-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 00:00:45 2023 UTC
-- File 20220609_001519_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.403066 to 27.282001
-each separately from 3s to 27s and from 281712us to 403355us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.878357 in 1 packets: 18
MIU packet time from 2022-06-09T00:15:25.403.066.000 to 2022-06-09T00:15:49.282.001.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.878357 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.0646975 to 29.0646114
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999139 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220609_001519_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4677646 to 5.4680536
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.8782710  -- OK for SL --
   MIU time from 2022-06-09T00:15:25.403.066.000 to 2022-06-09T00:15:25.403.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.878357

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.3463246 to 29.3466136
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-09T00:15:49.281.712.000 to 2022-06-09T00:15:49.282.001.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.467764600 corrected by 40bit 10MHz counter of 5.064697500 obtained at 1Hz counter of 3. Result: 3.403068551
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-09T00:15:25.402.925.725
   Max 1st sample MIU time among all SL packets: 2022-06-09T00:15:49.281.860.725
   Min 1st sample XO time among all SL packets: 3.402928276
   Max 1st sample XO time among all SL packets: 27.281862912
   TM values from -263 to 238 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-09T00:15:25.402.925.725 (XO: 3.402928276 - 3.403234277, t=42.0 degC)  Rank: 00080
   TM values: Max =   238  Min =  -263  Ave =    -4.54  Std =    31.30
   Min TM value:-263 at an offset of 149.551 usec -> 2022-06-09T00:15:25.403.075.275 (XO: 3.403077826)
   Max TM value:238 at an offset of 151.913 usec -> 2022-06-09T00:15:25.403.077.638 (XO: 3.403080189)
   Max dTM/dt [1/us]:2.41e+03 at an offset of 109.325 usec -> 2022-06-09T00:15:25.403.035.050 (XO: 3.403037601)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 106.950 usec -> 2022-06-09T00:15:25.403.032.675 (XO: 3.403035226)

 - Block number: 2 starting at 2022-06-09T00:15:49.281.571.725 (XO: 27.281573911 - 27.281879912, t=42.0 degC)  Rank: 00087
   TM values: Max =   230  Min =  -255  Ave =    -4.49  Std =    27.95
   Min TM value:-255 at an offset of 108.788 usec -> 2022-06-09T00:15:49.281.680.512 (XO: 27.281682698)
   Max TM value:230 at an offset of 109.038 usec -> 2022-06-09T00:15:49.281.680.762 (XO: 27.281682948)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 108.925 usec -> 2022-06-09T00:15:49.281.680.650 (XO: 27.281682836)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 106.513 usec -> 2022-06-09T00:15:49.281.678.237 (XO: 27.281680423)
