-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 15 21:02:09 2023 UTC
-- File 20220608_153225_1229.dat, 9625  packets found.
APID:
      1229 (EW) in 9625 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 9625 packets (all packets)
Total number of packets: 
      36720 in 9625 packets (all packets)
Packet No from 1 to 9625
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 36.129985 to 49.551189
-each separately from 36s to 49s and from 129985us to 979474us.
   Differences in neigboring packets: 
      0.000000 in 9622 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -9.571715 in 1 packets: 816
      -3.849489 in 1 packets: 4488
MIU packet time from 2022-06-08T15:33:04.129.985.000 to 2022-06-08T15:33:17.551.189.000
   Differences in neigboring packets: 
      0.000000 in 9622 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -9.571715 in 1 packets: 816
      -3.849489 in 1 packets: 4488
MIU orbit number: 
      60000 in 9625 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 9625 packets (all packets)
MIU orbit information / number of events: 
      0 in 9625 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 9625 packets (all packets)
Instrument anomalies: 
      0 (OK) in 9625 packets (all packets)
FPGA configuration version: 
      211 in 9625 packets (all packets)
10 MHz counter at the last 1Hz pulse from 38.2526232 to 51.2525781
   Differences in neigboring packets: 
      0.0000000 in 9622 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -9.9999653 in 1 packets: 816
      -2.9999896 in 1 packets: 4488
Temperature FPGA: 
      00 -> 55.0 degC in 9625 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 5953 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C5 -> 43.6 degC in 3672 packets: 817, 818, 819, 820, 821, 822, 823, 824, 825, 826, 827, 828, 829, 830, 831, 832, 833, 834, ...
Temperature ant 3: 
      01 -> 108.0 degC in 9625 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 9625 packets (all packets)
Number of external events: 
      3 in 816 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 3672 packets: 817, 818, 819, 820, 821, 822, 823, 824, 825, 826, 827, 828, 829, 830, 831, 832, 833, 834, ...
      1 in 5137 packets: 4489, 4490, 4491, 4492, 4493, 4494, 4495, 4496, 4497, 4498, 4499, 4500, 4501, 4502, 4503, 4504, 4505, 4506, ...
Number of generated alerts: 
      3 in 816 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 3672 packets: 817, 818, 819, 820, 821, 822, 823, 824, 825, 826, 827, 828, 829, 830, 831, 832, 833, 834, ...
      1 in 5137 packets: 4489, 4490, 4491, 4492, 4493, 4494, 4495, 4496, 4497, 4498, 4499, 4500, 4501, 4502, 4503, 4504, 4505, 4506, ...
Survey memory error: 
      0 in 9625 packets (all packets)
Event memory error: 
      0 in 9625 packets (all packets)
Configuration error: 
      0 in 9625 packets (all packets)
SDRAM CRC error: 
      0 in 9625 packets (all packets)
Radio gain: 
      0 in 9625 packets (all packets)
Radio bandwidth: 
      0 in 9625 packets (all packets)
Digital gain: 
      0 in 9625 packets (all packets)
Analog gain: 
      0 in 9625 packets (all packets)

EVENT DATA READER
-- File 20220608_153225_1229.dat, 47 EW pages found. Extra packets: 37.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 816 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 3672 packets: 817, 818, 819, 820, 821, 822, 823, 824, 825, 826, 827, 828, 829, 830, 831, 832, 833, 834, ...
      1 in 5100 packets: 4489, 4490, 4491, 4492, 4493, 4494, 4495, 4496, 4497, 4498, 4499, 4500, 4501, 4502, 4503, 4504, 4505, 4506, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 38.3271210 to 51.7968160
      Difference between the neighboring packets: 
      0.0000170 in 9585 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -9.6202150 in 1 packets: 816
      -3.9118870 in 1 packets: 4488
   Memory page number from 0 to 46
      Difference between the neighboring packets: 
      0 in 9541 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 46 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 25
      Difference between the neighboring packets: 
      0 in 9541 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 44 packets: 204, 408, 612, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
      -13 in 1 packets: 816
      -17 in 1 packets: 4488
   MIU event flag 
      1 in 9588 packets (all packets)
   Time triggered event flag 
      0 in 9588 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-06-08T15:33:04.074.496.631
   Min EW header abs. time found among all events            : 2022-06-08T15:33:04.129.985.000
   Max EW event (1st sample) abs. time found among all events: 2022-06-08T15:33:17.495.700.631
   Max EW header abs. time found among all events            : 2022-06-08T15:33:17.551.189.000
   TM values from -254 to 245 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 5100
    Page no. 1:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 25  ordered packets: 5100
   TM values from -246 to 231 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=36 C1MHz=0.129985
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 5099 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 38.3271210
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-06-08T15:33:04.129.985.000
   Event 1st sample abs. time : 2022-06-08T15:33:04.074.496.631)
   --- Rel. time correction for 1 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 36, 32bit 10MHz counter = 38.327121000 corrected by 40bit 10MHz counter of 38.252623200 obtained at 1Hz counter of 36. Result: 36.074498061
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              36.074497886
   Event data descriptor XO time (reconstructed seconds from XP start)    :              36.074498061
   Sampling frequency correction factor for this event:     0.999996500

