-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 15 14:42:24 2023 UTC
-- File 20220607_123444_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.129121 to 23.603017
-each separately from 7s to 23s and from 129121us to 603017us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.473318 in 1 packets: 18
MIU packet time from 2022-06-07T12:34:54.129.121.000 to 2022-06-07T12:35:10.603.017.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.473318 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7504484 to 24.7503933
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999449 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      00 -> 55.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220607_123444_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8795709 to 8.8798599
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4732630  -- OK for SL --
   MIU time from 2022-06-07T12:34:54.129.121.000 to 2022-06-07T12:34:54.129.410.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.473318

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.3531229 to 25.3534119
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-07T12:35:10.602.728.000 to 2022-06-07T12:35:10.603.017.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.879570900 corrected by 40bit 10MHz counter of 8.750448400 obtained at 1Hz counter of 7. Result: 7.129122952
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-07T12:34:54.128.980.725
   Max 1st sample MIU time among all SL packets: 2022-06-07T12:35:10.602.876.725
   Min 1st sample XO time among all SL packets: 7.128982676
   Max 1st sample XO time among all SL packets: 23.602880375
   TM values from -267 to 239 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-07T12:34:54.128.980.725 (XO: 7.128982676 - 7.129288678, t=43.6 degC)  Rank: 00081
   TM values: Max =   239  Min =  -267  Ave =    -4.39  Std =    31.47
   Min TM value:-267 at an offset of 159.263 usec -> 2022-06-07T12:34:54.129.139.988 (XO: 7.129141939)
   Max TM value:239 at an offset of 161.651 usec -> 2022-06-07T12:34:54.129.142.375 (XO: 7.129144327)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 161.526 usec -> 2022-06-07T12:34:54.129.142.250 (XO: 7.129144202)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 159.163 usec -> 2022-06-07T12:34:54.129.139.888 (XO: 7.129141839)

 - Block number: 2 starting at 2022-06-07T12:35:10.602.587.725 (XO: 23.602591374 - 23.602897375, t=43.6 degC)  Rank: 00077
   TM values: Max =   159  Min =  -180  Ave =    -4.44  Std =    22.23
   Min TM value:-180 at an offset of 103.500 usec -> 2022-06-07T12:35:10.602.691.225 (XO: 23.602694874)
   Max TM value:159 at an offset of 161.576 usec -> 2022-06-07T12:35:10.602.749.300 (XO: 23.602752949)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 161.513 usec -> 2022-06-07T12:35:10.602.749.238 (XO: 23.602752887)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 132.025 usec -> 2022-06-07T12:35:10.602.719.750 (XO: 23.602723399)
