-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 04:46:33 2023 UTC
-- File 20220618_131858_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.914567 to 33.444913
-each separately from 1s to 33s and from 444624us to 914856us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.579709 in 1 packets: 18
      12.949770 in 1 packets: 36
MIU packet time from 2022-06-18T13:19:02.914.567.000 to 2022-06-18T13:19:34.444.913.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.579709 in 1 packets: 18
      12.949770 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.2554270 to 35.2553232
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999384 in 1 packets: 18
      12.9999578 in 1 packets: 36
Temperature FPGA: 
      FE -> 57.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220618_131858_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1699966 to 4.1702856
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5796470  -- OK for SL --
   MIU time from 2022-06-18T13:19:02.914.567.000 to 2022-06-18T13:19:02.914.856.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.579709

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7499326 to 22.7502216
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9497270  -- OK for SL --
   MIU time from 2022-06-18T13:19:21.494.565.000 to 2022-06-18T13:19:21.494.854.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.949770

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6999486 to 35.7002376
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-18T13:19:34.444.624.000 to 2022-06-18T13:19:34.444.913.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 4.169996600 corrected by 40bit 10MHz counter of 3.255427000 obtained at 1Hz counter of 1. Result: 1.914572527
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2022-06-18T13:19:02.914.426.725
   Max 1st sample MIU time among all SL packets: 2022-06-18T13:19:34.444.772.725
   Min 1st sample XO time among all SL packets: 1.914432251
   Max 1st sample XO time among all SL packets: 33.444775548
   TM values from -271 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-18T13:19:02.914.426.725 (XO: 1.914432251 - 1.914738252, t=45.2 degC)  Rank: 00082
   TM values: Max =   210  Min =  -242  Ave =    -4.51  Std =    26.39
   Min TM value:-242 at an offset of 122.350 usec -> 2022-06-18T13:19:02.914.549.075 (XO: 1.914554602)
   Max TM value:210 at an offset of 124.625 usec -> 2022-06-18T13:19:02.914.551.350 (XO: 1.914556877)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 124.475 usec -> 2022-06-18T13:19:02.914.551.200 (XO: 1.914556727)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 126.975 usec -> 2022-06-18T13:19:02.914.553.700 (XO: 1.914559227)

 - Block number: 2 starting at 2022-06-18T13:19:21.494.424.725 (XO: 20.494428557 - 20.494734558, t=45.2 degC)  Rank: 00080
   TM values: Max =   221  Min =  -271  Ave =    -4.49  Std =    23.52
   Min TM value:-271 at an offset of 149.375 usec -> 2022-06-18T13:19:21.494.574.100 (XO: 20.494577932)
   Max TM value:221 at an offset of 125.863 usec -> 2022-06-18T13:19:21.494.550.587 (XO: 20.494554420)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 151.676 usec -> 2022-06-18T13:19:21.494.576.400 (XO: 20.494580232)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 149.263 usec -> 2022-06-18T13:19:21.494.573.987 (XO: 20.494577820)

 - Block number: 3 starting at 2022-06-18T13:19:34.444.483.725 (XO: 33.444486547 - 33.444792548, t=45.2 degC)  Rank: 00079
   TM values: Max =   222  Min =  -230  Ave =    -4.52  Std =    29.09
   Min TM value:-230 at an offset of 126.288 usec -> 2022-06-18T13:19:34.444.610.012 (XO: 33.444612835)
   Max TM value:222 at an offset of 115.000 usec -> 2022-06-18T13:19:34.444.598.725 (XO: 33.444601548)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 141.388 usec -> 2022-06-18T13:19:34.444.625.112 (XO: 33.444627935)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 115.100 usec -> 2022-06-18T13:19:34.444.598.825 (XO: 33.444601648)
