-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 15 15:56:12 2023 UTC
-- File 20220607_180819_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841253us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 5, 8, 9, 11, 12, 15, 16
      0.833943 in 5 packets: 3, 6, 10, 14, 17
      0.833939 in 2 packets: 7, 13
MIU packet time from 2022-06-07T18:08:22.000.000.000 to 2022-06-07T18:08:36.177.027.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 5, 8, 9, 11, 12, 15, 16
      0.833943 in 5 packets: 3, 6, 10, 14, 17
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.3418718 to 15.3418233
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999966 in 5 packets: 2, 4, 8, 12, 16
      0.9999965 in 9 packets: 3, 5, 6, 9, 10, 11, 14, 15, 17
   XO PPM: 
      -3.4 in 5 packets: 2, 4, 8, 12, 16
      -3.5 in 9 packets: 3, 5, 6, 9, 10, 11, 14, 15, 17
Temperature FPGA: 
      FF -> 56.3 degC in 15 packets: 1, 2, 3, 5, 6, 7, 8, 9, 10, 11, 12, 15, 16, 17, 18
      FE -> 57.6 degC in 3 packets: 4, 13, 14
Temperature XO: 
      C5 -> 43.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220607_180819_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.3418719 to 15.5188519
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-07T18:08:22.000.000.000 to 2022-06-07T18:08:36.177.027.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 5, 8, 9, 11, 12, 15, 16
      0.833943 in 5 packets: 3, 6, 10, 14, 17
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.341871900 corrected by 40bit 10MHz counter of 1.341871800 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SR packets: 2022-06-07T18:08:21.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-06-07T18:08:36.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177029145
   TM values from -174 to 187 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-06-07T18:08:21.999.999.925 (XO: 0.000000025 - 0.000017025, t=43.6 degC)
   TM values: Max =   187  Min =  -174  Ave =    -4.55  Std =    65.27

 - Packet number: 2 starting at 2022-06-07T18:08:22.833.937.925 (XO: 0.833942860 - 0.833959860, t=43.6 degC)
   TM values: Max =   119  Min =  -137  Ave =    -4.36  Std =    47.07

 - Packet number: 3 starting at 2022-06-07T18:08:23.667.881.925 (XO: 1.667885763 - 1.667902763, t=43.6 degC)
   TM values: Max =   116  Min =  -136  Ave =    -4.67  Std =    46.57

 - Packet number: 4 starting at 2022-06-07T18:08:24.501.824.925 (XO: 2.501828631 - 2.501845631, t=43.6 degC)
   TM values: Max =   134  Min =  -163  Ave =    -4.39  Std =    53.03

 - Packet number: 5 starting at 2022-06-07T18:08:25.335.768.925 (XO: 3.335771500 - 3.335788500, t=43.6 degC)
   TM values: Max =   153  Min =  -151  Ave =    -4.59  Std =    54.82

 - Packet number: 6 starting at 2022-06-07T18:08:26.169.712.925 (XO: 4.169714419 - 4.169731419, t=43.6 degC)
   TM values: Max =   157  Min =  -160  Ave =    -4.33  Std =    61.32

 - Packet number: 7 starting at 2022-06-07T18:08:27.003.655.925 (XO: 5.003657337 - 5.003674337, t=43.6 degC)
   TM values: Max =   152  Min =  -170  Ave =    -4.54  Std =    58.27

 - Packet number: 8 starting at 2022-06-07T18:08:27.837.594.925 (XO: 5.837600173 - 5.837617173, t=43.6 degC)
   TM values: Max =   113  Min =  -112  Ave =    -4.32  Std =    44.10

 - Packet number: 9 starting at 2022-06-07T18:08:28.671.538.925 (XO: 6.671543075 - 6.671560075, t=43.6 degC)
   TM values: Max =   120  Min =  -133  Ave =    -4.36  Std =    54.34

 - Packet number: 10 starting at 2022-06-07T18:08:29.505.482.925 (XO: 7.505485994 - 7.505502994, t=43.6 degC)
   TM values: Max =   143  Min =  -142  Ave =    -4.61  Std =    58.80

 - Packet number: 11 starting at 2022-06-07T18:08:30.339.425.925 (XO: 8.339428913 - 8.339445913, t=43.6 degC)
   TM values: Max =   145  Min =  -169  Ave =    -4.55  Std =    60.47

 - Packet number: 12 starting at 2022-06-07T18:08:31.173.369.925 (XO: 9.173371814 - 9.173388815, t=43.6 degC)
   TM values: Max =   131  Min =  -114  Ave =    -4.45  Std =    51.98

 - Packet number: 13 starting at 2022-06-07T18:08:32.007.313.925 (XO: 10.007314651 - 10.007331651, t=43.6 degC)
   TM values: Max =   119  Min =  -134  Ave =    -4.60  Std =    42.12

 - Packet number: 14 starting at 2022-06-07T18:08:32.841.252.925 (XO: 10.841257569 - 10.841274569, t=43.6 degC)
   TM values: Max =   109  Min =  -121  Ave =    -4.22  Std =    38.81

 - Packet number: 15 starting at 2022-06-07T18:08:33.675.195.925 (XO: 11.675200488 - 11.675217488, t=43.6 degC)
   TM values: Max =    87  Min =   -97  Ave =    -4.52  Std =    33.56

 - Packet number: 16 starting at 2022-06-07T18:08:34.509.139.925 (XO: 12.509143356 - 12.509160356, t=43.6 degC)
   TM values: Max =    97  Min =   -96  Ave =    -4.29  Std =    33.98

 - Packet number: 17 starting at 2022-06-07T18:08:35.343.083.925 (XO: 13.343086226 - 13.343103226, t=43.6 degC)
   TM values: Max =    83  Min =  -105  Ave =    -4.63  Std =    32.16

 - Packet number: 18 starting at 2022-06-07T18:08:36.177.026.925 (XO: 14.177029145 - 14.177046145, t=43.6 degC)
   TM values: Max =    89  Min =  -102  Ave =    -4.24  Std =    36.78
