-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 13 17:39:27 2023 UTC
-- File 20220603_115528_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.017117 to 61.660234
-each separately from 5s to 61s and from 17117us to 660234us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.570908 in 1 packets: 18
      8.789981 in 1 packets: 36
      11.160324 in 1 packets: 54
      13.120459 in 1 packets: 72
MIU packet time from 2022-06-03T11:55:36.017.117.000 to 2022-06-03T11:56:32.660.234.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.570908 in 1 packets: 18
      8.789981 in 1 packets: 36
      11.160324 in 1 packets: 54
      13.120459 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8795601 to 62.8793713
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999226 in 1 packets: 18
      8.9999698 in 1 packets: 36
      10.9999631 in 1 packets: 54
      12.9999557 in 1 packets: 72
Temperature FPGA: 
      00 -> 55.0 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FF -> 56.3 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C5 -> 43.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220603_115528_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.8966781 to 6.8969671
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5708310  -- OK for SL --
   MIU time from 2022-06-03T11:55:36.017.117.000 to 2022-06-03T11:55:36.017.406.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.570908

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.4677981 to 30.4680871
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7899510  -- OK for SL --
   MIU time from 2022-06-03T11:55:59.588.314.000 to 2022-06-03T11:55:59.588.603.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.789981

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.2580381 to 39.2583271
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1602870  -- OK for SL --
   MIU time from 2022-06-03T11:56:08.378.584.000 to 2022-06-03T11:56:08.378.873.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.160324

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.4186141 to 50.4189031
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1204150  -- OK for SL --
   MIU time from 2022-06-03T11:56:19.539.197.000 to 2022-06-03T11:56:19.539.486.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.120459

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.5393181 to 63.5396071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-03T11:56:32.659.945.000 to 2022-06-03T11:56:32.660.234.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.896678100 corrected by 40bit 10MHz counter of 6.879560100 obtained at 1Hz counter of 5. Result: 5.017118058
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-03T11:55:36.016.976.725
   Max 1st sample MIU time among all SL packets: 2022-06-03T11:56:32.660.093.725
   Min 1st sample XO time among all SL packets: 5.016977783
   Max 1st sample XO time among all SL packets: 61.660097769
   TM values from -271 to 244 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-03T11:55:36.016.976.725 (XO: 5.016977783 - 5.017283784, t=43.6 degC)  Rank: 00085
   TM values: Max =   236  Min =  -253  Ave =    -4.45  Std =    28.90
   Min TM value:-253 at an offset of 145.225 usec -> 2022-06-03T11:55:36.017.121.950 (XO: 5.017123008)
   Max TM value:236 at an offset of 75.125 usec -> 2022-06-03T11:55:36.017.051.850 (XO: 5.017052908)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 74.988 usec -> 2022-06-03T11:55:36.017.051.712 (XO: 5.017052770)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 72.575 usec -> 2022-06-03T11:55:36.017.049.300 (XO: 5.017050358)

 - Block number: 2 starting at 2022-06-03T11:55:59.588.173.725 (XO: 28.588177066 - 28.588483067, t=43.6 degC)  Rank: 00079
   TM values: Max =   244  Min =  -271  Ave =    -4.44  Std =    17.54
   Min TM value:-271 at an offset of 119.800 usec -> 2022-06-03T11:55:59.588.293.525 (XO: 28.588296866)
   Max TM value:244 at an offset of 117.850 usec -> 2022-06-03T11:55:59.588.291.575 (XO: 28.588294916)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 122.000 usec -> 2022-06-03T11:55:59.588.295.725 (XO: 28.588299066)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 119.688 usec -> 2022-06-03T11:55:59.588.293.412 (XO: 28.588296754)

 - Block number: 3 starting at 2022-06-03T11:56:08.378.443.725 (XO: 37.378446612 - 37.378752613, t=43.6 degC)  Rank: 00078
   TM values: Max =   230  Min =  -248  Ave =    -4.45  Std =    30.79
   Min TM value:-248 at an offset of 116.988 usec -> 2022-06-03T11:56:08.378.560.712 (XO: 37.378563600)
   Max TM value:230 at an offset of 119.363 usec -> 2022-06-03T11:56:08.378.563.087 (XO: 37.378565975)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 119.213 usec -> 2022-06-03T11:56:08.378.562.937 (XO: 37.378565825)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 119.025 usec -> 2022-06-03T11:56:08.378.562.750 (XO: 37.378565637)

 - Block number: 4 starting at 2022-06-03T11:56:19.539.056.725 (XO: 48.539060004 - 48.539366005, t=43.6 degC)  Rank: 00084
   TM values: Max =   239  Min =  -246  Ave =    -4.35  Std =    27.38
   Min TM value:-246 at an offset of 76.275 usec -> 2022-06-03T11:56:19.539.133.000 (XO: 48.539136279)
   Max TM value:239 at an offset of 76.513 usec -> 2022-06-03T11:56:19.539.133.237 (XO: 48.539136517)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 76.388 usec -> 2022-06-03T11:56:19.539.133.112 (XO: 48.539136392)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 73.988 usec -> 2022-06-03T11:56:19.539.130.712 (XO: 48.539133992)

 - Block number: 5 starting at 2022-06-03T11:56:32.659.804.725 (XO: 61.659808768 - 61.660114769, t=43.6 degC)  Rank: 00075
   TM values: Max =   216  Min =  -239  Ave =    -4.46  Std =    23.15
   Min TM value:-239 at an offset of 150.438 usec -> 2022-06-03T11:56:32.659.955.163 (XO: 61.659959206)
   Max TM value:216 at an offset of 150.688 usec -> 2022-06-03T11:56:32.659.955.413 (XO: 61.659959456)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 150.563 usec -> 2022-06-03T11:56:32.659.955.288 (XO: 61.659959331)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 118.975 usec -> 2022-06-03T11:56:32.659.923.700 (XO: 61.659927744)
