-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 02:17:12 2023 UTC
-- File 20220613_152646_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.217668 to 101.777478
-each separately from 1s to 101s and from 161220us to 973295us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.943263 in 1 packets: 18
      12.622119 in 1 packets: 36
      12.662060 in 1 packets: 54
      26.385062 in 1 packets: 72
      9.141389 in 1 packets: 90
      20.803894 in 1 packets: 108
MIU packet time from 2022-06-13T15:26:50.217.668.000 to 2022-06-13T15:28:30.777.478.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.943263 in 1 packets: 18
      12.622119 in 1 packets: 36
      12.662060 in 1 packets: 54
      26.385062 in 1 packets: 72
      9.141389 in 1 packets: 90
      20.803894 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.5713061 to 102.5709687
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999358 in 1 packets: 18
      11.9999595 in 1 packets: 36
      12.9999562 in 1 packets: 54
      25.9999122 in 1 packets: 72
      8.9999696 in 1 packets: 90
      20.9999293 in 1 packets: 108
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 108 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C5 -> 43.6 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 53 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 53 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220613_152646_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7889757 to 2.7892647
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9431990  -- OK for SL --
   MIU time from 2022-06-13T15:26:50.217.668.000 to 2022-06-13T15:26:50.217.957.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.943263

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7324637 to 21.7327527
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6220790  -- OK for SL --
   MIU time from 2022-06-13T15:27:09.161.220.000 to 2022-06-13T15:27:09.161.509.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.622119

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.3548317 to 34.3551207
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6620150  -- OK for SL --
   MIU time from 2022-06-13T15:27:21.783.628.000 to 2022-06-13T15:27:21.783.917.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.662060

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.0171357 to 47.0174247
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.3849750  -- OK for SL --
   MIU time from 2022-06-13T15:27:34.445.977.000 to 2022-06-13T15:27:34.446.266.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.385062

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 73.4023997 to 73.4026887
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1413590  -- OK for SL --
   MIU time from 2022-06-13T15:28:00.831.328.000 to 2022-06-13T15:28:00.831.617.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.141389

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.5440477 to 82.5443367
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8038230  -- OK for SL --
   MIU time from 2022-06-13T15:28:09.973.006.000 to 2022-06-13T15:28:09.973.295.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.803894

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 103.3481597 to 103.3484487
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-13T15:28:30.777.189.000 to 2022-06-13T15:28:30.777.478.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.788975700 corrected by 40bit 10MHz counter of 2.571306100 obtained at 1Hz counter of 1. Result: 1.217670340
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-13T15:26:50.217.527.725
   Max 1st sample MIU time among all SL packets: 2022-06-13T15:28:30.777.337.725
   Min 1st sample XO time among all SL packets: 1.217530065
   Max 1st sample XO time among all SL packets: 101.777342290
   TM values from -262 to 233 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-13T15:26:50.217.527.725 (XO: 1.217530065 - 1.217836066, t=43.6 degC)  Rank: 00074
   TM values: Max =   218  Min =  -243  Ave =    -4.48  Std =    33.99
   Min TM value:-243 at an offset of 81.675 usec -> 2022-06-13T15:26:50.217.609.400 (XO: 1.217611740)
   Max TM value:218 at an offset of 210.526 usec -> 2022-06-13T15:26:50.217.738.250 (XO: 1.217740590)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 210.376 usec -> 2022-06-13T15:26:50.217.738.100 (XO: 1.217740440)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 220.926 usec -> 2022-06-13T15:26:50.217.748.650 (XO: 1.217750990)

 - Block number: 2 starting at 2022-06-13T15:27:09.161.079.725 (XO: 20.161082073 - 20.161388074, t=43.6 degC)  Rank: 00085
   TM values: Max =   213  Min =  -220  Ave =    -4.39  Std =    29.87
   Min TM value:-220 at an offset of 113.963 usec -> 2022-06-13T15:27:09.161.193.687 (XO: 20.161196036)
   Max TM value:213 at an offset of 158.813 usec -> 2022-06-13T15:27:09.161.238.538 (XO: 20.161240886)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 114.113 usec -> 2022-06-13T15:27:09.161.193.837 (XO: 20.161196186)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 227.901 usec -> 2022-06-13T15:27:09.161.307.625 (XO: 20.161309973)

