-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 14:55:43 2023 UTC
-- File 20220616_004552_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.688346 to 62.118874
-each separately from 9s to 62s and from 28416us to 988785us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.219785 in 1 packets: 18
      12.119707 in 1 packets: 36
      20.959791 in 1 packets: 54
      11.129800 in 1 packets: 72
MIU packet time from 2022-06-16T00:46:04.688.346.000 to 2022-06-16T00:46:57.118.874.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.219785 in 1 packets: 18
      12.119707 in 1 packets: 36
      20.959791 in 1 packets: 54
      11.129800 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.5133621 to 63.5131873
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999736 in 1 packets: 18
      12.9999571 in 1 packets: 36
      19.9999340 in 1 packets: 54
      11.9999605 in 1 packets: 72
Temperature FPGA: 
      FE -> 57.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      00 -> 55.0 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C5 -> 43.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220616_004552_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.2017095 to 11.2019985
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2197590  -- OK for SL --
   MIU time from 2022-06-16T00:46:04.688.346.000 to 2022-06-16T00:46:04.688.635.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.219785

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4217575 to 19.4220465
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1196630  -- OK for SL --
   MIU time from 2022-06-16T00:46:12.908.420.000 to 2022-06-16T00:46:12.908.709.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.119707

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5417095 to 31.5419985
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9597270  -- OK for SL --
   MIU time from 2022-06-16T00:46:25.028.416.000 to 2022-06-16T00:46:25.028.705.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.959791

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.5017255 to 52.5020145
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1297590  -- OK for SL --
   MIU time from 2022-06-16T00:46:45.988.496.000 to 2022-06-16T00:46:45.988.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.129800

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.6317735 to 63.6320625
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-16T00:46:57.118.585.000 to 2022-06-16T00:46:57.118.874.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.201709500 corrected by 40bit 10MHz counter of 10.513362100 obtained at 1Hz counter of 9. Result: 9.688349672
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2022-06-16T00:46:04.688.205.725
   Max 1st sample MIU time among all SL packets: 2022-06-16T00:46:57.118.733.725
   Min 1st sample XO time among all SL packets: 9.688209396
   Max 1st sample XO time among all SL packets: 62.118735317
   TM values from -274 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-16T00:46:04.688.205.725 (XO: 9.688209396 - 9.688515397, t=43.6 degC)  Rank: 00081
   TM values: Max =   210  Min =  -229  Ave =    -4.45  Std =    25.80
   Min TM value:-229 at an offset of 123.613 usec -> 2022-06-16T00:46:04.688.329.337 (XO: 9.688333009)
   Max TM value:210 at an offset of 178.051 usec -> 2022-06-16T00:46:04.688.383.775 (XO: 9.688387447)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 123.738 usec -> 2022-06-16T00:46:04.688.329.462 (XO: 9.688333134)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 123.538 usec -> 2022-06-16T00:46:04.688.329.262 (XO: 9.688332934)

 - Block number: 2 starting at 2022-06-16T00:46:12.908.279.725 (XO: 17.908284432 - 17.908590432, t=43.6 degC)  Rank: 00079
   TM values: Max =   237  Min =  -256  Ave =    -4.36  Std =    25.60
   Min TM value:-256 at an offset of 115.800 usec -> 2022-06-16T00:46:12.908.395.525 (XO: 17.908400232)
   Max TM value:237 at an offset of 118.163 usec -> 2022-06-16T00:46:12.908.397.887 (XO: 17.908402594)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 118.038 usec -> 2022-06-16T00:46:12.908.397.762 (XO: 17.908402469)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 91.500 usec -> 2022-06-16T00:46:12.908.371.225 (XO: 17.908375932)

 - Block number: 3 starting at 2022-06-16T00:46:25.028.275.725 (XO: 30.028276518 - 30.028582519, t=43.6 degC)  Rank: 00077
   TM values: Max =   241  Min =  -274  Ave =    -4.48  Std =    28.40
   Min TM value:-274 at an offset of 139.575 usec -> 2022-06-16T00:46:25.028.415.300 (XO: 30.028416094)
   Max TM value:241 at an offset of 182.888 usec -> 2022-06-16T00:46:25.028.458.613 (XO: 30.028459406)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 154.588 usec -> 2022-06-16T00:46:25.028.430.313 (XO: 30.028431106)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 95.588 usec -> 2022-06-16T00:46:25.028.371.312 (XO: 30.028372106)

 - Block number: 4 starting at 2022-06-16T00:46:45.988.355.725 (XO: 50.988361687 - 50.988667688, t=43.6 degC)  Rank: 00082
   TM values: Max =   224  Min =  -268  Ave =    -4.47  Std =    27.81
   Min TM value:-268 at an offset of 109.463 usec -> 2022-06-16T00:46:45.988.465.187 (XO: 50.988471149)
   Max TM value:224 at an offset of 111.800 usec -> 2022-06-16T00:46:45.988.467.525 (XO: 50.988473487)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 111.675 usec -> 2022-06-16T00:46:45.988.467.400 (XO: 50.988473362)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 109.300 usec -> 2022-06-16T00:46:45.988.465.025 (XO: 50.988470987)

 - Block number: 5 starting at 2022-06-16T00:46:57.118.444.725 (XO: 62.118446316 - 62.118752317, t=43.6 degC)  Rank: 00076
   TM values: Max =   243  Min =  -244  Ave =    -4.52  Std =    29.21
   Min TM value:-244 at an offset of 109.663 usec -> 2022-06-16T00:46:57.118.554.387 (XO: 62.118555979)
   Max TM value:243 at an offset of 112.013 usec -> 2022-06-16T00:46:57.118.556.737 (XO: 62.118558329)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 170.438 usec -> 2022-06-16T00:46:57.118.615.163 (XO: 62.118616754)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 109.538 usec -> 2022-06-16T00:46:57.118.554.262 (XO: 62.118555854)
