-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 08:52:53 2023 UTC
-- File 20220610_005259_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177028
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2022-06-10T00:53:02.000.000.000 to 2022-06-10T00:53:16.177.028.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7470352 to 15.7469862
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 10 packets: 2, 3, 4, 10, 11, 12, 14, 15, 16, 17
      0.9999966 in 2 packets: 5, 8
      0.9999964 in 2 packets: 6, 9
   XO PPM: 
      -3.5 in 10 packets: 2, 3, 4, 10, 11, 12, 14, 15, 16, 17
      -3.4 in 2 packets: 5, 8
      -3.6 in 2 packets: 6, 9
Temperature FPGA: 
      01 -> 53.7 degC in 13 packets: 1, 3, 4, 5, 6, 7, 8, 10, 11, 12, 16, 17, 18
      00 -> 55.0 degC in 1 packets: 2
      02 -> 52.4 degC in 4 packets: 9, 13, 14, 15
Temperature XO: 
      C4 -> 42.0 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      1 in 1 packets: 14
      2 in 4 packets: 15, 16, 17, 18
Number of generated alerts: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      2 in 1 packets: 14
      3 in 4 packets: 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220610_005259_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7470353 to 15.9240153
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-10T00:53:02.000.000.000 to 2022-06-10T00:53:16.177.028.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.747035300 corrected by 40bit 10MHz counter of 1.747035200 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996600
   Min 1st sample MIU time among all SR packets: 2022-06-10T00:53:01.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-06-10T00:53:16.177.027.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177029645
   TM values from -157 to 154 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-06-10T00:53:01.999.999.925 (XO: 0.000000025 - 0.000017025, t=42.0 degC)
   TM values: Max =    13  Min =   -26  Ave =    -4.71  Std =     6.32

 - Packet number: 2 starting at 2022-06-10T00:53:02.833.938.925 (XO: 0.833942944 - 0.833959944, t=42.0 degC)
   TM values: Max =    12  Min =   -26  Ave =    -4.66  Std =     5.84

 - Packet number: 3 starting at 2022-06-10T00:53:03.667.881.925 (XO: 1.667885863 - 1.667902863, t=42.0 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.42  Std =     5.33

 - Packet number: 4 starting at 2022-06-10T00:53:04.501.825.925 (XO: 2.501828781 - 2.501845781, t=42.0 degC)
   TM values: Max =    10  Min =   -22  Ave =    -4.56  Std =     5.37

 - Packet number: 5 starting at 2022-06-10T00:53:05.335.768.925 (XO: 3.335771667 - 3.335788667, t=42.0 degC)
   TM values: Max =    11  Min =   -24  Ave =    -4.50  Std =     5.83

 - Packet number: 6 starting at 2022-06-10T00:53:06.169.712.925 (XO: 4.169714536 - 4.169731536, t=42.0 degC)
   TM values: Max =    13  Min =   -23  Ave =    -4.87  Std =     5.44

 - Packet number: 7 starting at 2022-06-10T00:53:07.003.655.925 (XO: 5.003657537 - 5.003674537, t=42.0 degC)
   TM values: Max =    12  Min =   -21  Ave =    -4.74  Std =     5.44

 - Packet number: 8 starting at 2022-06-10T00:53:07.837.595.925 (XO: 5.837600373 - 5.837617373, t=42.0 degC)
   TM values: Max =    14  Min =   -21  Ave =    -4.34  Std =     5.49

 - Packet number: 9 starting at 2022-06-10T00:53:08.671.539.925 (XO: 6.671543343 - 6.671560343, t=42.0 degC)
   TM values: Max =   154  Min =  -157  Ave =    -4.80  Std =    38.03

 - Packet number: 10 starting at 2022-06-10T00:53:09.505.482.925 (XO: 7.505486294 - 7.505503294, t=42.0 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.44  Std =     4.25

 - Packet number: 11 starting at 2022-06-10T00:53:10.339.426.925 (XO: 8.339429213 - 8.339446213, t=42.0 degC)
   TM values: Max =    18  Min =   -27  Ave =    -4.87  Std =     6.70

 - Packet number: 12 starting at 2022-06-10T00:53:11.173.369.925 (XO: 9.173372132 - 9.173389132, t=42.0 degC)
   TM values: Max =    14  Min =   -25  Ave =    -4.50  Std =     6.62

 - Packet number: 13 starting at 2022-06-10T00:53:12.007.313.925 (XO: 10.007315051 - 10.007332051, t=42.0 degC)
   TM values: Max =    11  Min =   -25  Ave =    -4.49  Std =     5.98

 - Packet number: 14 starting at 2022-06-10T00:53:12.841.253.925 (XO: 10.841257969 - 10.841274969, t=42.0 degC)
   TM values: Max =    37  Min =   -46  Ave =    -4.41  Std =    10.39

 - Packet number: 15 starting at 2022-06-10T00:53:13.675.196.925 (XO: 11.675200888 - 11.675217888, t=42.0 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.59  Std =     6.54

 - Packet number: 16 starting at 2022-06-10T00:53:14.509.140.925 (XO: 12.509143807 - 12.509160807, t=42.0 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.43  Std =     5.99

 - Packet number: 17 starting at 2022-06-10T00:53:15.343.083.925 (XO: 13.343086726 - 13.343103726, t=42.0 degC)
   TM values: Max =    14  Min =   -24  Ave =    -4.43  Std =     6.51

 - Packet number: 18 starting at 2022-06-10T00:53:16.177.027.925 (XO: 14.177029645 - 14.177046645, t=42.0 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.53  Std =     6.76
