-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 11:32:34 2023 UTC
-- File 20220619_144612_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.374048 to 46.636995
-each separately from 1s to 46s and from 374048us to 895294us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.520668 in 1 packets: 18
      26.611143 in 1 packets: 36
      4.129980 in 1 packets: 54
MIU packet time from 2022-06-19T14:46:16.374.048.000 to 2022-06-19T14:47:01.636.995.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.520668 in 1 packets: 18
      26.611143 in 1 packets: 36
      4.129980 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.1032442 to 48.1030982
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999546 in 1 packets: 18
      26.9999124 in 1 packets: 36
      3.9999870 in 1 packets: 54
Temperature FPGA: 
      FF -> 56.3 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C6 -> 45.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220619_144612_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4772939 to 3.4775829
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5206230  -- OK for SL --
   MIU time from 2022-06-19T14:46:16.374.048.000 to 2022-06-19T14:46:16.374.337.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.520668

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9982059 to 17.9984949
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.6110550  -- OK for SL --
   MIU time from 2022-06-19T14:46:30.895.005.000 to 2022-06-19T14:46:30.895.294.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.611143

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.6095499 to 44.6098389
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.1299670  -- OK for SL --
   MIU time from 2022-06-19T14:46:57.506.437.000 to 2022-06-19T14:46:57.506.726.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.129980

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.7398059 to 48.7400949
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-19T14:47:01.636.706.000 to 2022-06-19T14:47:01.636.995.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.477293900 corrected by 40bit 10MHz counter of 3.103244200 obtained at 1Hz counter of 1. Result: 1.374050897
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996800 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2022-06-19T14:46:16.373.907.725
   Max 1st sample MIU time among all SL packets: 2022-06-19T14:47:01.636.854.725
   Min 1st sample XO time among all SL packets: 1.373910622
   Max 1st sample XO time among all SL packets: 46.636858463
   TM values from -267 to 241 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-19T14:46:16.373.907.725 (XO: 1.373910622 - 1.374216622, t=45.2 degC)  Rank: 00080
   TM values: Max =   241  Min =  -266  Ave =    -4.55  Std =    36.62
   Min TM value:-266 at an offset of 130.663 usec -> 2022-06-19T14:46:16.374.038.387 (XO: 1.374041284)
   Max TM value:241 at an offset of 204.651 usec -> 2022-06-19T14:46:16.374.112.375 (XO: 1.374115272)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 176.276 usec -> 2022-06-19T14:46:16.374.084.000 (XO: 1.374086897)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 189.726 usec -> 2022-06-19T14:46:16.374.097.450 (XO: 1.374100347)

 - Block number: 2 starting at 2022-06-19T14:46:30.894.864.725 (XO: 15.894869689 - 15.895175690, t=45.2 degC)  Rank: 00084
   TM values: Max =   223  Min =  -267  Ave =    -4.51  Std =    26.38
   Min TM value:-267 at an offset of 172.426 usec -> 2022-06-19T14:46:30.895.037.150 (XO: 15.895042114)
   Max TM value:223 at an offset of 115.725 usec -> 2022-06-19T14:46:30.894.980.450 (XO: 15.894985414)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 115.613 usec -> 2022-06-19T14:46:30.894.980.337 (XO: 15.894985301)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 172.301 usec -> 2022-06-19T14:46:30.895.037.025 (XO: 15.895041989)

 - Block number: 3 starting at 2022-06-19T14:46:57.506.296.725 (XO: 42.506300045 - 42.506606046, t=45.2 degC)  Rank: 00077
   TM values: Max =   207  Min =  -229  Ave =    -4.54  Std =    27.55
   Min TM value:-229 at an offset of 130.100 usec -> 2022-06-19T14:46:57.506.426.825 (XO: 42.506430146)
   Max TM value:207 at an offset of 129.825 usec -> 2022-06-19T14:46:57.506.426.550 (XO: 42.506429871)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 219.563 usec -> 2022-06-19T14:46:57.506.516.288 (XO: 42.506519608)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 129.925 usec -> 2022-06-19T14:46:57.506.426.650 (XO: 42.506429971)

 - Block number: 4 starting at 2022-06-19T14:47:01.636.565.725 (XO: 46.636569462 - 46.636875463, t=45.2 degC)  Rank: 00083
   TM values: Max =   210  Min =  -236  Ave =    -4.45  Std =    26.15
   Min TM value:-236 at an offset of 139.750 usec -> 2022-06-19T14:47:01.636.705.475 (XO: 46.636709212)
   Max TM value:210 at an offset of 115.263 usec -> 2022-06-19T14:47:01.636.680.987 (XO: 46.636684725)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 112.913 usec -> 2022-06-19T14:47:01.636.678.637 (XO: 46.636682375)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 115.388 usec -> 2022-06-19T14:47:01.636.681.112 (XO: 46.636684850)
