-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar 18 23:52:38 2023 UTC
-- File 20220613_095931_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.151737 to 25.411958
-each separately from 3s to 25s and from 151737us to 411958us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.259643 in 1 packets: 18
MIU packet time from 2022-06-13T09:59:37.151.737.000 to 2022-06-13T09:59:59.411.958.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.259643 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.6406490 to 26.6405730
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999240 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220613_095931_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.7923875 to 4.7926765
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.2595670  -- OK for SL --
   MIU time from 2022-06-13T09:59:37.151.737.000 to 2022-06-13T09:59:37.152.026.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.259643

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0522435 to 27.0525325
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-13T09:59:59.411.669.000 to 2022-06-13T09:59:59.411.958.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.792387500 corrected by 40bit 10MHz counter of 4.640649000 obtained at 1Hz counter of 3. Result: 3.151739016
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-13T09:59:37.151.596.725
   Max 1st sample MIU time among all SL packets: 2022-06-13T09:59:59.411.817.725
   Min 1st sample XO time among all SL packets: 3.151598740
   Max 1st sample XO time among all SL packets: 25.411820666
   TM values from -253 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-13T09:59:37.151.596.725 (XO: 3.151598740 - 3.151904741, t=43.6 degC)  Rank: 00080
   TM values: Max =   227  Min =  -253  Ave =    -4.41  Std =    24.10
   Min TM value:-253 at an offset of 126.100 usec -> 2022-06-13T09:59:37.151.722.825 (XO: 3.151724841)
   Max TM value:227 at an offset of 125.838 usec -> 2022-06-13T09:59:37.151.722.562 (XO: 3.151724578)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 170.751 usec -> 2022-06-13T09:59:37.151.767.475 (XO: 3.151769491)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 125.938 usec -> 2022-06-13T09:59:37.151.722.662 (XO: 3.151724678)

 - Block number: 2 starting at 2022-06-13T09:59:59.411.528.725 (XO: 25.411531665 - 25.411837666, t=43.6 degC)  Rank: 00082
   TM values: Max =   197  Min =  -235  Ave =    -4.47  Std =    26.91
   Min TM value:-235 at an offset of 151.751 usec -> 2022-06-13T09:59:59.411.680.475 (XO: 25.411683416)
   Max TM value:197 at an offset of 127.313 usec -> 2022-06-13T09:59:59.411.656.037 (XO: 25.411658978)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 81.563 usec -> 2022-06-13T09:59:59.411.610.287 (XO: 25.411613228)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 127.450 usec -> 2022-06-13T09:59:59.411.656.175 (XO: 25.411659116)
