-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 19:50:26 2023 UTC
-- File 20220621_142759_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.744374 to 82.929053
-each separately from 4s to 82s and from 62088us to 929053us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.099172 in 1 packets: 18
      22.808717 in 1 packets: 36
      8.408958 in 1 packets: 54
      21.397893 in 1 packets: 72
      14.468205 in 1 packets: 90
MIU packet time from 2022-06-21T14:28:06.744.374.000 to 2022-06-21T14:29:24.929.053.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.099172 in 1 packets: 18
      22.808717 in 1 packets: 36
      8.408958 in 1 packets: 54
      21.397893 in 1 packets: 72
      14.468205 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.2502495 to 84.2499868
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999629 in 1 packets: 18
      22.9999224 in 1 packets: 36
      8.9999697 in 1 packets: 54
      20.9999293 in 1 packets: 72
      13.9999530 in 1 packets: 90
Temperature FPGA: 
      FE -> 57.6 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FD -> 58.9 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      FC -> 60.2 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature XO: 
      C6 -> 45.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220621_142759_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9946255 to 6.9949145
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0991350  -- OK for SL --
   MIU time from 2022-06-21T14:28:06.744.374.000 to 2022-06-21T14:28:06.744.663.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.099172

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0940495 to 18.0943385
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.8086390  -- OK for SL --
   MIU time from 2022-06-21T14:28:17.843.835.000 to 2022-06-21T14:28:17.844.124.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.808717

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.9029775 to 40.9032665
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4089270  -- OK for SL --
   MIU time from 2022-06-21T14:28:40.652.841.000 to 2022-06-21T14:28:40.653.130.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.408958

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.3121935 to 49.3124825
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3978230  -- OK for SL --
   MIU time from 2022-06-21T14:28:49.062.088.000 to 2022-06-21T14:28:49.062.377.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.397893

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.7103055 to 70.7105945
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4681590  -- OK for SL --
   MIU time from 2022-06-21T14:29:10.460.270.000 to 2022-06-21T14:29:10.460.559.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.468205

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 85.1787535 to 85.1790425
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-21T14:29:24.928.764.000 to 2022-06-21T14:29:24.929.053.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.994625500 corrected by 40bit 10MHz counter of 6.250249500 obtained at 1Hz counter of 4. Result: 4.744378531
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-21T14:28:06.744.233.725
   Max 1st sample MIU time among all SL packets: 2022-06-21T14:29:24.928.912.725
   Min 1st sample XO time among all SL packets: 4.744238255
   Max 1st sample XO time among all SL packets: 82.928918583
   TM values from -279 to 241 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-21T14:28:06.744.233.725 (XO: 4.744238255 - 4.744544256, t=45.2 degC)  Rank: 00087
   TM values: Max =   223  Min =  -279  Ave =    -4.44  Std =    32.67
   Min TM value:-279 at an offset of 175.238 usec -> 2022-06-21T14:28:06.744.408.963 (XO: 4.744413494)
   Max TM value:223 at an offset of 119.675 usec -> 2022-06-21T14:28:06.744.353.400 (XO: 4.744357931)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 177.301 usec -> 2022-06-21T14:28:06.744.411.025 (XO: 4.744415556)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 151.138 usec -> 2022-06-21T14:28:06.744.384.863 (XO: 4.744389393)

 - Block number: 2 starting at 2022-06-21T14:28:17.843.694.725 (XO: 15.843699694 - 15.844005695, t=45.2 degC)  Rank: 00075
   TM values: Max =   221  Min =  -229  Ave =    -4.33  Std =    24.23
   Min TM value:-229 at an offset of 124.225 usec -> 2022-06-21T14:28:17.843.818.950 (XO: 15.843823919)
   Max TM value:221 at an offset of 126.475 usec -> 2022-06-21T14:28:17.843.821.200 (XO: 15.843826169)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 126.313 usec -> 2022-06-21T14:28:17.843.821.037 (XO: 15.843826007)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 123.988 usec -> 2022-06-21T14:28:17.843.818.712 (XO: 15.843823682)

 - Block number: 3 starting at 2022-06-21T14:28:40.652.700.725 (XO: 38.652704579 - 38.653010580, t=45.2 degC)  Rank: 00074
   TM values: Max =   231  Min =  -269  Ave =    -4.36  Std =    21.29
   Min TM value:-269 at an offset of 115.338 usec -> 2022-06-21T14:28:40.652.816.062 (XO: 38.652819917)
   Max TM value:231 at an offset of 117.588 usec -> 2022-06-21T14:28:40.652.818.312 (XO: 38.652822167)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 117.475 usec -> 2022-06-21T14:28:40.652.818.200 (XO: 38.652822054)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 115.163 usec -> 2022-06-21T14:28:40.652.815.887 (XO: 38.652819742)

 - Block number: 4 starting at 2022-06-21T14:28:49.061.947.725 (XO: 47.061948929 - 47.062254930, t=45.2 degC)  Rank: 00082
   TM values: Max =   241  Min =  -278  Ave =    -4.46  Std =    31.56
   Min TM value:-278 at an offset of 118.563 usec -> 2022-06-21T14:28:49.062.066.287 (XO: 47.062067492)
   Max TM value:241 at an offset of 118.288 usec -> 2022-06-21T14:28:49.062.066.012 (XO: 47.062067217)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 75.100 usec -> 2022-06-21T14:28:49.062.022.825 (XO: 47.062024030)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 118.438 usec -> 2022-06-21T14:28:49.062.066.162 (XO: 47.062067367)

 - Block number: 5 starting at 2022-06-21T14:29:10.460.129.725 (XO: 68.460132989 - 68.460438990, t=45.2 degC)  Rank: 00076
   TM values: Max =   231  Min =  -265  Ave =    -4.36  Std =    37.96
   Min TM value:-265 at an offset of 199.801 usec -> 2022-06-21T14:29:10.460.329.525 (XO: 68.460332790)
   Max TM value:231 at an offset of 174.376 usec -> 2022-06-21T14:29:10.460.304.100 (XO: 68.460307365)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 73.825 usec -> 2022-06-21T14:29:10.460.203.550 (XO: 68.460206815)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 187.563 usec -> 2022-06-21T14:29:10.460.317.288 (XO: 68.460320553)

 - Block number: 6 starting at 2022-06-21T14:29:24.928.623.725 (XO: 82.928629582 - 82.928935583, t=45.2 degC)  Rank: 00076
   TM values: Max =   228  Min =  -236  Ave =    -4.44  Std =    36.34
   Min TM value:-236 at an offset of 272.838 usec -> 2022-06-21T14:29:24.928.896.563 (XO: 82.928902421)
   Max TM value:228 at an offset of 128.038 usec -> 2022-06-21T14:29:24.928.751.762 (XO: 82.928757620)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 275.001 usec -> 2022-06-21T14:29:24.928.898.725 (XO: 82.928904583)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 248.913 usec -> 2022-06-21T14:29:24.928.872.638 (XO: 82.928878496)
