-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 19:20:58 2023 UTC
-- File 20220616_145054_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.151281 to 15.011516
-each separately from 12s to 15s and from 11227us to 151570us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.859657 in 1 packets: 18
MIU packet time from 2022-06-16T14:51:09.151.281.000 to 2022-06-16T14:51:12.011.516.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.859657 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7539857 to 16.7539758
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999901 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FE -> 57.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220616_145054_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9052684 to 13.9055574
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.8596470  -- OK for SL --
   MIU time from 2022-06-16T14:51:09.151.281.000 to 2022-06-16T14:51:09.151.570.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.859657

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7652044 to 16.7654934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-16T14:51:12.011.227.000 to 2022-06-16T14:51:12.011.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.905268400 corrected by 40bit 10MHz counter of 13.753985700 obtained at 1Hz counter of 12. Result: 12.151283199
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-16T14:51:09.151.140.725
   Max 1st sample MIU time among all SL packets: 2022-06-16T14:51:12.011.375.725
   Min 1st sample XO time among all SL packets: 12.151142924
   Max 1st sample XO time among all SL packets: 15.011377363
   TM values from -288 to 237 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-16T14:51:09.151.140.725 (XO: 12.151142924 - 12.151448925, t=43.6 degC)  Rank: 00077
   TM values: Max =   223  Min =  -288  Ave =    -4.45  Std =    35.35
   Min TM value:-288 at an offset of 130.913 usec -> 2022-06-16T14:51:09.151.271.637 (XO: 12.151273837)
   Max TM value:223 at an offset of 173.951 usec -> 2022-06-16T14:51:09.151.314.675 (XO: 12.151316874)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 145.888 usec -> 2022-06-16T14:51:09.151.286.612 (XO: 12.151288812)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 176.276 usec -> 2022-06-16T14:51:09.151.317.000 (XO: 12.151319199)

 - Block number: 2 starting at 2022-06-16T14:51:12.011.086.725 (XO: 15.011088362 - 15.011394363, t=43.6 degC)  Rank: 00086
   TM values: Max =   237  Min =  -257  Ave =    -4.53  Std =    28.52
   Min TM value:-257 at an offset of 167.663 usec -> 2022-06-16T14:51:12.011.254.388 (XO: 15.011256025)
   Max TM value:237 at an offset of 282.401 usec -> 2022-06-16T14:51:12.011.369.125 (XO: 15.011370763)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 282.276 usec -> 2022-06-16T14:51:12.011.369.000 (XO: 15.011370638)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 167.538 usec -> 2022-06-16T14:51:12.011.254.263 (XO: 15.011255900)
