-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 05:42:14 2023 UTC
-- File 20220614_134919_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841252us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 6, 8, 10, 11, 14, 16, 17
      0.833943 in 5 packets: 3, 5, 9, 12, 15
      0.833939 in 2 packets: 7, 13
MIU packet time from 2022-06-14T13:49:22.000.000.000 to 2022-06-14T13:49:36.177.027.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 6, 8, 10, 11, 14, 16, 17
      0.833943 in 5 packets: 3, 5, 9, 12, 15
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2484007 to 16.2483529
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999966 in 12 packets: 2, 3, 4, 5, 8, 9, 10, 11, 12, 15, 16, 17
      0.9999965 in 2 packets: 6, 14
   XO PPM: 
      -3.4 in 12 packets: 2, 3, 4, 5, 8, 9, 10, 11, 12, 15, 16, 17
      -3.5 in 2 packets: 6, 14
Temperature FPGA: 
      01 -> 53.7 degC in 4 packets: 1, 7, 8, 11
      00 -> 55.0 degC in 7 packets: 2, 3, 5, 10, 15, 17, 18
      FF -> 56.3 degC in 7 packets: 4, 6, 9, 12, 13, 14, 16
Temperature XO: 
      C5 -> 43.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 15 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15
      1 in 3 packets: 16, 17, 18
Number of generated alerts: 
      0 in 15 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15
      1 in 3 packets: 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220614_134919_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2484008 to 16.4253808
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-14T13:49:22.000.000.000 to 2022-06-14T13:49:36.177.027.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 6, 8, 10, 11, 14, 16, 17
      0.833943 in 5 packets: 3, 5, 9, 12, 15
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.248400800 corrected by 40bit 10MHz counter of 2.248400700 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SR packets: 2022-06-14T13:49:21.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-06-14T13:49:36.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028427
   TM values from -20 to 12 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-06-14T13:49:21.999.999.925 (XO: 0.000000025 - 0.000017025, t=43.6 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.13  Std =     6.01

 - Packet number: 2 starting at 2022-06-14T13:49:22.833.937.925 (XO: 0.833942860 - 0.833959860, t=43.6 degC)
   TM values: Max =     5  Min =   -14  Ave =    -4.51  Std =     3.57

 - Packet number: 3 starting at 2022-06-14T13:49:23.667.881.925 (XO: 1.667885696 - 1.667902696, t=43.6 degC)
   TM values: Max =    11  Min =   -20  Ave =    -4.32  Std =     7.48

 - Packet number: 4 starting at 2022-06-14T13:49:24.501.824.925 (XO: 2.501828531 - 2.501845531, t=43.6 degC)
   TM values: Max =     9  Min =   -15  Ave =    -4.39  Std =     4.47

 - Packet number: 5 starting at 2022-06-14T13:49:25.335.768.925 (XO: 3.335771367 - 3.335788367, t=43.6 degC)
   TM values: Max =     4  Min =   -16  Ave =    -4.27  Std =     3.24

 - Packet number: 6 starting at 2022-06-14T13:49:26.169.711.925 (XO: 4.169714219 - 4.169731219, t=43.6 degC)
   TM values: Max =     7  Min =   -17  Ave =    -4.54  Std =     4.82

 - Packet number: 7 starting at 2022-06-14T13:49:27.003.655.925 (XO: 5.003657137 - 5.003674137, t=43.6 degC)
   TM values: Max =     8  Min =   -17  Ave =    -4.27  Std =     4.63

 - Packet number: 8 starting at 2022-06-14T13:49:27.837.594.925 (XO: 5.837599973 - 5.837616973, t=43.6 degC)
   TM values: Max =    12  Min =   -19  Ave =    -4.37  Std =     5.83

 - Packet number: 9 starting at 2022-06-14T13:49:28.671.538.925 (XO: 6.671542808 - 6.671559808, t=43.6 degC)
   TM values: Max =     7  Min =   -16  Ave =    -4.53  Std =     4.93

 - Packet number: 10 starting at 2022-06-14T13:49:29.505.481.925 (XO: 7.505485644 - 7.505502644, t=43.6 degC)
   TM values: Max =     8  Min =   -20  Ave =    -4.40  Std =     4.77

 - Packet number: 11 starting at 2022-06-14T13:49:30.339.425.925 (XO: 8.339428479 - 8.339445479, t=43.6 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.32  Std =     5.02

 - Packet number: 12 starting at 2022-06-14T13:49:31.173.369.925 (XO: 9.173371314 - 9.173388315, t=43.6 degC)
   TM values: Max =     9  Min =   -16  Ave =    -4.09  Std =     5.21

 - Packet number: 13 starting at 2022-06-14T13:49:32.007.312.925 (XO: 10.007314151 - 10.007331151, t=43.6 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.41  Std =     5.27

 - Packet number: 14 starting at 2022-06-14T13:49:32.841.251.925 (XO: 10.841257069 - 10.841274069, t=43.6 degC)
   TM values: Max =     7  Min =   -18  Ave =    -4.19  Std =     5.30

 - Packet number: 15 starting at 2022-06-14T13:49:33.675.195.925 (XO: 11.675199921 - 11.675216921, t=43.6 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.53  Std =     6.46

 - Packet number: 16 starting at 2022-06-14T13:49:34.509.138.925 (XO: 12.509142756 - 12.509159756, t=43.6 degC)
   TM values: Max =    11  Min =   -20  Ave =    -4.12  Std =     7.17

 - Packet number: 17 starting at 2022-06-14T13:49:35.343.082.925 (XO: 13.343085591 - 13.343102592, t=43.6 degC)
   TM values: Max =     4  Min =   -12  Ave =    -4.57  Std =     2.94

 - Packet number: 18 starting at 2022-06-14T13:49:36.177.026.925 (XO: 14.177028427 - 14.177045427, t=43.6 degC)
   TM values: Max =     8  Min =   -18  Ave =    -4.31  Std =     5.29
