-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 06:03:57 2023 UTC
-- File 20220614_142536_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.681249 to 50.773034
-each separately from 5s to 50s and from 121729us to 773034us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.440191 in 1 packets: 18
      18.450107 in 1 packets: 36
      13.200331 in 1 packets: 54
MIU packet time from 2022-06-14T14:25:44.681.249.000 to 2022-06-14T14:26:29.773.034.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.440191 in 1 packets: 18
      18.450107 in 1 packets: 36
      13.200331 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.2502006 to 52.2500485
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999527 in 1 packets: 18
      17.9999391 in 1 packets: 36
      12.9999561 in 1 packets: 54
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      FE -> 57.6 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C5 -> 43.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220614_142536_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.9314513 to 7.9317403
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4401430  -- OK for SL --
   MIU time from 2022-06-14T14:25:44.681.249.000 to 2022-06-14T14:25:44.681.538.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.440191

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3718833 to 21.3721723
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4500470  -- OK for SL --
   MIU time from 2022-06-14T14:25:58.121.729.000 to 2022-06-14T14:25:58.122.018.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.450107

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.8222193 to 39.8225083
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2002870  -- OK for SL --
   MIU time from 2022-06-14T14:26:16.572.125.000 to 2022-06-14T14:26:16.572.414.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.200331

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.0227953 to 53.0230843
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-14T14:26:29.772.745.000 to 2022-06-14T14:26:29.773.034.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.931451300 corrected by 40bit 10MHz counter of 7.250200600 obtained at 1Hz counter of 5. Result: 5.681252948
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-14T14:25:44.681.108.725
   Max 1st sample MIU time among all SL packets: 2022-06-14T14:26:29.772.893.725
   Min 1st sample XO time among all SL packets: 5.681112673
   Max 1st sample XO time among all SL packets: 50.772898153
   TM values from -302 to 238 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-14T14:25:44.681.108.725 (XO: 5.681112673 - 5.681418674, t=43.6 degC)  Rank: 00078
   TM values: Max =   215  Min =  -240  Ave =    -4.41  Std =    23.95
   Min TM value:-240 at an offset of 156.501 usec -> 2022-06-14T14:25:44.681.265.225 (XO: 5.681269173)
   Max TM value:215 at an offset of 156.238 usec -> 2022-06-14T14:25:44.681.264.963 (XO: 5.681268911)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 156.176 usec -> 2022-06-14T14:25:44.681.264.900 (XO: 5.681268848)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 156.338 usec -> 2022-06-14T14:25:44.681.265.063 (XO: 5.681269011)

 - Block number: 2 starting at 2022-06-14T14:25:58.121.588.725 (XO: 19.121590138 - 19.121896139, t=43.6 degC)  Rank: 00080
   TM values: Max =   226  Min =  -275  Ave =    -4.47  Std =    31.70
   Min TM value:-275 at an offset of 119.650 usec -> 2022-06-14T14:25:58.121.708.375 (XO: 19.121709789)
   Max TM value:226 at an offset of 117.650 usec -> 2022-06-14T14:25:58.121.706.375 (XO: 19.121707789)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 117.525 usec -> 2022-06-14T14:25:58.121.706.250 (XO: 19.121707664)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 119.538 usec -> 2022-06-14T14:25:58.121.708.262 (XO: 19.121709676)

 - Block number: 3 starting at 2022-06-14T14:26:16.571.984.725 (XO: 37.571988570 - 37.572294571, t=43.6 degC)  Rank: 00077
   TM values: Max =   224  Min =  -302  Ave =    -4.58  Std =    33.24
   Min TM value:-302 at an offset of 208.401 usec -> 2022-06-14T14:26:16.572.193.125 (XO: 37.572196970)
   Max TM value:224 at an offset of 206.401 usec -> 2022-06-14T14:26:16.572.191.125 (XO: 37.572194970)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 206.251 usec -> 2022-06-14T14:26:16.572.190.975 (XO: 37.572194820)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 118.338 usec -> 2022-06-14T14:26:16.572.103.062 (XO: 37.572106908)

 - Block number: 4 starting at 2022-06-14T14:26:29.772.604.725 (XO: 50.772609152 - 50.772915153, t=43.6 degC)  Rank: 00084
   TM values: Max =   238  Min =  -256  Ave =    -4.43  Std =    27.29
   Min TM value:-256 at an offset of 91.888 usec -> 2022-06-14T14:26:29.772.696.612 (XO: 50.772701040)
   Max TM value:238 at an offset of 148.988 usec -> 2022-06-14T14:26:29.772.753.713 (XO: 50.772758140)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 148.851 usec -> 2022-06-14T14:26:29.772.753.575 (XO: 50.772758002)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 91.650 usec -> 2022-06-14T14:26:29.772.696.375 (XO: 50.772700802)
