-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 05:40:07 2023 UTC
-- File 20220622_164033_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.775512 to 26.246246
-each separately from 9s to 26s and from 245957us to 775801us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.470156 in 1 packets: 18
MIU packet time from 2022-06-22T16:40:45.775.512.000 to 2022-06-22T16:41:02.246.246.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.470156 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.2269070 to 28.2268459
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999389 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220622_164033_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.0024197 to 12.0027087
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4700950  -- OK for SL --
   MIU time from 2022-06-22T16:40:45.775.512.000 to 2022-06-22T16:40:45.775.801.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.470156

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.4728037 to 28.4730927
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-22T16:41:02.245.957.000 to 2022-06-22T16:41:02.246.246.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 12.002419700 corrected by 40bit 10MHz counter of 11.226907000 obtained at 1Hz counter of 9. Result: 9.775515492
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-22T16:40:45.775.371.725
   Max 1st sample MIU time among all SL packets: 2022-06-22T16:41:02.246.105.725
   Min 1st sample XO time among all SL packets: 9.775375216
   Max 1st sample XO time among all SL packets: 26.246107411
   TM values from -284 to 258 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-22T16:40:45.775.371.725 (XO: 9.775375216 - 9.775681217, t=37.2 degC)  Rank: 00081
   TM values: Max =   258  Min =  -277  Ave =    -4.60  Std =    30.36
   Min TM value:-277 at an offset of 128.250 usec -> 2022-06-22T16:40:45.775.499.975 (XO: 9.775503467)
   Max TM value:258 at an offset of 125.763 usec -> 2022-06-22T16:40:45.775.497.487 (XO: 9.775500979)
   Max dTM/dt [1/us]:2.31e+03 at an offset of 128.388 usec -> 2022-06-22T16:40:45.775.500.112 (XO: 9.775503604)
   Min dTM/dt [1/us]:-2.71e+03 at an offset of 125.875 usec -> 2022-06-22T16:40:45.775.497.600 (XO: 9.775501092)

 - Block number: 2 starting at 2022-06-22T16:41:02.245.816.725 (XO: 26.245818410 - 26.246124411, t=37.2 degC)  Rank: 00082
   TM values: Max =   237  Min =  -284  Ave =    -4.58  Std =    32.10
   Min TM value:-284 at an offset of 128.713 usec -> 2022-06-22T16:41:02.245.945.437 (XO: 26.245947123)
   Max TM value:237 at an offset of 126.238 usec -> 2022-06-22T16:41:02.245.942.962 (XO: 26.245944648)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 128.825 usec -> 2022-06-22T16:41:02.245.945.550 (XO: 26.245947235)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 126.350 usec -> 2022-06-22T16:41:02.245.943.075 (XO: 26.245944760)
