-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 03:35:47 2023 UTC
-- File 20220528_221218_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.960178 to 90.787781
-each separately from 8s to 90s and from 90225us to 960467us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.163736 in 1 packets: 18
      12.466646 in 1 packets: 36
      18.172763 in 1 packets: 54
      10.675284 in 1 packets: 72
      22.650173 in 1 packets: 90
      6.696978 in 1 packets: 108
MIU packet time from 2022-05-28T22:12:29.960.178.000 to 2022-05-28T22:13:51.787.781.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.163736 in 1 packets: 18
      12.466646 in 1 packets: 36
      18.172763 in 1 packets: 54
      10.675284 in 1 packets: 72
      22.650173 in 1 packets: 90
      6.696978 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.2612075 to 92.2609333
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999599 in 2 packets: 18, 36
      17.9999398 in 1 packets: 54
      10.9999633 in 1 packets: 72
      22.9999230 in 1 packets: 90
      5.9999799 in 1 packets: 108
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 90 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C5 -> 43.6 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 90 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 90 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220528_221218_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.2213874 to 11.2216764
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1636950  -- OK for SL --
   MIU time from 2022-05-28T22:12:29.960.178.000 to 2022-05-28T22:12:29.960.467.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.163736

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      101 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.3853714 to 22.3856604
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4666070  -- OK for SL --
   MIU time from 2022-05-28T22:12:41.124.203.000 to 2022-05-28T22:12:41.124.492.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.466646

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8522674 to 34.8525564
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1727030  -- OK for SL --
   MIU time from 2022-05-28T22:12:53.591.138.000 to 2022-05-28T22:12:53.591.427.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.172763

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.0252594 to 53.0255484
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6752470  -- OK for SL --
   MIU time from 2022-05-28T22:13:11.764.190.000 to 2022-05-28T22:13:11.764.479.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.675284

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.7007954 to 63.7010844
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6500950  -- OK for SL --
   MIU time from 2022-05-28T22:13:22.439.763.000 to 2022-05-28T22:13:22.440.052.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.650173

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 86.3511794 to 86.3514684
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6969590  -- OK for SL --
   MIU time from 2022-05-28T22:13:45.090.225.000 to 2022-05-28T22:13:45.090.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.696978

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 93.0484274 to 93.0487164
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-28T22:13:51.787.492.000 to 2022-05-28T22:13:51.787.781.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 11.221387400 corrected by 40bit 10MHz counter of 10.261207500 obtained at 1Hz counter of 8. Result: 8.960183069
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-05-28T22:12:29.960.037.725
   Max 1st sample MIU time among all SL packets: 2022-05-28T22:13:51.787.640.725
   Min 1st sample XO time among all SL packets: 8.960042793
   Max 1st sample XO time among all SL packets: 90.787645503
   TM values from -559 to 275 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-28T22:12:29.960.037.725 (XO: 8.960042793 - 8.960348794, t=43.6 degC)  Rank: 00069
   TM values: Max =   211  Min =  -233  Ave =    -4.28  Std =    33.34
   Min TM value:-233 at an offset of 127.575 usec -> 2022-05-28T22:12:29.960.165.300 (XO: 8.960170369)
   Max TM value:211 at an offset of 213.988 usec -> 2022-05-28T22:12:29.960.251.713 (XO: 8.960256781)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 127.713 usec -> 2022-05-28T22:12:29.960.165.437 (XO: 8.960170506)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 125.263 usec -> 2022-05-28T22:12:29.960.162.987 (XO: 8.960168056)

 - Block number: 2 starting at 2022-05-28T22:12:41.124.062.725 (XO: 20.124064134 - 20.124370135, t=43.6 degC)  Rank: 00101
   TM values: Max =   275  Min =  -559  Ave =    -4.25  Std =    33.11
   Min TM value:-559 at an offset of 125.263 usec -> 2022-05-28T22:12:41.124.187.987 (XO: 20.124189397)
   Max TM value:275 at an offset of 129.400 usec -> 2022-05-28T22:12:41.124.192.125 (XO: 20.124193535)
   Max dTM/dt [1/us]:4.01e+03 at an offset of 125.313 usec -> 2022-05-28T22:12:41.124.188.037 (XO: 20.124189447)
   Min dTM/dt [1/us]:-3.97e+03 at an offset of 125.225 usec -> 2022-05-28T22:12:41.124.187.950 (XO: 20.124189360)

