-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 05:37:54 2023 UTC
-- File 20220622_163858_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.192935 to 15.973344
-each separately from 13s to 15s and from 192935us to 973344us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.779831 in 1 packets: 18
MIU packet time from 2022-06-22T16:39:14.192.935.000 to 2022-06-22T16:39:16.973.344.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.779831 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.1185223 to 17.1185151
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.9999928 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220622_163858_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.3114580 to 15.3117470
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.7798230  -- OK for SL --
   MIU time from 2022-06-22T16:39:14.192.935.000 to 2022-06-22T16:39:14.193.224.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.779831

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0915700 to 18.0918590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-22T16:39:16.973.055.000 to 2022-06-22T16:39:16.973.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.311458000 corrected by 40bit 10MHz counter of 15.118522300 obtained at 1Hz counter of 13. Result: 13.192936395
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-22T16:39:14.192.794.725
   Max 1st sample MIU time among all SL packets: 2022-06-22T16:39:16.973.203.725
   Min 1st sample XO time among all SL packets: 13.192796119
   Max 1st sample XO time among all SL packets: 15.973207031
   TM values from -285 to 242 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-22T16:39:14.192.794.725 (XO: 13.192796119 - 13.193102120, t=37.2 degC)  Rank: 00081
   TM values: Max =   242  Min =  -285  Ave =    -4.63  Std =    29.78
   Min TM value:-285 at an offset of 114.100 usec -> 2022-06-22T16:39:14.192.908.825 (XO: 13.192910219)
   Max TM value:242 at an offset of 111.600 usec -> 2022-06-22T16:39:14.192.906.325 (XO: 13.192907719)
   Max dTM/dt [1/us]:2.49e+03 at an offset of 114.200 usec -> 2022-06-22T16:39:14.192.908.925 (XO: 13.192910319)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 111.700 usec -> 2022-06-22T16:39:14.192.906.425 (XO: 13.192907819)

 - Block number: 2 starting at 2022-06-22T16:39:16.972.914.725 (XO: 15.972918030 - 15.973224031, t=37.2 degC)  Rank: 00082
   TM values: Max =   230  Min =  -284  Ave =    -4.64  Std =    22.02
   Min TM value:-284 at an offset of 111.950 usec -> 2022-06-22T16:39:16.973.026.675 (XO: 15.973029981)
   Max TM value:230 at an offset of 109.438 usec -> 2022-06-22T16:39:16.973.024.162 (XO: 15.973027468)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 112.050 usec -> 2022-06-22T16:39:16.973.026.775 (XO: 15.973030081)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 109.538 usec -> 2022-06-22T16:39:16.973.024.262 (XO: 15.973027568)
