-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 15 17:01:45 2023 UTC
-- File 20220608_060906_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.284303 to 24.498324
-each separately from 0s to 24s and from 284303us to 498324us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.213443 in 1 packets: 18
MIU packet time from 2022-06-08T06:09:09.284.303.000 to 2022-06-08T06:09:33.498.324.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.213443 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5458063 to 25.5457220
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999157 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      00 -> 55.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220608_060906_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8301110 to 1.8304000
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.2133590  -- OK for SL --
   MIU time from 2022-06-08T06:09:09.284.303.000 to 2022-06-08T06:09:09.284.592.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.213443

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0437590 to 26.0440480
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-08T06:09:33.498.035.000 to 2022-06-08T06:09:33.498.324.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.830111000 corrected by 40bit 10MHz counter of 1.545806300 obtained at 1Hz counter of 0. Result: 0.284305695
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-08T06:09:09.284.162.725
   Max 1st sample MIU time among all SL packets: 2022-06-08T06:09:33.498.183.725
   Min 1st sample XO time among all SL packets: 0.284165420
   Max 1st sample XO time among all SL packets: 24.498187469
   TM values from -274 to 229 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-08T06:09:09.284.162.725 (XO: 0.284165420 - 0.284471421, t=43.6 degC)  Rank: 00068
   TM values: Max =   224  Min =  -239  Ave =    -4.56  Std =    27.68
   Min TM value:-239 at an offset of 117.900 usec -> 2022-06-08T06:09:09.284.280.625 (XO: 0.284283320)
   Max TM value:224 at an offset of 150.151 usec -> 2022-06-08T06:09:09.284.312.875 (XO: 0.284315570)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 120.075 usec -> 2022-06-08T06:09:09.284.282.800 (XO: 0.284285495)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 117.713 usec -> 2022-06-08T06:09:09.284.280.437 (XO: 0.284283132)

 - Block number: 2 starting at 2022-06-08T06:09:33.497.894.725 (XO: 24.497898468 - 24.498204469, t=43.6 degC)  Rank: 00066
   TM values: Max =   229  Min =  -274  Ave =    -4.55  Std =    36.21
   Min TM value:-274 at an offset of 129.775 usec -> 2022-06-08T06:09:33.498.024.500 (XO: 24.498028243)
   Max TM value:229 at an offset of 160.163 usec -> 2022-06-08T06:09:33.498.054.888 (XO: 24.498058631)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 32.900 usec -> 2022-06-08T06:09:33.497.927.625 (XO: 24.497931368)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 145.588 usec -> 2022-06-08T06:09:33.498.040.313 (XO: 24.498044056)
