-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 13 09:21:00 2023 UTC
-- File 20220601_152615_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.280967 to 24.423245
-each separately from 5s to 24s and from 280967us to 423245us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.141700 in 1 packets: 18
MIU packet time from 2022-06-01T15:26:23.280.967.000 to 2022-06-01T15:26:42.423.245.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.141700 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.6587381 to 25.6586698
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999317 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220601_152615_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9397069 to 6.9399959
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1416310  -- OK for SL --
   MIU time from 2022-06-01T15:26:23.280.967.000 to 2022-06-01T15:26:23.281.256.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.141700

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0816269 to 26.0819159
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-01T15:26:42.422.956.000 to 2022-06-01T15:26:42.423.245.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.939706900 corrected by 40bit 10MHz counter of 6.658738100 obtained at 1Hz counter of 5. Result: 5.280969783
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-01T15:26:23.280.826.725
   Max 1st sample MIU time among all SL packets: 2022-06-01T15:26:42.423.104.725
   Min 1st sample XO time among all SL packets: 5.280829508
   Max 1st sample XO time among all SL packets: 24.423107348
   TM values from -248 to 215 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-01T15:26:23.280.826.725 (XO: 5.280829508 - 5.281135509, t=42.0 degC)  Rank: 00083
   TM values: Max =   213  Min =  -248  Ave =    -4.32  Std =    29.61
   Min TM value:-248 at an offset of 147.251 usec -> 2022-06-01T15:26:23.280.973.975 (XO: 5.280976758)
   Max TM value:213 at an offset of 149.663 usec -> 2022-06-01T15:26:23.280.976.388 (XO: 5.280979171)
   Max dTM/dt [1/us]:2.41e+03 at an offset of 149.538 usec -> 2022-06-01T15:26:23.280.976.263 (XO: 5.280979046)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 124.200 usec -> 2022-06-01T15:26:23.280.950.925 (XO: 5.280953708)

 - Block number: 2 starting at 2022-06-01T15:26:42.422.815.725 (XO: 24.422818347 - 24.423124348, t=42.0 degC)  Rank: 00079
   TM values: Max =   215  Min =  -226  Ave =    -4.37  Std =    27.86
   Min TM value:-226 at an offset of 122.613 usec -> 2022-06-01T15:26:42.422.938.337 (XO: 24.422940960)
   Max TM value:215 at an offset of 215.576 usec -> 2022-06-01T15:26:42.423.031.300 (XO: 24.423033923)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 122.725 usec -> 2022-06-01T15:26:42.422.938.450 (XO: 24.422941073)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 215.688 usec -> 2022-06-01T15:26:42.423.031.413 (XO: 24.423034035)
