-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 15 23:40:43 2023 UTC
-- File 20220608_234008_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.270507 to 19.931967
-each separately from 9s to 19s and from 270507us to 931967us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.660882 in 1 packets: 18
MIU packet time from 2022-06-08T23:40:20.270.507.000 to 2022-06-08T23:40:30.931.967.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.660882 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4016732 to 20.4016378
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999646 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220608_234008_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.6721813 to 10.6724703
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6608470  -- OK for SL --
   MIU time from 2022-06-08T23:40:20.270.507.000 to 2022-06-08T23:40:20.270.796.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.660882

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      103 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3333173 to 21.3336063
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-08T23:40:30.931.678.000 to 2022-06-08T23:40:30.931.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.672181300 corrected by 40bit 10MHz counter of 10.401673200 obtained at 1Hz counter of 9. Result: 9.270509074
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-08T23:40:20.270.366.725
   Max 1st sample MIU time among all SL packets: 2022-06-08T23:40:30.931.826.725
   Min 1st sample XO time among all SL packets: 9.270368798
   Max 1st sample XO time among all SL packets: 19.931831580
   TM values from -436 to 405 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-08T23:40:20.270.366.725 (XO: 9.270368798 - 9.270674799, t=42.0 degC)  Rank: 00079
   TM values: Max =   244  Min =  -247  Ave =    -4.48  Std =    24.62
   Min TM value:-247 at an offset of 114.425 usec -> 2022-06-08T23:40:20.270.481.150 (XO: 9.270483224)
   Max TM value:244 at an offset of 116.813 usec -> 2022-06-08T23:40:20.270.483.537 (XO: 9.270485611)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 116.688 usec -> 2022-06-08T23:40:20.270.483.412 (XO: 9.270485486)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 114.288 usec -> 2022-06-08T23:40:20.270.481.012 (XO: 9.270483086)

 - Block number: 2 starting at 2022-06-08T23:40:30.931.537.725 (XO: 19.931542579 - 19.931848580, t=42.0 degC)  Rank: 00103
   TM values: Max =   405  Min =  -436  Ave =    -4.40  Std =    48.54
   Min TM value:-436 at an offset of 105.300 usec -> 2022-06-08T23:40:30.931.643.025 (XO: 19.931647879)
   Max TM value:405 at an offset of 110.913 usec -> 2022-06-08T23:40:30.931.648.637 (XO: 19.931653491)
   Max dTM/dt [1/us]:3.18e+03 at an offset of 110.588 usec -> 2022-06-08T23:40:30.931.648.312 (XO: 19.931653166)
   Min dTM/dt [1/us]:-5.15e+03 at an offset of 150.913 usec -> 2022-06-08T23:40:30.931.688.638 (XO: 19.931693492)
