-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 06:30:17 2023 UTC
-- File 20220622_195921_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.514499 to 46.964376
-each separately from 5s to 46s and from 514423us to 964376us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.999635 in 1 packets: 18
      26.319453 in 1 packets: 36
      5.129633 in 1 packets: 54
MIU packet time from 2022-06-22T19:59:29.514.499.000 to 2022-06-22T20:00:10.964.376.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.999635 in 1 packets: 18
      26.319453 in 1 packets: 36
      5.129633 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.6937210 to 47.6935732
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999640 in 1 packets: 18
      25.9999062 in 1 packets: 36
      4.9999820 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220622_195921_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2082206 to 7.2085096
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9995990  -- OK for SL --
   MIU time from 2022-06-22T19:59:29.514.499.000 to 2022-06-22T19:59:29.514.788.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.999635

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.2081086 to 17.2083976
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.3193590  -- OK for SL --
   MIU time from 2022-06-22T19:59:39.514.423.000 to 2022-06-22T19:59:39.514.712.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.319453

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.5277566 to 43.5280456
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1296150  -- OK for SL --
   MIU time from 2022-06-22T20:00:05.834.165.000 to 2022-06-22T20:00:05.834.454.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.129633

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.6576606 to 48.6579496
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-22T20:00:10.964.087.000 to 2022-06-22T20:00:10.964.376.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.208220600 corrected by 40bit 10MHz counter of 6.693721000 obtained at 1Hz counter of 5. Result: 5.514501452
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-22T19:59:29.514.358.725
   Max 1st sample MIU time among all SL packets: 2022-06-22T20:00:10.964.235.725
   Min 1st sample XO time among all SL packets: 5.514361177
   Max 1st sample XO time among all SL packets: 46.964239596
   TM values from -319 to 248 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-22T19:59:29.514.358.725 (XO: 5.514361177 - 5.514667178, t=37.2 degC)  Rank: 00076
   TM values: Max =   193  Min =  -269  Ave =    -4.53  Std =    24.02
   Min TM value:-269 at an offset of 121.550 usec -> 2022-06-22T19:59:29.514.480.275 (XO: 5.514482727)
   Max TM value:193 at an offset of 119.038 usec -> 2022-06-22T19:59:29.514.477.762 (XO: 5.514480215)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 121.688 usec -> 2022-06-22T19:59:29.514.480.412 (XO: 5.514482865)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 119.163 usec -> 2022-06-22T19:59:29.514.477.887 (XO: 5.514480340)

 - Block number: 2 starting at 2022-06-22T19:59:39.514.282.725 (XO: 15.514285176 - 15.514591178, t=37.2 degC)  Rank: 00081
   TM values: Max =   248  Min =  -319  Ave =    -4.62  Std =    27.78
   Min TM value:-319 at an offset of 86.400 usec -> 2022-06-22T19:59:39.514.369.125 (XO: 15.514371577)
   Max TM value:248 at an offset of 83.950 usec -> 2022-06-22T19:59:39.514.366.675 (XO: 15.514369127)
   Max dTM/dt [1/us]:2.69e+03 at an offset of 86.513 usec -> 2022-06-22T19:59:39.514.369.237 (XO: 15.514371689)
   Min dTM/dt [1/us]:-2.76e+03 at an offset of 84.075 usec -> 2022-06-22T19:59:39.514.366.800 (XO: 15.514369252)

 - Block number: 3 starting at 2022-06-22T20:00:05.834.024.725 (XO: 41.834028128 - 41.834334129, t=37.2 degC)  Rank: 00076
   TM values: Max =   238  Min =  -296  Ave =    -4.55  Std =    31.84
   Min TM value:-296 at an offset of 110.888 usec -> 2022-06-22T20:00:05.834.135.612 (XO: 41.834139015)
   Max TM value:238 at an offset of 111.100 usec -> 2022-06-22T20:00:05.834.135.825 (XO: 41.834139228)
   Max dTM/dt [1/us]:2.85e+03 at an offset of 110.963 usec -> 2022-06-22T20:00:05.834.135.687 (XO: 41.834139090)
   Min dTM/dt [1/us]:-2.96e+03 at an offset of 108.525 usec -> 2022-06-22T20:00:05.834.133.250 (XO: 41.834136653)

 - Block number: 4 starting at 2022-06-22T20:00:10.963.946.725 (XO: 46.963950595 - 46.964256596, t=37.2 degC)  Rank: 00082
   TM values: Max =   234  Min =  -295  Ave =    -4.59  Std =    34.14
   Min TM value:-295 at an offset of 123.150 usec -> 2022-06-22T20:00:10.964.069.875 (XO: 46.964073746)
   Max TM value:234 at an offset of 123.413 usec -> 2022-06-22T20:00:10.964.070.137 (XO: 46.964074008)
   Max dTM/dt [1/us]:2.64e+03 at an offset of 123.288 usec -> 2022-06-22T20:00:10.964.070.012 (XO: 46.964073883)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 120.800 usec -> 2022-06-22T20:00:10.964.067.525 (XO: 46.964071396)
