-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 20:14:58 2023 UTC
-- File 20220621_150323_1229.dat, 8656  packets found.
APID:
      1229 (EW) in 8656 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 8656 packets (all packets)
Total number of packets: 
      36720 in 8656 packets (all packets)
Packet No from 1 to 8656
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 14.468521 to 73.473271
-each separately from 14s to 73s and from 468521us to 473271us.
   Differences in neigboring packets: 
      0.000000 in 8654 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -59.004750 in 1 packets: 3876
MIU packet time from 2022-06-21T15:03:40.468.521.000 to 2022-06-21T15:04:39.473.271.000
   Differences in neigboring packets: 
      0.000000 in 8654 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -59.004750 in 1 packets: 3876
MIU orbit number: 
      60000 in 8656 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 8656 packets (all packets)
MIU orbit information / number of events: 
      0 in 8656 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 8656 packets (all packets)
Instrument anomalies: 
      0 (OK) in 8656 packets (all packets)
FPGA configuration version: 
      211 in 8656 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8181154 to 74.8179174
   Differences in neigboring packets: 
      0.0000000 in 8654 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -58.9998020 in 1 packets: 3876
Temperature FPGA: 
      FE -> 57.6 degC in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FF -> 56.3 degC in 4780 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
Temperature XO: 
      C6 -> 45.2 degC in 8656 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 8656 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 8656 packets (all packets)
Number of external events: 
      3 in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 4780 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
Number of generated alerts: 
      3 in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 4780 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
Survey memory error: 
      0 in 8656 packets (all packets)
Event memory error: 
      0 in 8656 packets (all packets)
Configuration error: 
      0 in 8656 packets (all packets)
SDRAM CRC error: 
      0 in 8656 packets (all packets)
Radio gain: 
      0 in 8656 packets (all packets)
Radio bandwidth: 
      0 in 8656 packets (all packets)
Digital gain: 
      0 in 8656 packets (all packets)
Analog gain: 
      0 in 8656 packets (all packets)

EVENT DATA READER
-- File 20220621_150323_1229.dat, 42 EW pages found. Extra packets: 88.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 4692 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 16.2311498 to 75.4437648
      Difference between the neighboring packets: 
      0.0000170 in 8566 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -59.2126150 in 1 packets: 3876
   Memory page number from 0 to 99
      Difference between the neighboring packets: 
      0 in 8526 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 40 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      59 in 1 packets: 3876
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 8526 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 40 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -59 in 1 packets: 3876
   MIU event flag 
      1 in 8568 packets (all packets)
   Time triggered event flag 
      0 in 8568 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-06-21T15:03:40.413.032.637
   Min EW header abs. time found among all events            : 2022-06-21T15:03:40.468.521.000
   Max EW event (1st sample) abs. time found among all events: 2022-06-21T15:04:39.417.782.637
   Max EW header abs. time found among all events            : 2022-06-21T15:04:39.473.271.000
   TM values from -283 to 249 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 4692
    Page no. 1:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 23  ordered packets: 4692
   TM values from -283 to 238 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=14 C1MHz=0.468521
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 4691 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 16.2311498
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-06-21T15:03:40.468.521.000
   Event 1st sample abs. time : 2022-06-21T15:03:40.413.032.637)
   --- Rel. time correction for 1 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.231149800 corrected by 40bit 10MHz counter of 15.818115400 obtained at 1Hz counter of 14. Result: 14.413035804
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              14.413035629
   Event data descriptor XO time (reconstructed seconds from XP start)    :              14.413035804
   Sampling frequency correction factor for this event:     0.999996600

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 3876
    Page no. 42:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 19  ordered packets: 3876
   TM values from -275 to 249 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=73 C1MHz=0.473271
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 3875 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 75.3778898
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-06-21T15:04:39.473.271.000
   Event 1st sample abs. time : 2022-06-21T15:04:39.417.782.637)
   --- Rel. time correction for 1 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 73, 32bit 10MHz counter = 75.377889800 corrected by 40bit 10MHz counter of 59.817967900 obtained at 1Hz counter of 58. Result: 73.559974804
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              73.559974629
   Event data descriptor XO time (reconstructed seconds from XP start)    :              73.559974804
   Sampling frequency correction factor for this event:     0.999996600
--
Extremes in separate events:

 - Event number: 1 from 2022-06-21T15:03:40.413.032.637 to 2022-06-21T15:03:40.492.796.908 (XO: 14.413035629 - 14.492799901)
   Min TM value:-283 at an offset of 55257.613 usec -> 2022-06-21T15:03:40.468.290.249 (XO: 14.468293242)
   Max TM value:238 at an offset of 55255.675 usec -> 2022-06-21T15:03:40.468.288.312 (XO: 14.468291305)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 65307.372 usec -> 2022-06-21T15:03:40.478.340.009 (XO: 14.478343001)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 55274.225 usec -> 2022-06-21T15:03:40.468.306.862 (XO: 14.468309855)
   Med: -4.00000   Ave: -4.4631674   Std: 8.2452954

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2022-06-21T15:04:39.417.782.637 to 2022-06-21T15:04:39.483.674.861 (XO: 73.559974629 - 73.625866853)
   Min TM value:-275 at an offset of 29857.014 usec -> 2022-06-21T15:04:39.447.639.651 (XO: 73.589831643)
   Max TM value:249 at an offset of 50000.708 usec -> 2022-06-21T15:04:39.467.783.344 (XO: 73.609975336)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 9824.671 usec -> 2022-06-21T15:04:39.427.607.307 (XO: 73.569799300)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 50000.808 usec -> 2022-06-21T15:04:39.467.783.444 (XO: 73.609975436)
   Med: -5.00000   Ave: -4.5184057   Std: 8.6376789
