-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar 19 01:29:25 2023 UTC
-- File 20220613_140447_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.796610 to 104.713260
-each separately from 11s to 104s and from 106694us to 885297us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.061277 in 1 packets: 18
      15.022322 in 1 packets: 36
      23.003932 in 1 packets: 54
      10.221397 in 1 packets: 72
      16.172644 in 1 packets: 90
      19.433055 in 1 packets: 108
MIU packet time from 2022-06-13T14:05:01.796.610.000 to 2022-06-13T14:06:34.713.260.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.061277 in 1 packets: 18
      15.022322 in 1 packets: 36
      23.003932 in 1 packets: 54
      10.221397 in 1 packets: 72
      16.172644 in 1 packets: 90
      19.433055 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8158084 to 105.8154957
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999698 in 1 packets: 18
      14.9999495 in 1 packets: 36
      22.9999229 in 1 packets: 54
      10.9999630 in 1 packets: 72
      15.9999461 in 1 packets: 90
      18.9999360 in 1 packets: 108
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C5 -> 43.6 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 90 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 18 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109
      3 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 90 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 18 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109
      3 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220613_140447_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.6124206 to 13.6127096
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0612470  -- OK for SL --
   MIU time from 2022-06-13T14:05:01.796.610.000 to 2022-06-13T14:05:01.796.899.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.061277

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6739566 to 22.6742456
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0222710  -- OK for SL --
   MIU time from 2022-06-13T14:05:10.858.176.000 to 2022-06-13T14:05:10.858.465.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.022322

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.6965166 to 37.6968056
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.0038550  -- OK for SL --
   MIU time from 2022-06-13T14:05:25.880.787.000 to 2022-06-13T14:05:25.881.076.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.003932

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.7006606 to 60.7009496
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.2213590  -- OK for SL --
   MIU time from 2022-06-13T14:05:48.885.008.000 to 2022-06-13T14:05:48.885.297.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.221397

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.9223086 to 70.9225976
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1725910  -- OK for SL --
   MIU time from 2022-06-13T14:05:59.106.694.000 to 2022-06-13T14:05:59.106.983.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.172644

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 87.0951886 to 87.0954776
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4329910  -- OK for SL --
   MIU time from 2022-06-13T14:06:15.279.627.000 to 2022-06-13T14:06:15.279.916.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.433055

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 106.5284686 to 106.5287576
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-13T14:06:34.712.971.000 to 2022-06-13T14:06:34.713.260.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.612420600 corrected by 40bit 10MHz counter of 12.815808400 obtained at 1Hz counter of 11. Result: 11.796614829
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-13T14:05:01.796.469.725
   Max 1st sample MIU time among all SL packets: 2022-06-13T14:06:34.713.119.725
   Min 1st sample XO time among all SL packets: 11.796474553
   Max 1st sample XO time among all SL packets: 104.713124050
   TM values from -285 to 237 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-13T14:05:01.796.469.725 (XO: 11.796474553 - 11.796780554, t=43.6 degC)  Rank: 00088
   TM values: Max =   225  Min =  -281  Ave =    -4.55  Std =    34.78
   Min TM value:-281 at an offset of 214.163 usec -> 2022-06-13T14:05:01.796.683.888 (XO: 11.796688717)
   Max TM value:225 at an offset of 128.463 usec -> 2022-06-13T14:05:01.796.598.187 (XO: 11.796603016)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 158.351 usec -> 2022-06-13T14:05:01.796.628.075 (XO: 11.796632904)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 142.850 usec -> 2022-06-13T14:05:01.796.612.575 (XO: 11.796617404)

 - Block number: 2 starting at 2022-06-13T14:05:10.858.035.725 (XO: 20.858041042 - 20.858347043, t=43.6 degC)  Rank: 00079
   TM values: Max =   237  Min =  -235  Ave =    -4.44  Std =    29.08
   Min TM value:-235 at an offset of 197.938 usec -> 2022-06-13T14:05:10.858.233.663 (XO: 20.858238981)
   Max TM value:237 at an offset of 200.326 usec -> 2022-06-13T14:05:10.858.236.050 (XO: 20.858241368)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 126.738 usec -> 2022-06-13T14:05:10.858.162.462 (XO: 20.858167780)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 129.250 usec -> 2022-06-13T14:05:10.858.164.975 (XO: 20.858170293)

