-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 07:08:25 2023 UTC
-- File 20220622_231115_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.316974 to 90.795199
-each separately from 5s to 90s and from 55652us to 996340us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.739255 in 1 packets: 18
      17.939244 in 1 packets: 36
      13.059312 in 1 packets: 54
      9.599504 in 1 packets: 72
      18.419299 in 1 packets: 90
      6.719588 in 1 packets: 108
MIU packet time from 2022-06-22T23:11:23.316.974.000 to 2022-06-22T23:12:48.795.199.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.739255 in 1 packets: 18
      17.939244 in 1 packets: 36
      13.059312 in 1 packets: 54
      9.599504 in 1 packets: 72
      18.419299 in 1 packets: 90
      6.719588 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.6198316 to 91.6195275
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999284 in 1 packets: 18
      16.9999392 in 1 packets: 36
      13.9999500 in 1 packets: 54
      8.9999678 in 1 packets: 72
      18.9999319 in 1 packets: 90
      5.9999786 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      05 -> 48.5 degC in 18 packets: 109, 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 108 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 108 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220622_231115_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9368067 to 6.9370957
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7391830  -- OK for SL --
   MIU time from 2022-06-22T23:11:23.316.974.000 to 2022-06-22T23:11:23.317.263.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.739255

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6762787 to 26.6765677
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9391830  -- OK for SL --
   MIU time from 2022-06-22T23:11:43.056.518.000 to 2022-06-22T23:11:43.056.807.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.939244

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.6157507 to 44.6160397
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0592630  -- OK for SL --
   MIU time from 2022-06-22T23:12:00.996.051.000 to 2022-06-22T23:12:00.996.340.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.059312

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.6753027 to 57.6755917
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5994710  -- OK for SL --
   MIU time from 2022-06-22T23:12:14.055.652.000 to 2022-06-22T23:12:14.055.941.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.599504

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.2750627 to 67.2753517
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4192310  -- OK for SL --
   MIU time from 2022-06-22T23:12:23.655.445.000 to 2022-06-22T23:12:23.655.734.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.419299

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 85.6945827 to 85.6948717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.7195670  -- OK for SL --
   MIU time from 2022-06-22T23:12:42.075.033.000 to 2022-06-22T23:12:42.075.322.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.719588

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 92.4144387 to 92.4147277
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-22T23:12:48.794.910.000 to 2022-06-22T23:12:48.795.199.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.936806700 corrected by 40bit 10MHz counter of 6.619831600 obtained at 1Hz counter of 5. Result: 5.316976241
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-22T23:11:23.316.833.725
   Max 1st sample MIU time among all SL packets: 2022-06-22T23:12:48.795.058.725
   Min 1st sample XO time among all SL packets: 5.316835966
   Max 1st sample XO time among all SL packets: 90.795062787
   TM values from -252 to 209 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-22T23:11:23.316.833.725 (XO: 5.316835966 - 5.317141967, t=37.2 degC)  Rank: 00083
   TM values: Max =   160  Min =  -228  Ave =    -4.53  Std =    18.75
   Min TM value:-228 at an offset of 130.463 usec -> 2022-06-22T23:11:23.316.964.187 (XO: 5.316966429)
   Max TM value:160 at an offset of 133.025 usec -> 2022-06-22T23:11:23.316.966.750 (XO: 5.316968991)
   Max dTM/dt [1/us]:1.36e+03 at an offset of 130.538 usec -> 2022-06-22T23:11:23.316.964.262 (XO: 5.316966504)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 127.988 usec -> 2022-06-22T23:11:23.316.961.712 (XO: 5.316963954)

 - Block number: 2 starting at 2022-06-22T23:11:43.056.377.725 (XO: 25.056378628 - 25.056684629, t=37.2 degC)  Rank: 00079
   TM values: Max =   157  Min =  -248  Ave =    -4.59  Std =    20.25
   Min TM value:-248 at an offset of 130.800 usec -> 2022-06-22T23:11:43.056.508.525 (XO: 25.056509428)
   Max TM value:157 at an offset of 133.525 usec -> 2022-06-22T23:11:43.056.511.250 (XO: 25.056512153)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 130.900 usec -> 2022-06-22T23:11:43.056.508.625 (XO: 25.056509528)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 128.363 usec -> 2022-06-22T23:11:43.056.506.087 (XO: 25.056506991)

