-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 10 07:09:40 2023 UTC
-- File 20220530_152514_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.057069 to 24.378359
-each separately from 7s to 24s and from 57069us to 378359us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.320712 in 1 packets: 18
MIU packet time from 2022-05-30T15:25:24.057.069.000 to 2022-05-30T15:25:41.378.359.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.320712 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7550672 to 25.7550097
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999425 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220530_152514_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8121376 to 8.8124266
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3206550  -- OK for SL --
   MIU time from 2022-05-30T15:25:24.057.069.000 to 2022-05-30T15:25:24.057.358.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.320712

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1330816 to 26.1333706
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-30T15:25:41.378.070.000 to 2022-05-30T15:25:41.378.359.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.812137600 corrected by 40bit 10MHz counter of 8.755067200 obtained at 1Hz counter of 7. Result: 7.057070594
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-05-30T15:25:24.056.928.725
   Max 1st sample MIU time among all SL packets: 2022-05-30T15:25:41.378.218.725
   Min 1st sample XO time among all SL packets: 7.056930319
   Max 1st sample XO time among all SL packets: 24.378221911
   TM values from -265 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-30T15:25:24.056.928.725 (XO: 7.056930319 - 7.057236320, t=42.0 degC)  Rank: 00090
   TM values: Max =   222  Min =  -255  Ave =    -4.45  Std =    27.07
   Min TM value:-255 at an offset of 121.613 usec -> 2022-05-30T15:25:24.057.050.337 (XO: 7.057051931)
   Max TM value:222 at an offset of 123.938 usec -> 2022-05-30T15:25:24.057.052.662 (XO: 7.057054256)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 153.663 usec -> 2022-05-30T15:25:24.057.082.388 (XO: 7.057083982)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 121.413 usec -> 2022-05-30T15:25:24.057.050.137 (XO: 7.057051731)

 - Block number: 2 starting at 2022-05-30T15:25:41.377.929.725 (XO: 24.377932910 - 24.378238911, t=42.0 degC)  Rank: 00083
   TM values: Max =   214  Min =  -265  Ave =    -4.52  Std =    31.35
   Min TM value:-265 at an offset of 207.476 usec -> 2022-05-30T15:25:41.378.137.200 (XO: 24.378140386)
   Max TM value:214 at an offset of 124.225 usec -> 2022-05-30T15:25:41.378.053.950 (XO: 24.378057135)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 124.113 usec -> 2022-05-30T15:25:41.378.053.837 (XO: 24.378057023)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 212.176 usec -> 2022-05-30T15:25:41.378.141.900 (XO: 24.378145086)
