-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 06:43:18 2023 UTC
-- File 20220609_144504_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.937776 to 30.158699
-each separately from 0s to 30s and from 158410us to 938065us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.900241 in 1 packets: 18
      1.319815 in 1 packets: 36
MIU packet time from 2022-06-09T14:45:07.937.776.000 to 2022-06-09T14:45:37.158.699.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.900241 in 1 packets: 18
      1.319815 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2468619 to 32.2467567
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999018 in 1 packets: 18
      1.9999930 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220609_144504_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1846397 to 3.1849287
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.9001430  -- OK for SL --
   MIU time from 2022-06-09T14:45:07.937.776.000 to 2022-06-09T14:45:07.938.065.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.900241

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.0850717 to 31.0853607
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.3198070  -- OK for SL --
   MIU time from 2022-06-09T14:45:35.838.306.000 to 2022-06-09T14:45:35.838.595.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.319815

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4051677 to 32.4054567
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-09T14:45:37.158.410.000 to 2022-06-09T14:45:37.158.699.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 3.184639700 corrected by 40bit 10MHz counter of 2.246861900 obtained at 1Hz counter of 0. Result: 0.937781082
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-09T14:45:07.937.635.725
   Max 1st sample MIU time among all SL packets: 2022-06-09T14:45:37.158.558.725
   Min 1st sample XO time among all SL packets: 0.937640807
   Max 1st sample XO time among all SL packets: 30.158560280
   TM values from -271 to 239 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-09T14:45:07.937.635.725 (XO: 0.937640807 - 0.937946808, t=42.0 degC)  Rank: 00085
   TM values: Max =   199  Min =  -232  Ave =    -4.56  Std =    24.72
   Min TM value:-232 at an offset of 122.088 usec -> 2022-06-09T14:45:07.937.757.812 (XO: 0.937762895)
   Max TM value:199 at an offset of 124.425 usec -> 2022-06-09T14:45:07.937.760.150 (XO: 0.937765232)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 124.300 usec -> 2022-06-09T14:45:07.937.760.025 (XO: 0.937765107)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 138.388 usec -> 2022-06-09T14:45:07.937.774.112 (XO: 0.937779195)

 - Block number: 2 starting at 2022-06-09T14:45:35.838.165.725 (XO: 28.838170659 - 28.838476660, t=42.0 degC)  Rank: 00081
   TM values: Max =   239  Min =  -271  Ave =    -4.58  Std =    30.23
   Min TM value:-271 at an offset of 172.051 usec -> 2022-06-09T14:45:35.838.337.775 (XO: 28.838342709)
   Max TM value:239 at an offset of 116.625 usec -> 2022-06-09T14:45:35.838.282.350 (XO: 28.838287284)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 174.313 usec -> 2022-06-09T14:45:35.838.340.038 (XO: 28.838344972)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 114.075 usec -> 2022-06-09T14:45:35.838.279.800 (XO: 28.838284734)

 - Block number: 3 starting at 2022-06-09T14:45:37.158.269.725 (XO: 30.158271279 - 30.158577280, t=42.0 degC)  Rank: 00080
   TM values: Max =   231  Min =  -263  Ave =    -4.64  Std =    22.84
   Min TM value:-263 at an offset of 124.688 usec -> 2022-06-09T14:45:37.158.394.412 (XO: 30.158395967)
   Max TM value:231 at an offset of 127.063 usec -> 2022-06-09T14:45:37.158.396.787 (XO: 30.158398342)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 126.950 usec -> 2022-06-09T14:45:37.158.396.675 (XO: 30.158398229)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 124.513 usec -> 2022-06-09T14:45:37.158.394.237 (XO: 30.158395792)