 - Block number: 3 starting at 2022-06-22T23:12:00.995.910.725 (XO: 42.995914810 - 42.996220811, t=37.2 degC)  Rank: 00073
   TM values: Max =   155  Min =  -223  Ave =    -4.59  Std =    20.28
   Min TM value:-223 at an offset of 119.838 usec -> 2022-06-22T23:12:00.996.030.562 (XO: 42.996034648)
   Max TM value:155 at an offset of 122.500 usec -> 2022-06-22T23:12:00.996.033.225 (XO: 42.996037311)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 119.975 usec -> 2022-06-22T23:12:00.996.030.700 (XO: 42.996034786)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 122.675 usec -> 2022-06-22T23:12:00.996.033.400 (XO: 42.996037486)

 - Block number: 4 starting at 2022-06-22T23:12:14.055.511.725 (XO: 56.055513425 - 56.055819426, t=37.2 degC)  Rank: 00079
   TM values: Max =   209  Min =  -250  Ave =    -4.48  Std =    19.10
   Min TM value:-250 at an offset of 134.300 usec -> 2022-06-22T23:12:14.055.646.025 (XO: 56.055647725)
   Max TM value:209 at an offset of 131.863 usec -> 2022-06-22T23:12:14.055.643.587 (XO: 56.055645288)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 134.438 usec -> 2022-06-22T23:12:14.055.646.162 (XO: 56.055647863)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 131.950 usec -> 2022-06-22T23:12:14.055.643.675 (XO: 56.055645375)

 - Block number: 5 starting at 2022-06-22T23:12:23.655.304.725 (XO: 65.655307784 - 65.655613785, t=37.2 degC)  Rank: 00080
   TM values: Max =   164  Min =  -252  Ave =    -4.48  Std =    18.46
   Min TM value:-252 at an offset of 119.113 usec -> 2022-06-22T23:12:23.655.423.837 (XO: 65.655426897)
   Max TM value:164 at an offset of 116.563 usec -> 2022-06-22T23:12:23.655.421.287 (XO: 65.655424347)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 119.238 usec -> 2022-06-22T23:12:23.655.423.962 (XO: 65.655427022)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 116.725 usec -> 2022-06-22T23:12:23.655.421.450 (XO: 65.655424510)

 - Block number: 6 starting at 2022-06-22T23:12:42.074.892.725 (XO: 84.074893787 - 84.075199788, t=37.2 degC)  Rank: 00080
   TM values: Max =   182  Min =  -247  Ave =    -4.69  Std =    30.94
   Min TM value:-247 at an offset of 189.701 usec -> 2022-06-22T23:12:42.075.082.425 (XO: 84.075083488)
   Max TM value:182 at an offset of 231.238 usec -> 2022-06-22T23:12:42.075.123.963 (XO: 84.075125025)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 189.813 usec -> 2022-06-22T23:12:42.075.082.538 (XO: 84.075083600)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 231.351 usec -> 2022-06-22T23:12:42.075.124.075 (XO: 84.075125138)

 - Block number: 7 starting at 2022-06-22T23:12:48.794.769.725 (XO: 90.794773786 - 90.795079787, t=37.2 degC)  Rank: 00086
   TM values: Max =   167  Min =  -207  Ave =    -4.58  Std =    19.18
   Min TM value:-207 at an offset of 128.688 usec -> 2022-06-22T23:12:48.794.898.412 (XO: 90.794902474)
   Max TM value:167 at an offset of 116.250 usec -> 2022-06-22T23:12:48.794.885.975 (XO: 90.794890037)
   Max dTM/dt [1/us]:1.23e+03 at an offset of 131.175 usec -> 2022-06-22T23:12:48.794.900.900 (XO: 90.794904962)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 116.388 usec -> 2022-06-22T23:12:48.794.886.112 (XO: 90.794890174)
