-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 03:02:47 2023 UTC
-- File 20220528_211345_1229.dat, 1496  packets found.
APID:
      1229 (EW) in 1496 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 1496 packets (all packets)
Total number of packets: 
      36720 in 1496 packets (all packets)
Packet No from 1 to 1496
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 105.812158 to 105.812158
-each separately from 105s to 105s and from 812158us to 812158us.
   Differences in neigboring packets: 
      0.000000 in 1495 packets (all packets)
MIU packet time from 2022-05-28T21:15:33.812.158.000 to 2022-05-28T21:15:33.812.158.000
   Differences in neigboring packets: 
      0.000000 in 1495 packets (all packets)
MIU orbit number: 
      60000 in 1496 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 1496 packets (all packets)
MIU orbit information / number of events: 
      0 in 1496 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 1496 packets (all packets)
Instrument anomalies: 
      0 (OK) in 1496 packets (all packets)
FPGA configuration version: 
      211 in 1496 packets (all packets)
10 MHz counter at the last 1Hz pulse from 106.7667408 to 106.7667408
   Differences in neigboring packets: 
      0.0000000 in 1495 packets (all packets)
Temperature FPGA: 
      00 -> 55.0 degC in 1496 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 1496 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 1496 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 1496 packets (all packets)
Number of external events: 
      1 in 1496 packets (all packets)
Number of generated alerts: 
      1 in 1496 packets (all packets)
Survey memory error: 
      0 in 1496 packets (all packets)
Event memory error: 
      0 in 1496 packets (all packets)
Configuration error: 
      0 in 1496 packets (all packets)
SDRAM CRC error: 
      0 in 1496 packets (all packets)
Radio gain: 
      0 in 1496 packets (all packets)
Radio bandwidth: 
      0 in 1496 packets (all packets)
Digital gain: 
      0 in 1496 packets (all packets)
Analog gain: 
      0 in 1496 packets (all packets)

EVENT DATA READER
-- File 20220528_211345_1229.dat, 7 EW pages found. Extra packets: 68.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 1
      1 in 1428 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 107.5823690 to 107.6066280
      Difference between the neighboring packets: 
      0.0000170 in 1427 packets (all packets)
   Memory page number from 0 to 6
      Difference between the neighboring packets: 
      0 in 1421 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 6 packets: 204, 408, 612, 816, 1020, 1224
   Event page number from 18 to 24
      Difference between the neighboring packets: 
      0 in 1421 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 6 packets: 204, 408, 612, 816, 1020, 1224
   MIU event flag 
      1 in 1428 packets (all packets)
   Time triggered event flag 
      0 in 1428 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-05-28T21:15:33.756.669.637
   Min EW header abs. time found among all events            : 2022-05-28T21:15:33.812.158.000
   Max EW event (1st sample) abs. time found among all events: 2022-05-28T21:15:33.756.669.637
   Max EW header abs. time found among all events            : 2022-05-28T21:15:33.812.158.000
   TM values from -294 to 211 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 1428
    Page no. 18:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 7  ordered packets: 1428
   TM values from -294 to 211 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=105 C1MHz=0.812158
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1427 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 107.5823690
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-28T21:15:33.812.158.000
   Event 1st sample abs. time : 2022-05-28T21:15:33.756.669.637)
   --- Rel. time correction for 1 points based on 134 XO calibration points at 1Hz counter from 0 to 133
   --- First point: 1Hz counter = 105, 32bit 10MHz counter = 107.582369000 corrected by 40bit 10MHz counter of 106.766740800 obtained at 1Hz counter of 105. Result: 105.815630973
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             105.815630798
   Event data descriptor XO time (reconstructed seconds from XP start)    :             105.815630973
   Sampling frequency correction factor for this event:     0.999996600
--
Extremes in separate events:

 - Event number: 1 from 2022-05-28T21:15:33.756.669.637 to 2022-05-28T21:15:33.780.945.719 (XO: 105.815630798 - 105.839906881)
   Min TM value:-294 at an offset of 3669.050 usec -> 2022-05-28T21:15:33.760.338.686 (XO: 105.819299848)
   Max TM value:211 at an offset of 21999.525 usec -> 2022-05-28T21:15:33.778.669.161 (XO: 105.837630323)
   Max dTM/dt [1/us]:2.92e+03 at an offset of 9640.670 usec -> 2022-05-28T21:15:33.766.310.307 (XO: 105.825271468)
   Min dTM/dt [1/us]:-2.94e+03 at an offset of 7647.476 usec -> 2022-05-28T21:15:33.764.317.113 (XO: 105.823278274)
   Med: -4.00000   Ave: -4.3110037   Std: 13.807601
