-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 05:54:20 2023 UTC
-- File 20220614_141238_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.248535 to 61.889727
-each separately from 12s to 61s and from 248535us to 889727us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.360073 in 1 packets: 18
      4.099788 in 1 packets: 36
      16.539898 in 1 packets: 54
      11.639988 in 1 packets: 72
MIU packet time from 2022-06-14T14:12:53.248.535.000 to 2022-06-14T14:13:42.889.727.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.360073 in 1 packets: 18
      4.099788 in 1 packets: 36
      16.539898 in 1 packets: 54
      11.639988 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7619280 to 62.7617601
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999418 in 1 packets: 18
      3.9999863 in 1 packets: 36
      16.9999417 in 1 packets: 54
      10.9999623 in 1 packets: 72
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      FE -> 57.6 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C5 -> 43.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220614_141238_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.0104642 to 14.0107532
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3600150  -- OK for SL --
   MIU time from 2022-06-14T14:12:53.248.535.000 to 2022-06-14T14:12:53.248.824.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.360073

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.3707682 to 31.3710572
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0997750  -- OK for SL --
   MIU time from 2022-06-14T14:13:10.608.897.000 to 2022-06-14T14:13:10.609.186.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.099788

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4708322 to 35.4711212
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5398390  -- OK for SL --
   MIU time from 2022-06-14T14:13:14.708.974.000 to 2022-06-14T14:13:14.709.263.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.539898

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.0109602 to 52.0112492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6399510  -- OK for SL --
   MIU time from 2022-06-14T14:13:31.249.161.000 to 2022-06-14T14:13:31.249.450.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.639988

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.6512002 to 63.6514892
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-14T14:13:42.889.438.000 to 2022-06-14T14:13:42.889.727.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.010464200 corrected by 40bit 10MHz counter of 13.761928000 obtained at 1Hz counter of 12. Result: 12.248537045
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-14T14:12:53.248.394.725
   Max 1st sample MIU time among all SL packets: 2022-06-14T14:13:42.889.586.725
   Min 1st sample XO time among all SL packets: 12.248396770
   Max 1st sample XO time among all SL packets: 61.889591939
   TM values from -275 to 232 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-14T14:12:53.248.394.725 (XO: 12.248396770 - 12.248702771, t=43.6 degC)  Rank: 00080
   TM values: Max =   232  Min =  -270  Ave =    -4.37  Std =    29.98
   Min TM value:-270 at an offset of 126.475 usec -> 2022-06-14T14:12:53.248.521.200 (XO: 12.248523245)
   Max TM value:232 at an offset of 157.226 usec -> 2022-06-14T14:12:53.248.551.950 (XO: 12.248553995)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 128.625 usec -> 2022-06-14T14:12:53.248.523.350 (XO: 12.248525395)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 86.588 usec -> 2022-06-14T14:12:53.248.481.312 (XO: 12.248483357)

 - Block number: 2 starting at 2022-06-14T14:13:10.608.756.725 (XO: 29.608760195 - 29.609066196, t=43.6 degC)  Rank: 00076
   TM values: Max =   197  Min =  -207  Ave =    -4.47  Std =    24.58
   Min TM value:-207 at an offset of 112.550 usec -> 2022-06-14T14:13:10.608.869.275 (XO: 29.608872745)
   Max TM value:197 at an offset of 115.075 usec -> 2022-06-14T14:13:10.608.871.800 (XO: 29.608875270)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 112.675 usec -> 2022-06-14T14:13:10.608.869.400 (XO: 29.608872870)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 115.188 usec -> 2022-06-14T14:13:10.608.871.912 (XO: 29.608875383)

 - Block number: 3 starting at 2022-06-14T14:13:14.708.833.725 (XO: 33.708838306 - 33.709144307, t=43.6 degC)  Rank: 00075
   TM values: Max =   229  Min =  -271  Ave =    -4.43  Std =    30.12
   Min TM value:-271 at an offset of 90.925 usec -> 2022-06-14T14:13:14.708.924.650 (XO: 33.708929231)
   Max TM value:229 at an offset of 90.688 usec -> 2022-06-14T14:13:14.708.924.412 (XO: 33.708928994)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 47.700 usec -> 2022-06-14T14:13:14.708.881.425 (XO: 33.708886006)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 90.825 usec -> 2022-06-14T14:13:14.708.924.550 (XO: 33.708929131)

 - Block number: 4 starting at 2022-06-14T14:13:31.249.020.725 (XO: 50.249022997 - 50.249328998, t=43.6 degC)  Rank: 00082
   TM values: Max =   227  Min =  -275  Ave =    -4.53  Std =    31.56
   Min TM value:-275 at an offset of 122.575 usec -> 2022-06-14T14:13:31.249.143.300 (XO: 50.249145572)
   Max TM value:227 at an offset of 155.063 usec -> 2022-06-14T14:13:31.249.175.788 (XO: 50.249178060)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 154.901 usec -> 2022-06-14T14:13:31.249.175.625 (XO: 50.249177897)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 122.438 usec -> 2022-06-14T14:13:31.249.143.162 (XO: 50.249145435)

 - Block number: 5 starting at 2022-06-14T14:13:42.889.297.725 (XO: 61.889302938 - 61.889608939, t=43.6 degC)  Rank: 00077
   TM values: Max =   223  Min =  -262  Ave =    -4.41  Std =    28.78
   Min TM value:-262 at an offset of 127.313 usec -> 2022-06-14T14:13:42.889.425.037 (XO: 61.889430251)
   Max TM value:223 at an offset of 129.650 usec -> 2022-06-14T14:13:42.889.427.375 (XO: 61.889432588)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 58.950 usec -> 2022-06-14T14:13:42.889.356.675 (XO: 61.889361888)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 56.550 usec -> 2022-06-14T14:13:42.889.354.275 (XO: 61.889359488)
