-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 00:54:05 2023 UTC
-- File 20220528_123501_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.371040 to 17.235696
-each separately from 3s to 17s and from 235407us to 371329us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.864078 in 1 packets: 18
MIU packet time from 2022-05-28T12:35:07.371.040.000 to 2022-05-28T12:35:21.235.696.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.864078 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.2348672 to 19.2348199
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999527 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220528_123501_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      139 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.6059085 to 5.6061975
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8640310  -- OK for SL --
   MIU time from 2022-05-28T12:35:07.371.040.000 to 2022-05-28T12:35:07.371.329.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.864078

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4702285 to 19.4705175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-28T12:35:21.235.407.000 to 2022-05-28T12:35:21.235.696.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.605908500 corrected by 40bit 10MHz counter of 5.234867200 obtained at 1Hz counter of 3. Result: 3.371042562
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-05-28T12:35:07.370.899.725
   Max 1st sample MIU time among all SL packets: 2022-05-28T12:35:21.235.555.725
   Min 1st sample XO time among all SL packets: 3.370902286
   Max 1st sample XO time among all SL packets: 17.235558126
   TM values from -913 to 259 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-28T12:35:07.370.899.725 (XO: 3.370902286 - 3.371208287, t=43.6 degC)  Rank: 00139
   TM values: Max =   259  Min =  -913  Ave =    -4.56  Std =    35.84
   Min TM value:-913 at an offset of 111.038 usec -> 2022-05-28T12:35:07.371.010.762 (XO: 3.371013324)
   Max TM value:259 at an offset of 112.288 usec -> 2022-05-28T12:35:07.371.012.012 (XO: 3.371014574)
   Max dTM/dt [1/us]:3.30e+03 at an offset of 111.363 usec -> 2022-05-28T12:35:07.371.011.087 (XO: 3.371013649)
   Min dTM/dt [1/us]:-4.14e+03 at an offset of 110.925 usec -> 2022-05-28T12:35:07.371.010.650 (XO: 3.371013211)

 - Block number: 2 starting at 2022-05-28T12:35:21.235.266.725 (XO: 17.235269125 - 17.235575126, t=43.6 degC)  Rank: 00073
   TM values: Max =   224  Min =  -224  Ave =    -4.46  Std =    27.47
   Min TM value:-224 at an offset of 179.326 usec -> 2022-05-28T12:35:21.235.446.050 (XO: 17.235448451)
   Max TM value:224 at an offset of 123.113 usec -> 2022-05-28T12:35:21.235.389.837 (XO: 17.235392238)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 181.501 usec -> 2022-05-28T12:35:21.235.448.225 (XO: 17.235450626)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 120.638 usec -> 2022-05-28T12:35:21.235.387.362 (XO: 17.235389763)
