-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 17 00:54:09 2023 UTC
-- File 20220612_154345_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.313929 to 47.602483
-each separately from 3s to 47s and from 281005us to 602483us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.064163 in 1 packets: 18
      13.902335 in 1 packets: 36
      6.320900 in 1 packets: 54
MIU packet time from 2022-06-12T15:43:51.313.929.000 to 2022-06-12T15:44:35.602.483.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.064163 in 1 packets: 18
      13.902335 in 1 packets: 36
      6.320900 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.2542522 to 49.2540992
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999166 in 1 packets: 18
      13.9999513 in 1 packets: 36
      5.9999791 in 1 packets: 54
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      FE -> 57.6 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C5 -> 43.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220612_154345_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.5681829 to 5.5684719
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0640790  -- OK for SL --
   MIU time from 2022-06-12T15:43:51.313.929.000 to 2022-06-12T15:43:51.314.218.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.064163

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.6325509 to 29.6328399
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.9022870  -- OK for SL --
   MIU time from 2022-06-12T15:44:15.378.381.000 to 2022-06-12T15:44:15.378.670.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.902335

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.5351269 to 43.5354159
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.3208790  -- OK for SL --
   MIU time from 2022-06-12T15:44:29.281.005.000 to 2022-06-12T15:44:29.281.294.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.320900

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.8562949 to 49.8565839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-12T15:44:35.602.194.000 to 2022-06-12T15:44:35.602.483.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.568182900 corrected by 40bit 10MHz counter of 5.254252200 obtained at 1Hz counter of 3. Result: 3.313931767
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-12T15:43:51.313.788.725
   Max 1st sample MIU time among all SL packets: 2022-06-12T15:44:35.602.342.725
   Min 1st sample XO time among all SL packets: 3.313791492
   Max 1st sample XO time among all SL packets: 47.602346473
   TM values from -239 to 236 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-12T15:43:51.313.788.725 (XO: 3.313791492 - 3.314097493, t=43.6 degC)  Rank: 00086
   TM values: Max =   236  Min =  -235  Ave =    -4.51  Std =    30.16
   Min TM value:-235 at an offset of 132.900 usec -> 2022-06-12T15:43:51.313.921.625 (XO: 3.313924392)
   Max TM value:236 at an offset of 159.163 usec -> 2022-06-12T15:43:51.313.947.888 (XO: 3.313950655)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 159.038 usec -> 2022-06-12T15:43:51.313.947.763 (XO: 3.313950530)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 191.038 usec -> 2022-06-12T15:43:51.313.979.763 (XO: 3.313982530)

 - Block number: 2 starting at 2022-06-12T15:44:15.378.240.725 (XO: 27.378243149 - 27.378549150, t=43.6 degC)  Rank: 00081
   TM values: Max =   215  Min =  -239  Ave =    -4.46  Std =    31.99
   Min TM value:-239 at an offset of 204.576 usec -> 2022-06-12T15:44:15.378.445.300 (XO: 27.378447725)
   Max TM value:215 at an offset of 204.826 usec -> 2022-06-12T15:44:15.378.445.550 (XO: 27.378447975)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 204.663 usec -> 2022-06-12T15:44:15.378.445.388 (XO: 27.378447812)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 121.113 usec -> 2022-06-12T15:44:15.378.361.837 (XO: 27.378364262)

 - Block number: 3 starting at 2022-06-12T15:44:29.280.864.725 (XO: 41.280867508 - 41.281173509, t=43.6 degC)  Rank: 00073
   TM values: Max =   212  Min =  -216  Ave =    -4.53  Std =    32.66
   Min TM value:-216 at an offset of 107.913 usec -> 2022-06-12T15:44:29.280.972.637 (XO: 41.280975421)
   Max TM value:212 at an offset of 165.588 usec -> 2022-06-12T15:44:29.281.030.313 (XO: 41.281033096)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 179.676 usec -> 2022-06-12T15:44:29.281.044.400 (XO: 41.281047184)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 165.688 usec -> 2022-06-12T15:44:29.281.030.413 (XO: 41.281033196)

 - Block number: 4 starting at 2022-06-12T15:44:35.602.053.725 (XO: 47.602057472 - 47.602363473, t=43.6 degC)  Rank: 00073
   TM values: Max =   171  Min =  -217  Ave =    -4.45  Std =    23.61
   Min TM value:-217 at an offset of 131.300 usec -> 2022-06-12T15:44:35.602.185.025 (XO: 47.602188772)
   Max TM value:171 at an offset of 131.050 usec -> 2022-06-12T15:44:35.602.184.775 (XO: 47.602188522)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 128.813 usec -> 2022-06-12T15:44:35.602.182.537 (XO: 47.602186285)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 131.200 usec -> 2022-06-12T15:44:35.602.184.925 (XO: 47.602188672)
