-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 09:17:55 2023 UTC
-- File 20220610_015209_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.588172 to 39.589093
-each separately from 9s to 39s and from 548601us to 589093us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.960140 in 1 packets: 18
      17.039914 in 1 packets: 36
MIU packet time from 2022-06-10T01:52:21.588.172.000 to 2022-06-10T01:52:51.589.093.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.960140 in 1 packets: 18
      17.039914 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7014763 to 40.7013721
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999549 in 1 packets: 18
      16.9999409 in 1 packets: 36
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220610_015209_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.2896493 to 11.2899383
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9600950  -- OK for SL --
   MIU time from 2022-06-10T01:52:21.588.172.000 to 2022-06-10T01:52:21.588.461.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.960140

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2500333 to 24.2503223
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0398550  -- OK for SL --
   MIU time from 2022-06-10T01:52:34.548.601.000 to 2022-06-10T01:52:34.548.890.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.039914

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.2901773 to 41.2904663
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-10T01:52:51.588.804.000 to 2022-06-10T01:52:51.589.093.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.289649300 corrected by 40bit 10MHz counter of 10.701476300 obtained at 1Hz counter of 9. Result: 9.588175000
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-10T01:52:21.588.031.725
   Max 1st sample MIU time among all SL packets: 2022-06-10T01:52:51.588.952.725
   Min 1st sample XO time among all SL packets: 9.588034724
   Max 1st sample XO time among all SL packets: 39.588955927
   TM values from -298 to 247 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-10T01:52:21.588.031.725 (XO: 9.588034724 - 9.588340725, t=42.0 degC)  Rank: 00083
   TM values: Max =   247  Min =  -267  Ave =    -4.45  Std =    35.05
   Min TM value:-267 at an offset of 122.675 usec -> 2022-06-10T01:52:21.588.154.400 (XO: 9.588157400)
   Max TM value:247 at an offset of 108.375 usec -> 2022-06-10T01:52:21.588.140.100 (XO: 9.588143100)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 108.238 usec -> 2022-06-10T01:52:21.588.139.962 (XO: 9.588142962)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 105.825 usec -> 2022-06-10T01:52:21.588.137.550 (XO: 9.588140550)

 - Block number: 2 starting at 2022-06-10T01:52:34.548.460.725 (XO: 22.548463745 - 22.548769746, t=42.0 degC)  Rank: 00074
   TM values: Max =   244  Min =  -298  Ave =    -4.47  Std =    23.58
   Min TM value:-298 at an offset of 84.400 usec -> 2022-06-10T01:52:34.548.545.125 (XO: 22.548548145)
   Max TM value:244 at an offset of 84.113 usec -> 2022-06-10T01:52:34.548.544.837 (XO: 22.548547857)
   Max dTM/dt [1/us]:2.58e+03 at an offset of 86.625 usec -> 2022-06-10T01:52:34.548.547.350 (XO: 22.548550370)
   Min dTM/dt [1/us]:-3.02e+03 at an offset of 84.213 usec -> 2022-06-10T01:52:34.548.544.937 (XO: 22.548547957)

 - Block number: 3 starting at 2022-06-10T01:52:51.588.663.725 (XO: 39.588666926 - 39.588972928, t=42.0 degC)  Rank: 00086
   TM values: Max =   244  Min =  -259  Ave =    -4.56  Std =    24.38
   Min TM value:-259 at an offset of 117.400 usec -> 2022-06-10T01:52:51.588.781.125 (XO: 39.588784327)
   Max TM value:244 at an offset of 119.763 usec -> 2022-06-10T01:52:51.588.783.487 (XO: 39.588786689)
   Max dTM/dt [1/us]:2.64e+03 at an offset of 119.625 usec -> 2022-06-10T01:52:51.588.783.350 (XO: 39.588786552)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 117.225 usec -> 2022-06-10T01:52:51.588.780.950 (XO: 39.588784152)
