-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 04:16:29 2023 UTC
-- File 20220614_112523_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.923064 to 16.865611
-each separately from 1s to 16s and from 865322us to 923353us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.941969 in 1 packets: 18
MIU packet time from 2022-06-14T11:25:27.923.064.000 to 2022-06-14T11:25:42.865.611.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.941969 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.2506342 to 18.2505841
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999499 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220614_112523_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1737001 to 4.1739891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9419190  -- OK for SL --
   MIU time from 2022-06-14T11:25:27.923.064.000 to 2022-06-14T11:25:27.923.353.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.941969

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1159081 to 19.1161971
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-14T11:25:42.865.322.000 to 2022-06-14T11:25:42.865.611.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 4.173700100 corrected by 40bit 10MHz counter of 3.250634200 obtained at 1Hz counter of 1. Result: 1.923069038
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-14T11:25:27.922.923.725
   Max 1st sample MIU time among all SL packets: 2022-06-14T11:25:42.865.470.725
   Min 1st sample XO time among all SL packets: 1.922928763
   Max 1st sample XO time among all SL packets: 16.865475581
   TM values from -286 to 214 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-14T11:25:27.922.923.725 (XO: 1.922928763 - 1.923234764, t=43.6 degC)  Rank: 00079
   TM values: Max =   214  Min =  -286  Ave =    -4.56  Std =    32.53
   Min TM value:-286 at an offset of 126.288 usec -> 2022-06-14T11:25:27.923.050.012 (XO: 1.923055051)
   Max TM value:214 at an offset of 128.600 usec -> 2022-06-14T11:25:27.923.052.325 (XO: 1.923057363)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 128.438 usec -> 2022-06-14T11:25:27.923.052.162 (XO: 1.923057201)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 126.100 usec -> 2022-06-14T11:25:27.923.049.825 (XO: 1.923054863)

 - Block number: 2 starting at 2022-06-14T11:25:42.865.181.725 (XO: 16.865186580 - 16.865492581, t=43.6 degC)  Rank: 00082
   TM values: Max =   207  Min =  -252  Ave =    -4.44  Std =    27.58
   Min TM value:-252 at an offset of 184.476 usec -> 2022-06-14T11:25:42.865.366.200 (XO: 16.865371056)
   Max TM value:207 at an offset of 184.176 usec -> 2022-06-14T11:25:42.865.365.900 (XO: 16.865370756)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 186.663 usec -> 2022-06-14T11:25:42.865.368.388 (XO: 16.865373243)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 184.276 usec -> 2022-06-14T11:25:42.865.366.000 (XO: 16.865370856)