 - Block number: 3 starting at 2022-05-28T22:12:53.590.997.725 (XO: 32.591001834 - 32.591307835, t=43.6 degC)  Rank: 00067
   TM values: Max =   207  Min =  -228  Ave =    -4.27  Std =    28.41
   Min TM value:-228 at an offset of 171.488 usec -> 2022-05-28T22:12:53.591.169.213 (XO: 32.591173322)
   Max TM value:207 at an offset of 114.800 usec -> 2022-05-28T22:12:53.591.112.525 (XO: 32.591116635)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 173.776 usec -> 2022-05-28T22:12:53.591.171.500 (XO: 32.591175610)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 144.163 usec -> 2022-05-28T22:12:53.591.141.887 (XO: 32.591145997)

 - Block number: 4 starting at 2022-05-28T22:13:11.764.049.725 (XO: 50.764054546 - 50.764360547, t=43.6 degC)  Rank: 00071
   TM values: Max =   194  Min =  -283  Ave =    -4.42  Std =    24.47
   Min TM value:-283 at an offset of 125.100 usec -> 2022-05-28T22:13:11.764.174.825 (XO: 50.764179647)
   Max TM value:194 at an offset of 127.375 usec -> 2022-05-28T22:13:11.764.177.100 (XO: 50.764181922)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 125.150 usec -> 2022-05-28T22:13:11.764.174.875 (XO: 50.764179697)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 124.950 usec -> 2022-05-28T22:13:11.764.174.675 (XO: 50.764179497)

 - Block number: 5 starting at 2022-05-28T22:13:22.439.622.725 (XO: 61.439626176 - 61.439932177, t=43.6 degC)  Rank: 00072
   TM values: Max =   193  Min =  -212  Ave =    -4.50  Std =    24.95
   Min TM value:-212 at an offset of 151.388 usec -> 2022-05-28T22:13:22.439.774.112 (XO: 61.439777564)
   Max TM value:193 at an offset of 126.788 usec -> 2022-05-28T22:13:22.439.749.512 (XO: 61.439752964)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 206.726 usec -> 2022-05-28T22:13:22.439.829.450 (XO: 61.439832901)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 126.913 usec -> 2022-05-28T22:13:22.439.749.637 (XO: 61.439753089)

 - Block number: 6 starting at 2022-05-28T22:13:45.090.084.725 (XO: 84.090086022 - 84.090392023, t=43.6 degC)  Rank: 00069
   TM values: Max =   206  Min =  -239  Ave =    -4.45  Std =    30.48
   Min TM value:-239 at an offset of 199.163 usec -> 2022-05-28T22:13:45.090.283.888 (XO: 84.090285185)
   Max TM value:206 at an offset of 72.100 usec -> 2022-05-28T22:13:45.090.156.825 (XO: 84.090158123)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 71.975 usec -> 2022-05-28T22:13:45.090.156.700 (XO: 84.090157998)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 199.063 usec -> 2022-05-28T22:13:45.090.283.788 (XO: 84.090285085)

 - Block number: 7 starting at 2022-05-28T22:13:51.787.351.725 (XO: 90.787356502 - 90.787662503, t=43.6 degC)  Rank: 00069
   TM values: Max =   181  Min =  -189  Ave =    -4.44  Std =    20.46
   Min TM value:-189 at an offset of 124.038 usec -> 2022-05-28T22:13:51.787.475.762 (XO: 90.787480540)
   Max TM value:181 at an offset of 126.600 usec -> 2022-05-28T22:13:51.787.478.325 (XO: 90.787483102)
   Max dTM/dt [1/us]:1.15e+03 at an offset of 126.500 usec -> 2022-05-28T22:13:51.787.478.225 (XO: 90.787483002)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 126.725 usec -> 2022-05-28T22:13:51.787.478.450 (XO: 90.787483227)
