-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 06:51:36 2023 UTC
-- File 20220614_152616_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177030
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 11 packets: 2, 3, 4, 5, 8, 10, 11, 12, 14, 15, 17
      0.833943 in 3 packets: 6, 9, 16
      0.833940 in 1 packets: 7
      0.833939 in 1 packets: 13
MIU packet time from 2022-06-14T15:26:19.000.000.000 to 2022-06-14T15:26:33.177.030.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 11 packets: 2, 3, 4, 5, 8, 10, 11, 12, 14, 15, 17
      0.833943 in 3 packets: 6, 9, 16
      0.833940 in 1 packets: 7
      0.833939 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8848067 to 15.8847555
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999963 in 8 packets: 2, 3, 4, 6, 9, 11, 14, 15
      0.9999964 in 6 packets: 5, 8, 10, 12, 16, 17
   XO PPM: 
      -3.7 in 8 packets: 2, 3, 4, 6, 9, 11, 14, 15
      -3.6 in 6 packets: 5, 8, 10, 12, 16, 17
Temperature FPGA: 
      00 -> 55.0 degC in 14 packets: 1, 3, 4, 6, 7, 8, 9, 11, 12, 13, 14, 15, 17, 18
      01 -> 53.7 degC in 1 packets: 2
      FF -> 56.3 degC in 3 packets: 5, 10, 16
Temperature XO: 
      C5 -> 43.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 3 packets: 1, 2, 3
      1 in 6 packets: 4, 5, 6, 7, 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 3 packets: 1, 2, 3
      1 in 6 packets: 4, 5, 6, 7, 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220614_152616_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8848068 to 16.0617868
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-14T15:26:19.000.000.000 to 2022-06-14T15:26:33.177.030.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 11 packets: 2, 3, 4, 5, 8, 10, 11, 12, 14, 15, 17
      0.833943 in 3 packets: 6, 9, 16
      0.833940 in 1 packets: 7
      0.833939 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.884806800 corrected by 40bit 10MHz counter of 1.884806700 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2022-06-14T15:26:18.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-06-14T15:26:33.177.029.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031862
   TM values from -196 to 200 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-06-14T15:26:18.999.999.925 (XO: 0.000000025 - 0.000017025, t=43.6 degC)
   TM values: Max =     8  Min =   -17  Ave =    -4.61  Std =     4.68

 - Packet number: 2 starting at 2022-06-14T15:26:19.833.937.925 (XO: 0.833943111 - 0.833960111, t=43.6 degC)
   TM values: Max =     9  Min =   -19  Ave =    -4.17  Std =     5.60

 - Packet number: 3 starting at 2022-06-14T15:26:20.667.881.925 (XO: 1.667886196 - 1.667903196, t=43.6 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.47  Std =     5.17

 - Packet number: 4 starting at 2022-06-14T15:26:21.501.825.925 (XO: 2.501829282 - 2.501846282, t=43.6 degC)
   TM values: Max =    10  Min =   -21  Ave =    -4.29  Std =     5.21

 - Packet number: 5 starting at 2022-06-14T15:26:22.335.769.925 (XO: 3.335772334 - 3.335789334, t=43.6 degC)
   TM values: Max =   200  Min =  -196  Ave =    -4.70  Std =    52.49

 - Packet number: 6 starting at 2022-06-14T15:26:23.169.713.925 (XO: 4.169715353 - 4.169732353, t=43.6 degC)
   TM values: Max =     9  Min =   -19  Ave =    -4.55  Std =     5.64

 - Packet number: 7 starting at 2022-06-14T15:26:24.003.656.925 (XO: 5.003658438 - 5.003675438, t=43.6 degC)
   TM values: Max =    13  Min =   -18  Ave =    -4.35  Std =     5.62

 - Packet number: 8 starting at 2022-06-14T15:26:24.837.596.925 (XO: 5.837601440 - 5.837618440, t=43.6 degC)
   TM values: Max =    10  Min =   -19  Ave =    -4.38  Std =     5.28

 - Packet number: 9 starting at 2022-06-14T15:26:25.671.540.925 (XO: 6.671544510 - 6.671561510, t=43.6 degC)
   TM values: Max =    12  Min =   -24  Ave =    -4.39  Std =     7.54

 - Packet number: 10 starting at 2022-06-14T15:26:26.505.483.925 (XO: 7.505487545 - 7.505504545, t=43.6 degC)
   TM values: Max =     8  Min =   -15  Ave =    -4.46  Std =     3.14

 - Packet number: 11 starting at 2022-06-14T15:26:27.339.427.925 (XO: 8.339430581 - 8.339447581, t=43.6 degC)
   TM values: Max =    10  Min =   -21  Ave =    -4.44  Std =     5.16

 - Packet number: 12 starting at 2022-06-14T15:26:28.173.371.925 (XO: 9.173373649 - 9.173390649, t=43.6 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.42  Std =     5.87

 - Packet number: 13 starting at 2022-06-14T15:26:29.007.315.925 (XO: 10.007316652 - 10.007333652, t=43.6 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.49  Std =     7.03

 - Packet number: 14 starting at 2022-06-14T15:26:29.841.254.925 (XO: 10.841259738 - 10.841276738, t=43.6 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.34  Std =     6.04

 - Packet number: 15 starting at 2022-06-14T15:26:30.675.198.925 (XO: 11.675202823 - 11.675219823, t=43.6 degC)
   TM values: Max =    12  Min =   -23  Ave =    -4.43  Std =     6.14

 - Packet number: 16 starting at 2022-06-14T15:26:31.509.142.925 (XO: 12.509145858 - 12.509162858, t=43.6 degC)
   TM values: Max =    12  Min =   -24  Ave =    -4.56  Std =     6.63

 - Packet number: 17 starting at 2022-06-14T15:26:32.343.085.925 (XO: 13.343088860 - 13.343105860, t=43.6 degC)
   TM values: Max =    10  Min =   -21  Ave =    -4.44  Std =     5.83

 - Packet number: 18 starting at 2022-06-14T15:26:33.177.029.925 (XO: 14.177031862 - 14.177048862, t=43.6 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.62  Std =     7.43