 - Block number: 3 starting at 2022-06-13T15:27:21.783.487.725 (XO: 32.783492689 - 32.783798690, t=43.6 degC)  Rank: 00075
   TM values: Max =   229  Min =  -261  Ave =    -4.48  Std =    28.32
   Min TM value:-261 at an offset of 151.226 usec -> 2022-06-13T15:27:21.783.638.950 (XO: 32.783643914)
   Max TM value:229 at an offset of 153.563 usec -> 2022-06-13T15:27:21.783.641.288 (XO: 32.783646252)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 153.451 usec -> 2022-06-13T15:27:21.783.641.175 (XO: 32.783646139)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 151.076 usec -> 2022-06-13T15:27:21.783.638.800 (XO: 32.783643764)

 - Block number: 4 starting at 2022-06-13T15:27:34.445.836.725 (XO: 45.445839341 - 45.446145342, t=43.6 degC)  Rank: 00076
   TM values: Max =   229  Min =  -245  Ave =    -4.36  Std =    35.60
   Min TM value:-245 at an offset of 127.138 usec -> 2022-06-13T15:27:34.445.963.862 (XO: 45.445966479)
   Max TM value:229 at an offset of 127.400 usec -> 2022-06-13T15:27:34.445.964.125 (XO: 45.445966741)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 127.238 usec -> 2022-06-13T15:27:34.445.963.962 (XO: 45.445966579)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 124.825 usec -> 2022-06-13T15:27:34.445.961.550 (XO: 45.445964166)

 - Block number: 5 starting at 2022-06-13T15:28:00.831.187.725 (XO: 71.831192368 - 71.831498369, t=43.6 degC)  Rank: 00081
   TM values: Max =   183  Min =  -214  Ave =    -4.47  Std =    23.36
   Min TM value:-214 at an offset of 123.563 usec -> 2022-06-13T15:28:00.831.311.287 (XO: 71.831315931)
   Max TM value:183 at an offset of 121.638 usec -> 2022-06-13T15:28:00.831.309.362 (XO: 71.831314006)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 125.788 usec -> 2022-06-13T15:28:00.831.313.512 (XO: 71.831318156)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 123.300 usec -> 2022-06-13T15:28:00.831.311.025 (XO: 71.831315668)

 - Block number: 6 starting at 2022-06-13T15:28:09.972.865.725 (XO: 80.972871235 - 80.973177236, t=43.6 degC)  Rank: 00074
   TM values: Max =   233  Min =  -262  Ave =    -4.43  Std =    30.64
   Min TM value:-262 at an offset of 259.888 usec -> 2022-06-13T15:28:09.973.125.613 (XO: 80.973131124)
   Max TM value:233 at an offset of 245.626 usec -> 2022-06-13T15:28:09.973.111.350 (XO: 80.973116861)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 245.501 usec -> 2022-06-13T15:28:09.973.111.225 (XO: 80.973116736)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 124.800 usec -> 2022-06-13T15:28:09.972.990.525 (XO: 80.972996036)

 - Block number: 7 starting at 2022-06-13T15:28:30.777.048.725 (XO: 101.777053289 - 101.777359290, t=43.6 degC)  Rank: 00080
   TM values: Max =   188  Min =  -226  Ave =    -4.50  Std =    27.49
   Min TM value:-226 at an offset of 208.826 usec -> 2022-06-13T15:28:30.777.257.550 (XO: 101.777262115)
   Max TM value:188 at an offset of 127.838 usec -> 2022-06-13T15:28:30.777.176.562 (XO: 101.777181127)
   Max dTM/dt [1/us]:1.41e+03 at an offset of 206.538 usec -> 2022-06-13T15:28:30.777.255.263 (XO: 101.777259827)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 208.726 usec -> 2022-06-13T15:28:30.777.257.450 (XO: 101.777262015)