 - Block number: 3 starting at 2022-06-13T14:05:25.880.646.725 (XO: 35.880651531 - 35.880957532, t=43.6 degC)  Rank: 00077
   TM values: Max =   216  Min =  -284  Ave =    -4.45  Std =    35.40
   Min TM value:-284 at an offset of 156.901 usec -> 2022-06-13T14:05:25.880.803.625 (XO: 35.880808432)
   Max TM value:216 at an offset of 159.213 usec -> 2022-06-13T14:05:25.880.805.938 (XO: 35.880810744)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 129.663 usec -> 2022-06-13T14:05:25.880.776.387 (XO: 35.880781194)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 190.451 usec -> 2022-06-13T14:05:25.880.837.175 (XO: 35.880841982)

 - Block number: 4 starting at 2022-06-13T14:05:48.884.867.725 (XO: 58.884872734 - 58.885178735, t=43.6 degC)  Rank: 00080
   TM values: Max =   222  Min =  -265  Ave =    -4.52  Std =    26.25
   Min TM value:-265 at an offset of 115.000 usec -> 2022-06-13T14:05:48.884.982.725 (XO: 58.884987734)
   Max TM value:222 at an offset of 157.701 usec -> 2022-06-13T14:05:48.885.025.425 (XO: 58.885030434)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 157.551 usec -> 2022-06-13T14:05:48.885.025.275 (XO: 58.885030284)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 114.850 usec -> 2022-06-13T14:05:48.884.982.575 (XO: 58.884987584)

 - Block number: 5 starting at 2022-06-13T14:05:59.106.553.725 (XO: 69.106555087 - 69.106861088, t=43.6 degC)  Rank: 00076
   TM values: Max =   215  Min =  -259  Ave =    -4.41  Std =    24.80
   Min TM value:-259 at an offset of 156.351 usec -> 2022-06-13T14:05:59.106.710.075 (XO: 69.106711438)
   Max TM value:215 at an offset of 156.076 usec -> 2022-06-13T14:05:59.106.709.800 (XO: 69.106711163)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 128.475 usec -> 2022-06-13T14:05:59.106.682.200 (XO: 69.106683563)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 156.188 usec -> 2022-06-13T14:05:59.106.709.913 (XO: 69.106711275)

 - Block number: 6 starting at 2022-06-13T14:06:15.279.486.725 (XO: 85.279489575 - 85.279795576, t=43.6 degC)  Rank: 00081
   TM values: Max =   230  Min =  -247  Ave =    -4.55  Std =    26.21
   Min TM value:-247 at an offset of 155.763 usec -> 2022-06-13T14:06:15.279.642.488 (XO: 85.279645338)
   Max TM value:230 at an offset of 128.713 usec -> 2022-06-13T14:06:15.279.615.437 (XO: 85.279618288)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 128.588 usec -> 2022-06-13T14:06:15.279.615.312 (XO: 85.279618163)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 155.601 usec -> 2022-06-13T14:06:15.279.642.325 (XO: 85.279645176)

 - Block number: 7 starting at 2022-06-13T14:06:34.712.830.725 (XO: 104.712835049 - 104.713141050, t=43.6 degC)  Rank: 00081
   TM values: Max =   228  Min =  -285  Ave =    -4.53  Std =    33.30
   Min TM value:-285 at an offset of 230.676 usec -> 2022-06-13T14:06:34.713.061.400 (XO: 104.713065724)
   Max TM value:228 at an offset of 126.425 usec -> 2022-06-13T14:06:34.712.957.150 (XO: 104.712961474)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 128.938 usec -> 2022-06-13T14:06:34.712.959.662 (XO: 104.712963987)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 126.513 usec -> 2022-06-13T14:06:34.712.957.237 (XO: 104.712961562)
