-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 22:15:50 2023 UTC
-- File 20220612_012506_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.013386 to 79.043935
-each separately from 8s to 79s and from 13386us to 943906us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.929942 in 1 packets: 18
      6.979703 in 1 packets: 36
      23.679825 in 1 packets: 54
      10.439653 in 1 packets: 72
      12.999692 in 1 packets: 90
MIU packet time from 2022-06-12T01:25:17.013.386.000 to 2022-06-12T01:26:28.043.935.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.929942 in 1 packets: 18
      6.979703 in 1 packets: 36
      23.679825 in 1 packets: 54
      10.439653 in 1 packets: 72
      12.999692 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.6012202 to 80.6009765
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999450 in 1 packets: 18
      6.9999759 in 1 packets: 36
      23.9999178 in 1 packets: 54
      10.9999623 in 1 packets: 72
      12.9999553 in 1 packets: 90
Temperature FPGA: 
      FF -> 56.3 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C5 -> 43.6 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 91
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 91
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220612_012506_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6146077 to 9.6148967
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9298870  -- OK for SL --
   MIU time from 2022-06-12T01:25:17.013.386.000 to 2022-06-12T01:25:17.013.675.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.929942

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.5447837 to 26.5450727
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.9796790  -- OK for SL --
   MIU time from 2022-06-12T01:25:33.943.617.000 to 2022-06-12T01:25:33.943.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.979703

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.5247517 to 33.5250407
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.6797430  -- OK for SL --
   MIU time from 2022-06-12T01:25:40.923.609.000 to 2022-06-12T01:25:40.923.898.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.679825

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.2047837 to 57.2050727
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4396150  -- OK for SL --
   MIU time from 2022-06-12T01:26:04.603.723.000 to 2022-06-12T01:26:04.604.012.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.439653

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.6446877 to 67.6449767
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9996470  -- OK for SL --
   MIU time from 2022-06-12T01:26:15.043.665.000 to 2022-06-12T01:26:15.043.954.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.999692

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.6446237 to 80.6449127
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-12T01:26:28.043.646.000 to 2022-06-12T01:26:28.043.935.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.614607700 corrected by 40bit 10MHz counter of 9.601220200 obtained at 1Hz counter of 8. Result: 8.013387547
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-12T01:25:17.013.245.725
   Max 1st sample MIU time among all SL packets: 2022-06-12T01:26:28.043.794.725
   Min 1st sample XO time among all SL packets: 8.013247271
   Max 1st sample XO time among all SL packets: 79.043796074
   TM values from -264 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-12T01:25:17.013.245.725 (XO: 8.013247271 - 8.013553272, t=43.6 degC)  Rank: 00079
   TM values: Max =   237  Min =  -217  Ave =    -4.48  Std =    31.57
   Min TM value:-217 at an offset of 128.050 usec -> 2022-06-12T01:25:17.013.373.775 (XO: 8.013375322)
   Max TM value:237 at an offset of 130.388 usec -> 2022-06-12T01:25:17.013.376.112 (XO: 8.013377659)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 230.451 usec -> 2022-06-12T01:25:17.013.476.175 (XO: 8.013477722)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 204.013 usec -> 2022-06-12T01:25:17.013.449.738 (XO: 8.013451285)

 - Block number: 2 starting at 2022-06-12T01:25:33.943.476.725 (XO: 24.943481527 - 24.943787528, t=43.6 degC)  Rank: 00078
   TM values: Max =   225  Min =  -225  Ave =    -4.41  Std =    29.71
   Min TM value:-225 at an offset of 116.175 usec -> 2022-06-12T01:25:33.943.592.900 (XO: 24.943597703)
   Max TM value:225 at an offset of 116.450 usec -> 2022-06-12T01:25:33.943.593.175 (XO: 24.943597978)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 116.300 usec -> 2022-06-12T01:25:33.943.593.025 (XO: 24.943597828)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 173.001 usec -> 2022-06-12T01:25:33.943.649.725 (XO: 24.943654528)

 - Block number: 3 starting at 2022-06-12T01:25:40.923.468.725 (XO: 31.923473465 - 31.923779466, t=43.6 degC)  Rank: 00078
   TM values: Max =   238  Min =  -264  Ave =    -4.48  Std =    34.51
   Min TM value:-264 at an offset of 127.675 usec -> 2022-06-12T01:25:40.923.596.400 (XO: 31.923601140)
   Max TM value:238 at an offset of 130.013 usec -> 2022-06-12T01:25:40.923.598.737 (XO: 31.923603478)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 129.875 usec -> 2022-06-12T01:25:40.923.598.600 (XO: 31.923603340)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 127.475 usec -> 2022-06-12T01:25:40.923.596.200 (XO: 31.923600940)

 - Block number: 4 starting at 2022-06-12T01:26:04.603.582.725 (XO: 55.603586577 - 55.603892578, t=43.6 degC)  Rank: 00077
   TM values: Max =   248  Min =  -240  Ave =    -4.52  Std =    30.24
   Min TM value:-240 at an offset of 118.638 usec -> 2022-06-12T01:26:04.603.701.362 (XO: 55.603705215)
   Max TM value:248 at an offset of 118.900 usec -> 2022-06-12T01:26:04.603.701.625 (XO: 55.603705478)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 204.726 usec -> 2022-06-12T01:26:04.603.787.450 (XO: 55.603791303)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 116.363 usec -> 2022-06-12T01:26:04.603.699.087 (XO: 55.603702940)

 - Block number: 5 starting at 2022-06-12T01:26:15.043.524.725 (XO: 66.043526377 - 66.043832378, t=43.6 degC)  Rank: 00078
   TM values: Max =   242  Min =  -263  Ave =    -4.56  Std =    27.87
   Min TM value:-263 at an offset of 133.438 usec -> 2022-06-12T01:26:15.043.658.162 (XO: 66.043659815)
   Max TM value:242 at an offset of 133.188 usec -> 2022-06-12T01:26:15.043.657.912 (XO: 66.043659565)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 135.650 usec -> 2022-06-12T01:26:15.043.660.375 (XO: 66.043662028)
   Min dTM/dt [1/us]:-2.70e+03 at an offset of 133.300 usec -> 2022-06-12T01:26:15.043.658.025 (XO: 66.043659678)

 - Block number: 6 starting at 2022-06-12T01:26:28.043.505.725 (XO: 79.043507073 - 79.043813074, t=43.6 degC)  Rank: 00084
   TM values: Max =   188  Min =  -239  Ave =    -4.57  Std =    20.84
   Min TM value:-239 at an offset of 126.975 usec -> 2022-06-12T01:26:28.043.632.700 (XO: 79.043634048)
   Max TM value:188 at an offset of 129.363 usec -> 2022-06-12T01:26:28.043.635.087 (XO: 79.043636436)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 129.275 usec -> 2022-06-12T01:26:28.043.635.000 (XO: 79.043636348)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 126.800 usec -> 2022-06-12T01:26:28.043.632.525 (XO: 79.043633873)
