-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 13:13:32 2023 UTC
-- File 20220623_121434_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.782202 to 21.543052
-each separately from 12s to 21s and from 542763us to 782491us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.760272 in 1 packets: 18
MIU packet time from 2022-06-23T12:14:49.782.202.000 to 2022-06-23T12:14:58.543.052.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.760272 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.2502154 to 23.2501822
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999668 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220623_121434_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.0324178 to 15.0327068
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7602390  -- OK for SL --
   MIU time from 2022-06-23T12:14:49.782.202.000 to 2022-06-23T12:14:49.782.491.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.760272

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.7929458 to 23.7932348
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-23T12:14:58.542.763.000 to 2022-06-23T12:14:58.543.052.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 15.032417800 corrected by 40bit 10MHz counter of 14.250215400 obtained at 1Hz counter of 12. Result: 12.782205294
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-23T12:14:49.782.061.725
   Max 1st sample MIU time among all SL packets: 2022-06-23T12:14:58.542.911.725
   Min 1st sample XO time among all SL packets: 12.782065019
   Max 1st sample XO time among all SL packets: 21.542914279
   TM values from -273 to 213 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-23T12:14:49.782.061.725 (XO: 12.782065019 - 12.782371020, t=37.2 degC)  Rank: 00082
   TM values: Max =   183  Min =  -272  Ave =    -4.55  Std =    26.52
   Min TM value:-272 at an offset of 115.225 usec -> 2022-06-23T12:14:49.782.176.950 (XO: 12.782180244)
   Max TM value:183 at an offset of 112.675 usec -> 2022-06-23T12:14:49.782.174.400 (XO: 12.782177694)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 115.313 usec -> 2022-06-23T12:14:49.782.177.037 (XO: 12.782180332)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 112.800 usec -> 2022-06-23T12:14:49.782.174.525 (XO: 12.782177819)

 - Block number: 2 starting at 2022-06-23T12:14:58.542.622.725 (XO: 21.542625278 - 21.542931280, t=37.2 degC)  Rank: 00081
   TM values: Max =   213  Min =  -273  Ave =    -4.55  Std =    22.37
   Min TM value:-273 at an offset of 131.413 usec -> 2022-06-23T12:14:58.542.754.137 (XO: 21.542756691)
   Max TM value:213 at an offset of 128.938 usec -> 2022-06-23T12:14:58.542.751.662 (XO: 21.542754216)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 131.525 usec -> 2022-06-23T12:14:58.542.754.250 (XO: 21.542756804)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 129.050 usec -> 2022-06-23T12:14:58.542.751.775 (XO: 21.542754329)
