: 18
      11.139496 in 1 packets: 36
      26.709171 in 1 packets: 54
      9.289568 in 1 packets: 72
      14.559367 in 1 packets: 90
      19.439542 in 1 packets: 108
MIU packet time from 2022-06-24T00:52:19.182.846.000 to 2022-06-24T00:53:49.441.661.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.119648 in 1 packets: 18
      11.139496 in 1 packets: 36
      26.709171 in 1 packets: 54
      9.289568 in 1 packets: 72
      14.559367 in 1 packets: 90
      19.439542 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.6089916 to 101.6086579
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999667 in 2 packets: 18, 72
      10.9999592 in 1 packets: 36
      26.9998999 in 1 packets: 54
      14.9999442 in 1 packets: 90
      18.9999296 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_005206_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7918384 to 11.7921274
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1196150  -- OK for SL --
   MIU time from 2022-06-24T00:52:19.182.846.000 to 2022-06-24T00:52:19.183.135.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.119648

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.9117424 to 20.9120314
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1394550  -- OK for SL --
   MIU time from 2022-06-24T00:52:28.302.783.000 to 2022-06-24T00:52:28.303.072.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.139496

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0514864 to 32.0517754
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.7090710  -- OK for SL --
   MIU time from 2022-06-24T00:52:39.442.568.000 to 2022-06-24T00:52:39.442.857.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.709171

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.7608464 to 58.7611354
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2895350  -- OK for SL --
   MIU time from 2022-06-24T00:53:06.152.028.000 to 2022-06-24T00:53:06.152.317.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.289568

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.0506704 to 68.0509594
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5593110  -- OK for SL --
   MIU time from 2022-06-24T00:53:15.441.885.000 to 2022-06-24T00:53:15.442.174.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.559367

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.6102704 to 82.6105594
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4394710  -- OK for SL --
   MIU time from 2022-06-24T00:53:30.001.541.000 to 2022-06-24T00:53:30.001.830.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.439542

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 102.0500304 to 102.0503194
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T00:53:49.441.372.000 to 2022-06-24T00:53:49.441.661.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 11.791838400 corrected by 40bit 10MHz counter of 11.608991600 obtained at 1Hz counter of 10. Result: 10.182847495
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T00:52:19.182.705.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T00:53:49.441.520.725
   Min 1st sample XO time among all SL packets: 10.182707219
   Max 1st sample XO time among all SL packets: 100.441522859
   TM values from -244 to 193 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T00:52:19.182.705.725 (XO: 10.182707219 - 10.183013220, t=37.2 degC)  Rank: 00072
   TM values: Max =   189  Min =  -229  Ave =    -4.43  Std =    24.20
   Min TM value:-229 at an offset of 120.713 usec -> 2022-06-24T00:52:19.182.826.437 (XO: 10.182827932)
   Max TM value:189 at an offset of 118.225 usec -> 2022-06-24T00:52:19.182.823.950 (XO: 10.182825445)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 120.888 usec -> 2022-06-24T00:52:19.182.826.612 (XO: 10.182828107)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 118.375 usec -> 2022-06-24T00:52:19.182.824.100 (XO: 10.182825595)

 - Block number: 2 starting at 2022-06-24T00:52:28.302.642.725 (XO: 19.302644945 - 19.302950946, t=37.2 degC)  Rank: 00081
   TM values: Max =   192  Min =  -231  Ave =    -4.56  Std =    20.90
   Min TM value:-231 at an offset of 120.125 usec -> 2022-06-24T00:52:28.302.762.850 (XO: 19.302765070)
   Max TM value:192 at an offset of 117.663 usec -> 2022-06-24T00:52:28.302.760.387 (XO: 19.302762608)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 120.250 usec -> 2022-06-24T00:52:28.302.762.975 (XO: 19.302765195)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 117.788 usec -> 2022-06-24T00:52:28.302.760.512 (XO: 19.302762733)

 - Block number: 3 starting at 2022-06-24T00:52:39.442.427.725 (XO: 30.442430262 - 30.442736263, t=37.2 degC)  Rank: 00075
   TM values: Max =   155  Min =  -244  Ave =    -4.46  Std =    20.68
   Min TM value:-244 at an offset of 120.050 usec -> 2022-06-24T00:52:39.442.547.775 (XO: 30.442550312)
   Max TM value:155 at an offset of 117.488 usec -> 2022-06-24T00:52:39.442.545.212 (XO: 30.442547750)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 120.163 usec -> 2022-06-24T00:52:39.442.547.887 (XO: 30.442550425)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 117.638 usec -> 2022-06-24T00:52:39.442.545.362 (XO: 30.442547900)

 - Block number: 4 starting at 2022-06-24T00:53:06.151.887.725 (XO: 57.151889287 - 57.152195288, t=37.2 degC)  Rank: 00077
   TM values: Max =   171  Min =  -241  Ave =    -4.31  Std =    29.02
   Min TM value:-241 at an offset of 107.113 usec -> 2022-06-24T00:53:06.151.994.837 (XO: 57.151996400)
   Max TM value:171 at an offset of 105.125 usec -> 2022-06-24T00:53:06.151.992.850 (XO: 57.151994412)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 104.938 usec -> 2022-06-24T00:53:06.151.992.662 (XO: 57.151994225)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 50.763 usec -> 2022-06-24T00:53:06.151.938.487 (XO: 57.151940050)

 - Block number: 5 starting at 2022-06-24T00:53:15.441.744.725 (XO: 66.441747659 - 66.442053661, t=37.2 degC)  Rank: 00079
   TM values: Max =   193  Min =  -242  Ave =    -4.35  Std =    21.30
   Min TM value:-242 at an offset of 92.513 usec -> 2022-06-24T00:53:15.441.837.237 (XO: 66.441840172)
   Max TM value:193 at an offset of 90.538 usec -> 2022-06-24T00:53:15.441.835.262 (XO: 66.441838197)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 90.350 usec -> 2022-06-24T00:53:15.441.835.075 (XO: 66.441838010)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 80.413 usec -> 2022-06-24T00:53:15.441.825.137 (XO: 66.441828072)

 - Block number: 6 starting at 2022-06-24T00:53:30.001.400.725 (XO: 81.001401830 - 81.001707831, t=37.2 degC)  Rank: 00081
   TM values: Max =   153  Min =  -212  Ave =    -4.50  Std =    28.06
   Min TM value:-212 at an offset of 164.076 usec -> 2022-06-24T00:53:30.001.564.800 (XO: 81.001565906)
   Max TM value:153 at an offset of 188.638 usec -> 2022-06-24T00:53:30.001.589.363 (XO: 81.001590468)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 164.176 usec -> 2022-06-24T00:53:30.001.564.900 (XO: 81.001566006)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 190.713 usec -> 2022-06-24T00:53:30.001.591.438 (XO: 81.001592543)

 - Block number: 7 starting at 2022-06-24T00:53:49.441.231.725 (XO: 100.441233858 - 100.441539859, t=37.2 degC)  Rank: 00076
   TM values: Max =   189  Min =  -225  Ave =    -4.44  Std =    23.88
   Min TM value:-225 at an offset of 119.888 usec -> 2022-06-24T00:53:49.441.351.612 (XO: 100.441353746)
   Max TM value:189 at an offset of 119.638 usec -> 2022-06-24T00:53:49.441.351.362 (XO: 100.441353496)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 122.238 usec -> 2022-06-24T00:53:49.441.353.962 (XO: 100.441356096)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 119.725 usec -> 2022-06-24T00:53:49.441.351.450 (XO: 100.441353583)
