-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 11:52:20 2023 UTC
-- File 20220615_151422_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177025
-each separately from 0s to 14s and from 0us to 841251us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU packet time from 2022-06-15T15:14:25.000.000.000 to 2022-06-15T15:14:39.177.025.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.6165427 to 15.6164967
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999968 in 4 packets: 2, 4, 11, 14
      0.9999966 in 2 packets: 3, 12
      0.9999967 in 8 packets: 5, 6, 8, 9, 10, 15, 16, 17
   XO PPM: 
      -3.2 in 4 packets: 2, 4, 11, 14
      -3.4 in 2 packets: 3, 12
      -3.3 in 8 packets: 5, 6, 8, 9, 10, 15, 16, 17
Temperature FPGA: 
      00 -> 55.0 degC in 3 packets: 1, 9, 11
      FF -> 56.3 degC in 10 packets: 2, 7, 8, 12, 13, 14, 15, 16, 17, 18
      FE -> 57.6 degC in 5 packets: 3, 4, 5, 6, 10
Temperature XO: 
      C5 -> 43.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 9 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9
      1 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 9 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9
      1 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220615_151422_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.6165428 to 15.7935228
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-15T15:14:25.000.000.000 to 2022-06-15T15:14:39.177.025.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.616542800 corrected by 40bit 10MHz counter of 1.616542700 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SR packets: 2022-06-15T15:14:24.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-06-15T15:14:39.177.024.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177026609
   TM values from -23 to 15 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-06-15T15:14:24.999.999.925 (XO: 0.000000025 - 0.000017025, t=43.6 degC)
   TM values: Max =    11  Min =   -16  Ave =    -4.39  Std =     4.44

 - Packet number: 2 starting at 2022-06-15T15:14:25.833.937.925 (XO: 0.833942694 - 0.833959694, t=43.6 degC)
   TM values: Max =     9  Min =   -15  Ave =    -4.34  Std =     4.47

 - Packet number: 3 starting at 2022-06-15T15:14:26.667.880.925 (XO: 1.667885496 - 1.667902496, t=43.6 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.44  Std =     5.07

 - Packet number: 4 starting at 2022-06-15T15:14:27.501.824.925 (XO: 2.501828231 - 2.501845231, t=43.6 degC)
   TM values: Max =    15  Min =   -23  Ave =    -4.65  Std =     7.99

 - Packet number: 5 starting at 2022-06-15T15:14:28.335.767.925 (XO: 3.335770933 - 3.335787933, t=43.6 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.14  Std =     6.01

 - Packet number: 6 starting at 2022-06-15T15:14:29.169.711.925 (XO: 4.169713685 - 4.169730685, t=43.6 degC)
   TM values: Max =    10  Min =   -15  Ave =    -4.44  Std =     4.25

 - Packet number: 7 starting at 2022-06-15T15:14:30.003.654.925 (XO: 5.003656437 - 5.003673437, t=43.6 degC)
   TM values: Max =    10  Min =   -20  Ave =    -4.39  Std =     5.66

 - Packet number: 8 starting at 2022-06-15T15:14:30.837.593.925 (XO: 5.837599189 - 5.837616189, t=43.6 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.18  Std =     5.40

 - Packet number: 9 starting at 2022-06-15T15:14:31.671.537.925 (XO: 6.671541941 - 6.671558941, t=43.6 degC)
   TM values: Max =     9  Min =   -19  Ave =    -4.59  Std =     5.14

 - Packet number: 10 starting at 2022-06-15T15:14:32.505.480.925 (XO: 7.505484693 - 7.505501693, t=43.6 degC)
   TM values: Max =    13  Min =   -19  Ave =    -4.15  Std =     6.06

 - Packet number: 11 starting at 2022-06-15T15:14:33.339.424.925 (XO: 8.339427411 - 8.339444411, t=43.6 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.43  Std =     5.05

 - Packet number: 12 starting at 2022-06-15T15:14:34.173.367.925 (XO: 9.173370114 - 9.173387115, t=43.6 degC)
   TM values: Max =     8  Min =   -17  Ave =    -4.61  Std =     5.00

 - Packet number: 13 starting at 2022-06-15T15:14:35.007.311.925 (XO: 10.007312948 - 10.007329948, t=43.6 degC)
   TM values: Max =     8  Min =   -16  Ave =    -4.40  Std =     4.31

 - Packet number: 14 starting at 2022-06-15T15:14:35.841.250.925 (XO: 10.841255617 - 10.841272617, t=43.6 degC)
   TM values: Max =    11  Min =   -20  Ave =    -4.29  Std =     5.47

 - Packet number: 15 starting at 2022-06-15T15:14:36.675.193.925 (XO: 11.675198353 - 11.675215353, t=43.6 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.56  Std =     5.69

 - Packet number: 16 starting at 2022-06-15T15:14:37.509.137.925 (XO: 12.509141105 - 12.509158105, t=43.6 degC)
   TM values: Max =    10  Min =   -20  Ave =    -4.16  Std =     5.29

 - Packet number: 17 starting at 2022-06-15T15:14:38.343.080.925 (XO: 13.343083857 - 13.343100857, t=43.6 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.55  Std =     5.54

 - Packet number: 18 starting at 2022-06-15T15:14:39.177.024.925 (XO: 14.177026609 - 14.177043609, t=43.6 degC)
   TM values: Max =     8  Min =   -21  Ave =    -4.37  Std =     4.46
