-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 13 09:10:39 2023 UTC
-- File 20220601_151332_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.167993 to 20.129347
-each separately from 4s to 20s and from 129058us to 168282us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.960776 in 1 packets: 18
MIU packet time from 2022-06-01T15:13:39.167.993.000 to 2022-06-01T15:13:55.129.347.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.960776 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.2375184 to 22.2374620
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999436 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C4 -> 42.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220601_151332_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4055130 to 6.4058020
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9607190  -- OK for SL --
   MIU time from 2022-06-01T15:13:39.167.993.000 to 2022-06-01T15:13:39.168.282.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.960776

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.3665210 to 22.3668100
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-01T15:13:55.129.058.000 to 2022-06-01T15:13:55.129.347.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.405513000 corrected by 40bit 10MHz counter of 6.237518400 obtained at 1Hz counter of 4. Result: 4.167995188
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-06-01T15:13:39.167.852.725
   Max 1st sample MIU time among all SL packets: 2022-06-01T15:13:55.129.206.725
   Min 1st sample XO time among all SL packets: 4.167854912
   Max 1st sample XO time among all SL packets: 20.129208190
   TM values from -274 to 210 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-01T15:13:39.167.852.725 (XO: 4.167854912 - 4.168160914, t=43.6 degC)  Rank: 00084
   TM values: Max =   200  Min =  -251  Ave =    -4.34  Std =    26.50
   Min TM value:-251 at an offset of 122.075 usec -> 2022-06-01T15:13:39.167.974.800 (XO: 4.167976988)
   Max TM value:200 at an offset of 124.450 usec -> 2022-06-01T15:13:39.167.977.175 (XO: 4.167979363)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 124.325 usec -> 2022-06-01T15:13:39.167.977.050 (XO: 4.167979238)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 126.813 usec -> 2022-06-01T15:13:39.167.979.537 (XO: 4.167981725)

 - Block number: 2 starting at 2022-06-01T15:13:55.128.917.725 (XO: 20.128919189 - 20.129225190, t=42.0 degC)  Rank: 00077
   TM values: Max =   210  Min =  -274  Ave =    -4.47  Std =    19.57
   Min TM value:-274 at an offset of 109.300 usec -> 2022-06-01T15:13:55.129.027.025 (XO: 20.129028490)
   Max TM value:210 at an offset of 111.650 usec -> 2022-06-01T15:13:55.129.029.375 (XO: 20.129030840)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 111.525 usec -> 2022-06-01T15:13:55.129.029.250 (XO: 20.129030715)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 109.163 usec -> 2022-06-01T15:13:55.129.026.887 (XO: 20.129028352)
