-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 22:11:15 2023 UTC
-- File 20220617_104817_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.496021 to 19.006426
-each separately from 8s to 19s and from 6137us to 496310us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.509827 in 1 packets: 18
MIU packet time from 2022-06-17T10:48:28.496.021.000 to 2022-06-17T10:48:39.006.426.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.509827 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.0995605 to 21.0995249
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999644 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220617_104817_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5955834 to 10.5958724
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5097910  -- OK for SL --
   MIU time from 2022-06-17T10:48:28.496.021.000 to 2022-06-17T10:48:28.496.310.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.509827

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1056634 to 21.1059524
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-17T10:48:39.006.137.000 to 2022-06-17T10:48:39.006.426.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.595583400 corrected by 40bit 10MHz counter of 10.099560500 obtained at 1Hz counter of 8. Result: 8.496024537
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-17T10:48:28.495.880.725
   Max 1st sample MIU time among all SL packets: 2022-06-17T10:48:39.006.285.725
   Min 1st sample XO time among all SL packets: 8.495884261
   Max 1st sample XO time among all SL packets: 19.006287246
   TM values from -221 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-17T10:48:28.495.880.725 (XO: 8.495884261 - 8.496190262, t=43.6 degC)  Rank: 00084
   TM values: Max =   201  Min =  -221  Ave =    -4.47  Std =    29.63
   Min TM value:-221 at an offset of 96.738 usec -> 2022-06-17T10:48:28.495.977.462 (XO: 8.495980999)
   Max TM value:201 at an offset of 96.475 usec -> 2022-06-17T10:48:28.495.977.200 (XO: 8.495980737)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 151.976 usec -> 2022-06-17T10:48:28.496.032.700 (XO: 8.496036237)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 154.438 usec -> 2022-06-17T10:48:28.496.035.163 (XO: 8.496038699)

 - Block number: 2 starting at 2022-06-17T10:48:39.005.996.725 (XO: 19.005998245 - 19.006304246, t=43.6 degC)  Rank: 00087
   TM values: Max =   195  Min =  -221  Ave =    -4.42  Std =    19.39
   Min TM value:-221 at an offset of 115.150 usec -> 2022-06-17T10:48:39.006.111.875 (XO: 19.006113395)
   Max TM value:195 at an offset of 117.550 usec -> 2022-06-17T10:48:39.006.114.275 (XO: 19.006115795)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 117.438 usec -> 2022-06-17T10:48:39.006.114.162 (XO: 19.006115683)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 103.500 usec -> 2022-06-17T10:48:39.006.100.225 (XO: 19.006101745)
