-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 08:15:21 2023 UTC
-- File 20220530_124436_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.523550 to 28.823614
-each separately from 14s to 28s and from 523550us to 823614us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.299486 in 1 packets: 18
MIU packet time from 2022-05-30T12:44:53.523.550.000 to 2022-05-30T12:45:07.823.614.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.299486 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7600666 to 29.7600195
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999529 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220530_124436_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2836185 to 16.2839075
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2994390  -- OK for SL --
   MIU time from 2022-05-30T12:44:53.523.550.000 to 2022-05-30T12:44:53.523.839.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.299486

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.5833465 to 30.5836355
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-30T12:45:07.823.325.000 to 2022-05-30T12:45:07.823.614.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.283618500 corrected by 40bit 10MHz counter of 15.760066600 obtained at 1Hz counter of 14. Result: 14.523553680
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-05-30T12:44:53.523.409.725
   Max 1st sample MIU time among all SL packets: 2022-05-30T12:45:07.823.473.725
   Min 1st sample XO time among all SL packets: 14.523413405
   Max 1st sample XO time among all SL packets: 28.823478442
   TM values from -269 to 231 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-30T12:44:53.523.409.725 (XO: 14.523413405 - 14.523719406, t=42.0 degC)  Rank: 00078
   TM values: Max =   208  Min =  -256  Ave =    -4.36  Std =    31.50
   Min TM value:-256 at an offset of 244.163 usec -> 2022-05-30T12:44:53.523.653.888 (XO: 14.523657568)
   Max TM value:208 at an offset of 176.088 usec -> 2022-05-30T12:44:53.523.585.813 (XO: 14.523589493)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 259.401 usec -> 2022-05-30T12:44:53.523.669.125 (XO: 14.523672805)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 176.188 usec -> 2022-05-30T12:44:53.523.585.913 (XO: 14.523589593)

 - Block number: 2 starting at 2022-05-30T12:45:07.823.184.725 (XO: 28.823189442 - 28.823495443, t=42.0 degC)  Rank: 00080
   TM values: Max =   231  Min =  -269  Ave =    -4.38  Std =    25.50
   Min TM value:-269 at an offset of 128.538 usec -> 2022-05-30T12:45:07.823.313.262 (XO: 28.823317979)
   Max TM value:231 at an offset of 128.263 usec -> 2022-05-30T12:45:07.823.312.987 (XO: 28.823317704)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 130.775 usec -> 2022-05-30T12:45:07.823.315.500 (XO: 28.823320217)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 128.350 usec -> 2022-05-30T12:45:07.823.313.075 (XO: 28.823317792)
