-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 15:30:45 2023 UTC
-- File 20220610_234909_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.396799 to 15.592659
-each separately from 5s to 15s and from 396799us to 592659us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.195282 in 1 packets: 18
MIU packet time from 2022-06-10T23:49:17.396.799.000 to 2022-06-10T23:49:27.592.659.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.195282 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7582879 to 16.7582526
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999647 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220610_234909_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.1550879 to 7.1553769
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1952470  -- OK for SL --
   MIU time from 2022-06-10T23:49:17.396.799.000 to 2022-06-10T23:49:17.397.088.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.195282

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3506239 to 17.3509129
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-10T23:49:27.592.370.000 to 2022-06-10T23:49:27.592.659.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.155087900 corrected by 40bit 10MHz counter of 6.758287900 obtained at 1Hz counter of 5. Result: 5.396801389
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-10T23:49:17.396.658.725
   Max 1st sample MIU time among all SL packets: 2022-06-10T23:49:27.592.518.725
   Min 1st sample XO time among all SL packets: 5.396661113
   Max 1st sample XO time among all SL packets: 15.592522158
   TM values from -273 to 251 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-10T23:49:17.396.658.725 (XO: 5.396661113 - 5.396967114, t=43.6 degC)  Rank: 00080
   TM values: Max =   240  Min =  -273  Ave =    -4.45  Std =    29.07
   Min TM value:-273 at an offset of 202.876 usec -> 2022-06-10T23:49:17.396.861.600 (XO: 5.396863989)
   Max TM value:240 at an offset of 133.475 usec -> 2022-06-10T23:49:17.396.792.200 (XO: 5.396794589)
   Max dTM/dt [1/us]:2.40e+03 at an offset of 133.350 usec -> 2022-06-10T23:49:17.396.792.075 (XO: 5.396794464)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 202.763 usec -> 2022-06-10T23:49:17.396.861.488 (XO: 5.396863877)

 - Block number: 2 starting at 2022-06-10T23:49:27.592.229.725 (XO: 15.592233157 - 15.592539158, t=43.6 degC)  Rank: 00082
   TM values: Max =   251  Min =  -254  Ave =    -4.44  Std =    22.37
   Min TM value:-254 at an offset of 130.138 usec -> 2022-06-10T23:49:27.592.359.862 (XO: 15.592363295)
   Max TM value:251 at an offset of 132.475 usec -> 2022-06-10T23:49:27.592.362.200 (XO: 15.592365633)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 132.363 usec -> 2022-06-10T23:49:27.592.362.087 (XO: 15.592365520)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 129.988 usec -> 2022-06-10T23:49:27.592.359.712 (XO: 15.592363145)
