-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 00:40:37 2023 UTC
-- File 20220528_120847_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.593648 to 38.175440
-each separately from 5s to 38s and from 175151us to 593937us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.787928 in 1 packets: 18
      20.792997 in 1 packets: 36
MIU packet time from 2022-05-28T12:08:55.593.648.000 to 2022-05-28T12:09:28.175.440.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.787928 in 1 packets: 18
      20.792997 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7647921 to 39.7646813
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999597 in 1 packets: 18
      20.9999295 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C4 -> 42.0 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220528_120847_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3584419 to 7.3587309
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7878870  -- OK for SL --
   MIU time from 2022-05-28T12:08:55.593.648.000 to 2022-05-28T12:08:55.593.937.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.787928

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1466179 to 19.1469069
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7929270  -- OK for SL --
   MIU time from 2022-05-28T12:09:07.381.865.000 to 2022-05-28T12:09:07.382.154.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.792997

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.9398339 to 39.9401229
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-28T12:09:28.175.151.000 to 2022-05-28T12:09:28.175.440.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.358441900 corrected by 40bit 10MHz counter of 6.764792100 obtained at 1Hz counter of 5. Result: 5.593651818
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-05-28T12:08:55.593.507.725
   Max 1st sample MIU time among all SL packets: 2022-05-28T12:09:28.175.299.725
   Min 1st sample XO time among all SL packets: 5.593511543
   Max 1st sample XO time among all SL packets: 38.175301921
   TM values from -281 to 228 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-28T12:08:55.593.507.725 (XO: 5.593511543 - 5.593817544, t=43.6 degC)  Rank: 00080
   TM values: Max =   187  Min =  -231  Ave =    -4.45  Std =    23.58
   Min TM value:-231 at an offset of 126.750 usec -> 2022-05-28T12:08:55.593.634.475 (XO: 5.593638293)
   Max TM value:187 at an offset of 156.726 usec -> 2022-05-28T12:08:55.593.664.450 (XO: 5.593668268)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 126.863 usec -> 2022-05-28T12:08:55.593.634.587 (XO: 5.593638406)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 159.088 usec -> 2022-05-28T12:08:55.593.666.813 (XO: 5.593670631)

 - Block number: 2 starting at 2022-05-28T12:09:07.381.724.725 (XO: 17.381727085 - 17.382033086, t=43.6 degC)  Rank: 00084
   TM values: Max =   228  Min =  -281  Ave =    -4.35  Std =    31.37
   Min TM value:-281 at an offset of 157.826 usec -> 2022-05-28T12:09:07.381.882.550 (XO: 17.381884910)
   Max TM value:228 at an offset of 128.238 usec -> 2022-05-28T12:09:07.381.852.962 (XO: 17.381855323)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 157.888 usec -> 2022-05-28T12:09:07.381.882.613 (XO: 17.381884973)
   Min dTM/dt [1/us]:-2.72e+03 at an offset of 203.813 usec -> 2022-05-28T12:09:07.381.928.538 (XO: 17.381930898)

 - Block number: 3 starting at 2022-05-28T12:09:28.175.010.725 (XO: 38.175012920 - 38.175318921, t=42.0 degC)  Rank: 00081
   TM values: Max =   221  Min =  -254  Ave =    -4.36  Std =    29.24
   Min TM value:-254 at an offset of 141.500 usec -> 2022-05-28T12:09:28.175.152.225 (XO: 38.175154421)
   Max TM value:221 at an offset of 143.850 usec -> 2022-05-28T12:09:28.175.154.575 (XO: 38.175156771)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 143.700 usec -> 2022-05-28T12:09:28.175.154.425 (XO: 38.175156621)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 117.188 usec -> 2022-05-28T12:09:28.175.127.912 (XO: 38.175130108)
