-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 05:21:19 2023 UTC
-- File 20220609_132450_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.153961 to 61.095134
-each separately from 5s to 61s and from 94845us to 915051us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.680236 in 1 packets: 18
      25.280058 in 1 packets: 36
      7.799640 in 1 packets: 54
      9.179794 in 1 packets: 72
MIU packet time from 2022-06-09T13:24:58.153.961.000 to 2022-06-09T13:25:54.095.134.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.680236 in 1 packets: 18
      25.280058 in 1 packets: 36
      7.799640 in 1 packets: 54
      9.179794 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.2560811 to 63.2558849
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999545 in 1 packets: 18
      25.9999088 in 1 packets: 36
      6.9999755 in 1 packets: 54
      9.9999650 in 1 packets: 72
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      01 -> 53.7 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C4 -> 42.0 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 1 packets: 37
      3 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220609_132450_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4100430 to 7.4103320
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6801910  -- OK for SL --
   MIU time from 2022-06-09T13:24:58.153.961.000 to 2022-06-09T13:24:58.154.250.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.680236

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0905230 to 21.0908120
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.2799670  -- OK for SL --
   MIU time from 2022-06-09T13:25:11.834.486.000 to 2022-06-09T13:25:11.834.775.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.280058

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.3707790 to 46.3710680
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7996150  -- OK for SL --
   MIU time from 2022-06-09T13:25:37.114.833.000 to 2022-06-09T13:25:37.115.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.799640

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.1706830 to 54.1709720
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1797590  -- OK for SL --
   MIU time from 2022-06-09T13:25:44.914.762.000 to 2022-06-09T13:25:44.915.051.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.179794

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.3507310 to 63.3510200
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-09T13:25:54.094.845.000 to 2022-06-09T13:25:54.095.134.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.410043000 corrected by 40bit 10MHz counter of 7.256081100 obtained at 1Hz counter of 5. Result: 5.153962439
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-09T13:24:58.153.820.725
   Max 1st sample MIU time among all SL packets: 2022-06-09T13:25:54.094.993.725
   Min 1st sample XO time among all SL packets: 5.153822163
   Max 1st sample XO time among all SL packets: 61.094995157
   TM values from -268 to 229 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-09T13:24:58.153.820.725 (XO: 5.153822163 - 5.154128164, t=42.0 degC)  Rank: 00085
   TM values: Max =   226  Min =  -257  Ave =    -4.48  Std =    31.90
   Min TM value:-257 at an offset of 127.338 usec -> 2022-06-09T13:24:58.153.948.062 (XO: 5.153949501)
   Max TM value:226 at an offset of 127.063 usec -> 2022-06-09T13:24:58.153.947.787 (XO: 5.153949226)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 129.588 usec -> 2022-06-09T13:24:58.153.950.312 (XO: 5.153951751)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 127.163 usec -> 2022-06-09T13:24:58.153.947.887 (XO: 5.153949326)

 - Block number: 2 starting at 2022-06-09T13:25:11.834.345.725 (XO: 18.834350129 - 18.834656130, t=42.0 degC)  Rank: 00078
   TM values: Max =   229  Min =  -268  Ave =    -4.48  Std =    18.80
   Min TM value:-268 at an offset of 111.525 usec -> 2022-06-09T13:25:11.834.457.250 (XO: 18.834461654)
   Max TM value:229 at an offset of 111.263 usec -> 2022-06-09T13:25:11.834.456.987 (XO: 18.834461392)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 113.775 usec -> 2022-06-09T13:25:11.834.459.500 (XO: 18.834463904)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 111.363 usec -> 2022-06-09T13:25:11.834.457.087 (XO: 18.834461492)

 - Block number: 3 starting at 2022-06-09T13:25:37.114.692.725 (XO: 44.114694726 - 44.115000728, t=42.0 degC)  Rank: 00091
   TM values: Max =   209  Min =  -229  Ave =    -4.46  Std =    26.42
   Min TM value:-229 at an offset of 155.001 usec -> 2022-06-09T13:25:37.114.847.725 (XO: 44.114849727)
   Max TM value:209 at an offset of 128.275 usec -> 2022-06-09T13:25:37.114.821.000 (XO: 44.114823002)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 155.088 usec -> 2022-06-09T13:25:37.114.847.813 (XO: 44.114849814)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 128.400 usec -> 2022-06-09T13:25:37.114.821.125 (XO: 44.114823127)

 - Block number: 4 starting at 2022-06-09T13:25:44.914.621.725 (XO: 51.914626026 - 51.914932027, t=42.0 degC)  Rank: 00077
   TM values: Max =   200  Min =  -218  Ave =    -4.52  Std =    24.33
   Min TM value:-218 at an offset of 101.463 usec -> 2022-06-09T13:25:44.914.723.187 (XO: 51.914727489)
   Max TM value:200 at an offset of 103.988 usec -> 2022-06-09T13:25:44.914.725.712 (XO: 51.914730014)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 101.600 usec -> 2022-06-09T13:25:44.914.723.325 (XO: 51.914727627)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 115.400 usec -> 2022-06-09T13:25:44.914.737.125 (XO: 51.914741427)

 - Block number: 5 starting at 2022-06-09T13:25:54.094.704.725 (XO: 61.094706156 - 61.095012158, t=42.0 degC)  Rank: 00079
   TM values: Max =   229  Min =  -256  Ave =    -4.67  Std =    29.92
   Min TM value:-256 at an offset of 183.876 usec -> 2022-06-09T13:25:54.094.888.600 (XO: 61.094890032)
   Max TM value:229 at an offset of 186.226 usec -> 2022-06-09T13:25:54.094.890.950 (XO: 61.094892382)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 126.288 usec -> 2022-06-09T13:25:54.094.831.012 (XO: 61.094832444)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 126.113 usec -> 2022-06-09T13:25:54.094.830.837 (XO: 61.094832269)