 Event no. 2 total number of packets: 3672
    Page no. 1:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 18  ordered packets: 3672
   TM values from -254 to 245 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=39 C1MHz=0.979474
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 3671 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 42.1766010
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-06-08T15:33:07.979.474.000
   Event 1st sample abs. time : 2022-06-08T15:33:07.923.985.631)
   --- Rel. time correction for 1 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 39, 32bit 10MHz counter = 42.176601000 corrected by 40bit 10MHz counter of 41.252612800 obtained at 1Hz counter of 39. Result: 39.923991434
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              39.923991259
   Event data descriptor XO time (reconstructed seconds from XP start)    :              39.923991434
   Sampling frequency correction factor for this event:     0.999996500

 Event no. 3 total number of packets: 816
    Page no. 11:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 4  ordered packets: 816
   TM values from -210 to 206 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=49 C1MHz=0.551189
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 815 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 51.7829610
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-06-08T15:33:17.551.189.000
   Event 1st sample abs. time : 2022-06-08T15:33:17.495.700.631)
   --- Rel. time correction for 1 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 49, 32bit 10MHz counter = 51.782961000 corrected by 40bit 10MHz counter of 45.252598900 obtained at 1Hz counter of 43. Result: 49.530384956
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              49.530384781
   Event data descriptor XO time (reconstructed seconds from XP start)    :              49.530384956
   Sampling frequency correction factor for this event:     0.999996500
--
Extremes in separate events:

 - Event number: 1 from 2022-06-08T15:33:04.074.496.631 to 2022-06-08T15:33:04.161.196.934 (XO: 36.074497886 - 36.161198189)
   Min TM value:-246 at an offset of 12739.695 usec -> 2022-06-08T15:33:04.087.236.325 (XO: 36.087237580)
   Max TM value:231 at an offset of 52787.260 usec -> 2022-06-08T15:33:04.127.283.890 (XO: 36.127285145)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 72793.355 usec -> 2022-06-08T15:33:04.147.289.985 (XO: 36.147291241)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 72790.980 usec -> 2022-06-08T15:33:04.147.287.610 (XO: 36.147288866)
   Med: -4.00000   Ave: -4.4182119   Std: 8.0874805

 - Event number: 2 from 2022-06-08T15:33:07.923.985.631 to 2022-06-08T15:33:07.986.409.849 (XO: 39.923991259 - 39.986415477)
   Min TM value:-254 at an offset of 53639.563 usec -> 2022-06-08T15:33:07.977.625.193 (XO: 39.977630822)
   Max TM value:245 at an offset of 13365.359 usec -> 2022-06-08T15:33:07.937.350.990 (XO: 39.937356618)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 53641.813 usec -> 2022-06-08T15:33:07.977.627.443 (XO: 39.977633072)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 33576.080 usec -> 2022-06-08T15:33:07.957.561.711 (XO: 39.957567339)
   Med: -4.00000   Ave: -4.3768589   Std: 8.1547287

 - Event number: 3 from 2022-06-08T15:33:17.495.700.631 to 2022-06-08T15:33:17.509.572.679 (XO: 49.530384781 - 49.544256830)
   Min TM value:-210 at an offset of 8057.791 usec -> 2022-06-08T15:33:17.503.758.421 (XO: 49.538442572)
   Max TM value:206 at an offset of 8104.378 usec -> 2022-06-08T15:33:17.503.805.009 (XO: 49.538489160)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 8057.928 usec -> 2022-06-08T15:33:17.503.758.559 (XO: 49.538442710)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 8055.441 usec -> 2022-06-08T15:33:17.503.756.071 (XO: 49.538440222)
   Med: -4.00000   Ave: -4.3608122   Std: 7.8051504
