-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 03:12:34 2023 UTC
-- File 20220609_094817_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.420131 to 67.716188
-each separately from 1s to 67s and from 196941us to 959701us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.538992 in 1 packets: 18
      13.358832 in 1 packets: 36
      15.878119 in 1 packets: 54
      13.518669 in 1 packets: 72
MIU packet time from 2022-06-09T09:48:21.420.131.000 to 2022-06-09T09:49:27.716.188.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.538992 in 1 packets: 18
      13.358832 in 1 packets: 36
      15.878119 in 1 packets: 54
      13.518669 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7554092 to 68.7551776
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999193 in 1 packets: 18
      13.9999509 in 1 packets: 36
      15.9999439 in 1 packets: 54
      12.9999543 in 1 packets: 72
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      02 -> 52.4 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C4 -> 42.0 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220609_094817_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1755417 to 3.1758307
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5389110  -- OK for SL --
   MIU time from 2022-06-09T09:48:21.420.131.000 to 2022-06-09T09:48:21.420.420.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.538992

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7147417 to 26.7150307
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3587830  -- OK for SL --
   MIU time from 2022-06-09T09:48:44.959.412.000 to 2022-06-09T09:48:44.959.701.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.358832

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.0738137 to 40.0741027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8780630  -- OK for SL --
   MIU time from 2022-06-09T09:48:58.318.533.000 to 2022-06-09T09:48:58.318.822.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.878119

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.9521657 to 55.9524547
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5186230  -- OK for SL --
   MIU time from 2022-06-09T09:49:14.196.941.000 to 2022-06-09T09:49:14.197.230.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.518669

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.4710777 to 69.4713667
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-09T09:49:27.715.899.000 to 2022-06-09T09:49:27.716.188.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.175541700 corrected by 40bit 10MHz counter of 2.755409200 obtained at 1Hz counter of 1. Result: 1.420133970
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-09T09:48:21.419.990.725
   Max 1st sample MIU time among all SL packets: 2022-06-09T09:49:27.716.047.725
   Min 1st sample XO time among all SL packets: 1.419993695
   Max 1st sample XO time among all SL packets: 67.716051331
   TM values from -246 to 216 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-09T09:48:21.419.990.725 (XO: 1.419993695 - 1.420299696, t=42.0 degC)  Rank: 00079
   TM values: Max =   216  Min =  -246  Ave =    -4.50  Std =    23.74
   Min TM value:-246 at an offset of 145.738 usec -> 2022-06-09T09:48:21.420.136.463 (XO: 1.420139433)
   Max TM value:216 at an offset of 118.363 usec -> 2022-06-09T09:48:21.420.109.087 (XO: 1.420112058)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 160.938 usec -> 2022-06-09T09:48:21.420.151.663 (XO: 1.420154633)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 115.813 usec -> 2022-06-09T09:48:21.420.106.537 (XO: 1.420109508)

 - Block number: 2 starting at 2022-06-09T09:48:44.959.271.725 (XO: 24.959276282 - 24.959582284, t=42.0 degC)  Rank: 00078
   TM values: Max =   207  Min =  -236  Ave =    -4.45  Std =    22.57
   Min TM value:-236 at an offset of 115.388 usec -> 2022-06-09T09:48:44.959.387.112 (XO: 24.959391670)
   Max TM value:207 at an offset of 115.663 usec -> 2022-06-09T09:48:44.959.387.387 (XO: 24.959391945)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 115.500 usec -> 2022-06-09T09:48:44.959.387.225 (XO: 24.959391783)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 113.038 usec -> 2022-06-09T09:48:44.959.384.762 (XO: 24.959389320)

 - Block number: 3 starting at 2022-06-09T09:48:58.318.392.725 (XO: 38.318395139 - 38.318701140, t=42.0 degC)  Rank: 00081
   TM values: Max =   202  Min =  -232  Ave =    -4.51  Std =    27.04
   Min TM value:-232 at an offset of 152.426 usec -> 2022-06-09T09:48:58.318.545.150 (XO: 38.318547565)
   Max TM value:202 at an offset of 154.851 usec -> 2022-06-09T09:48:58.318.547.575 (XO: 38.318549990)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 154.713 usec -> 2022-06-09T09:48:58.318.547.438 (XO: 38.318549852)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 140.288 usec -> 2022-06-09T09:48:58.318.533.012 (XO: 38.318535427)

 - Block number: 4 starting at 2022-06-09T09:49:14.196.800.725 (XO: 54.196802814 - 54.197108815, t=42.0 degC)  Rank: 00081
   TM values: Max =   203  Min =  -240  Ave =    -4.42  Std =    25.43
   Min TM value:-240 at an offset of 154.013 usec -> 2022-06-09T09:49:14.196.954.738 (XO: 54.196956827)
   Max TM value:203 at an offset of 156.413 usec -> 2022-06-09T09:49:14.196.957.138 (XO: 54.196959227)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 156.263 usec -> 2022-06-09T09:49:14.196.956.988 (XO: 54.196959077)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 129.550 usec -> 2022-06-09T09:49:14.196.930.275 (XO: 54.196932364)

 - Block number: 5 starting at 2022-06-09T09:49:27.715.758.725 (XO: 67.715762330 - 67.716068331, t=42.0 degC)  Rank: 00077
   TM values: Max =   198  Min =  -226  Ave =    -4.38  Std =    26.40
   Min TM value:-226 at an offset of 130.063 usec -> 2022-06-09T09:49:27.715.888.787 (XO: 67.715892393)
   Max TM value:198 at an offset of 145.138 usec -> 2022-06-09T09:49:27.715.903.863 (XO: 67.715907468)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 145.001 usec -> 2022-06-09T09:49:27.715.903.725 (XO: 67.715907331)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 129.950 usec -> 2022-06-09T09:49:27.715.888.675 (XO: 67.715892281)
