-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 20:14:57 2023 UTC
-- File 20220621_150323_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.468247 to 54.409662
-each separately from 14s to 54s and from 28691us to 468536us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.560155 in 1 packets: 18
      10.079888 in 1 packets: 36
      17.300216 in 1 packets: 54
MIU packet time from 2022-06-21T15:03:40.468.247.000 to 2022-06-21T15:04:20.409.662.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.560155 in 1 packets: 18
      10.079888 in 1 packets: 36
      17.300216 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8181154 to 55.8179812
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999564 in 1 packets: 18
      9.9999665 in 1 packets: 36
      16.9999429 in 1 packets: 54
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FD -> 58.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      FE -> 57.6 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C6 -> 45.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220621_150323_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2863639 to 16.2866529
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5601110  -- OK for SL --
   MIU time from 2022-06-21T15:03:40.468.247.000 to 2022-06-21T15:03:40.468.536.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.560155

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8467639 to 28.8470529
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0798550  -- OK for SL --
   MIU time from 2022-06-21T15:03:53.028.691.000 to 2022-06-21T15:03:53.028.980.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.079888

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.9269079 to 38.9271969
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3001590  -- OK for SL --
   MIU time from 2022-06-21T15:04:03.108.868.000 to 2022-06-21T15:04:03.109.157.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.300216

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.2273559 to 56.2276449
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-21T15:04:20.409.373.000 to 2022-06-21T15:04:20.409.662.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.286363900 corrected by 40bit 10MHz counter of 15.818115400 obtained at 1Hz counter of 14. Result: 14.468250092
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-21T15:03:40.468.106.725
   Max 1st sample MIU time among all SL packets: 2022-06-21T15:04:20.409.521.725
   Min 1st sample XO time among all SL packets: 14.468109817
   Max 1st sample XO time among all SL packets: 54.409524776
   TM values from -283 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-21T15:03:40.468.106.725 (XO: 14.468109817 - 14.468415818, t=45.2 degC)  Rank: 00081
   TM values: Max =   238  Min =  -283  Ave =    -4.56  Std =    42.93
   Min TM value:-283 at an offset of 183.426 usec -> 2022-06-21T15:03:40.468.290.150 (XO: 14.468293242)
   Max TM value:238 at an offset of 181.488 usec -> 2022-06-21T15:03:40.468.288.213 (XO: 14.468291305)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 181.351 usec -> 2022-06-21T15:03:40.468.288.075 (XO: 14.468291167)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 200.038 usec -> 2022-06-21T15:03:40.468.306.763 (XO: 14.468309855)

 - Block number: 2 starting at 2022-06-21T15:03:53.028.550.725 (XO: 27.028551919 - 27.028857920, t=45.2 degC)  Rank: 00081
   TM values: Max =   210  Min =  -271  Ave =    -4.54  Std =    18.48
   Min TM value:-271 at an offset of 119.963 usec -> 2022-06-21T15:03:53.028.670.687 (XO: 27.028671882)
   Max TM value:210 at an offset of 119.663 usec -> 2022-06-21T15:03:53.028.670.387 (XO: 27.028671582)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 120.075 usec -> 2022-06-21T15:03:53.028.670.800 (XO: 27.028671995)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 119.763 usec -> 2022-06-21T15:03:53.028.670.487 (XO: 27.028671682)

 - Block number: 3 starting at 2022-06-21T15:04:03.108.727.725 (XO: 37.108729695 - 37.109035696, t=45.2 degC)  Rank: 00076
   TM values: Max =   211  Min =  -277  Ave =    -4.47  Std =    26.91
   Min TM value:-277 at an offset of 146.200 usec -> 2022-06-21T15:04:03.108.873.925 (XO: 37.108875895)
   Max TM value:211 at an offset of 121.563 usec -> 2022-06-21T15:04:03.108.849.287 (XO: 37.108851258)
   Max dTM/dt [1/us]:2.31e+03 at an offset of 148.263 usec -> 2022-06-21T15:04:03.108.875.988 (XO: 37.108877958)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 104.813 usec -> 2022-06-21T15:04:03.108.832.537 (XO: 37.108834508)

 - Block number: 4 starting at 2022-06-21T15:04:20.409.232.725 (XO: 54.409235775 - 54.409541776, t=45.2 degC)  Rank: 00080
   TM values: Max =   215  Min =  -240  Ave =    -4.44  Std =    31.52
   Min TM value:-240 at an offset of 246.026 usec -> 2022-06-21T15:04:20.409.478.750 (XO: 54.409481801)
   Max TM value:215 at an offset of 245.776 usec -> 2022-06-21T15:04:20.409.478.500 (XO: 54.409481551)
   Max dTM/dt [1/us]:1.34e+03 at an offset of 245.726 usec -> 2022-06-21T15:04:20.409.478.450 (XO: 54.409481501)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 245.926 usec -> 2022-06-21T15:04:20.409.478.650 (XO: 54.409481701)
