-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 23:11:24 2023 UTC
-- File 20220617_130011_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.602343 to 68.132904
-each separately from 0s to 68s and from 132615us to 782516us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.179595 in 1 packets: 18
      10.519716 in 1 packets: 36
      7.019782 in 1 packets: 54
      21.810023 in 1 packets: 72
MIU packet time from 2022-06-17T13:00:14.602.343.000 to 2022-06-17T13:01:22.132.904.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.179595 in 1 packets: 18
      10.519716 in 1 packets: 36
      7.019782 in 1 packets: 54
      21.810023 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7484416 to 69.7482187
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999084 in 1 packets: 18
      10.9999637 in 1 packets: 36
      6.9999771 in 1 packets: 54
      21.9999279 in 1 packets: 72
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C6 -> 45.2 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C5 -> 43.6 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220617_130011_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3507869 to 2.3510759
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.1795030  -- OK for SL --
   MIU time from 2022-06-17T13:00:14.602.343.000 to 2022-06-17T13:00:14.602.632.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.179595

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.5305789 to 30.5308679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5196790  -- OK for SL --
   MIU time from 2022-06-17T13:00:42.782.227.000 to 2022-06-17T13:00:42.782.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.519716

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.0505469 to 41.0508359
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.0197590  -- OK for SL --
   MIU time from 2022-06-17T13:00:53.302.232.000 to 2022-06-17T13:00:53.302.521.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.019782

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.0705949 to 48.0708839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8099510  -- OK for SL --
   MIU time from 2022-06-17T13:01:00.322.303.000 to 2022-06-17T13:01:00.322.592.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.810023

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.8808349 to 69.8811239
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-17T13:01:22.132.615.000 to 2022-06-17T13:01:22.132.904.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.350786900 corrected by 40bit 10MHz counter of 1.748441600 obtained at 1Hz counter of 0. Result: 0.602347288
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2022-06-17T13:00:14.602.202.725
   Max 1st sample MIU time among all SL packets: 2022-06-17T13:01:22.132.763.725
   Min 1st sample XO time among all SL packets: 0.602207012
   Max 1st sample XO time among all SL packets: 68.132765363
   TM values from -280 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-17T13:00:14.602.202.725 (XO: 0.602207012 - 0.602513013, t=45.2 degC)  Rank: 00081
   TM values: Max =   227  Min =  -224  Ave =    -4.44  Std =    29.19
   Min TM value:-224 at an offset of 109.288 usec -> 2022-06-17T13:00:14.602.312.012 (XO: 0.602316300)
   Max TM value:227 at an offset of 153.751 usec -> 2022-06-17T13:00:14.602.356.475 (XO: 0.602360763)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 153.601 usec -> 2022-06-17T13:00:14.602.356.325 (XO: 0.602360613)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 184.863 usec -> 2022-06-17T13:00:14.602.387.588 (XO: 0.602391875)

 - Block number: 2 starting at 2022-06-17T13:00:42.782.086.725 (XO: 28.782091206 - 28.782397207, t=45.2 degC)  Rank: 00079
   TM values: Max =   209  Min =  -280  Ave =    -4.48  Std =    23.38
   Min TM value:-280 at an offset of 149.438 usec -> 2022-06-17T13:00:42.782.236.162 (XO: 28.782240644)
   Max TM value:209 at an offset of 125.800 usec -> 2022-06-17T13:00:42.782.212.525 (XO: 28.782217006)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 151.701 usec -> 2022-06-17T13:00:42.782.238.425 (XO: 28.782242906)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 125.925 usec -> 2022-06-17T13:00:42.782.212.650 (XO: 28.782217131)

 - Block number: 3 starting at 2022-06-17T13:00:53.302.091.725 (XO: 39.302093892 - 39.302399893, t=43.6 degC)  Rank: 00079
   TM values: Max =   219  Min =  -233  Ave =    -4.50  Std =    30.62
   Min TM value:-233 at an offset of 67.325 usec -> 2022-06-17T13:00:53.302.159.050 (XO: 39.302161217)
   Max TM value:219 at an offset of 52.950 usec -> 2022-06-17T13:00:53.302.144.675 (XO: 39.302146842)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 52.825 usec -> 2022-06-17T13:00:53.302.144.550 (XO: 39.302146717)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 112.175 usec -> 2022-06-17T13:00:53.302.203.900 (XO: 39.302206067)

 - Block number: 4 starting at 2022-06-17T13:01:00.322.162.725 (XO: 46.322164888 - 46.322470889, t=45.2 degC)  Rank: 00082
   TM values: Max =   206  Min =  -264  Ave =    -4.42  Std =    30.78
   Min TM value:-264 at an offset of 119.950 usec -> 2022-06-17T13:01:00.322.282.675 (XO: 46.322284839)
   Max TM value:206 at an offset of 122.250 usec -> 2022-06-17T13:01:00.322.284.975 (XO: 46.322287139)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 192.338 usec -> 2022-06-17T13:01:00.322.355.063 (XO: 46.322357226)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 189.938 usec -> 2022-06-17T13:01:00.322.352.663 (XO: 46.322354826)

 - Block number: 5 starting at 2022-06-17T13:01:22.132.474.725 (XO: 68.132476362 - 68.132782363, t=45.2 degC)  Rank: 00078
   TM values: Max =   235  Min =  -225  Ave =    -4.34  Std =    31.45
   Min TM value:-225 at an offset of 123.863 usec -> 2022-06-17T13:01:22.132.598.587 (XO: 68.132600225)
   Max TM value:235 at an offset of 126.213 usec -> 2022-06-17T13:01:22.132.600.937 (XO: 68.132602575)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 126.088 usec -> 2022-06-17T13:01:22.132.600.812 (XO: 68.132602450)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 123.713 usec -> 2022-06-17T13:01:22.132.598.437 (XO: 68.132600075)
