-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 00:37:08 2023 UTC
-- File 20220528_120432_1229.dat, 9107  packets found.
APID:
      1229 (EW) in 9107 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 9107 packets (all packets)
Total number of packets: 
      36720 in 9107 packets (all packets)
Packet No from 1 to 9107
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 10.940066 to 29.823389
-each separately from 10s to 29s and from 483526us to 940066us.
   Differences in neigboring packets: 
      0.000000 in 9104 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      18.543460 in 1 packets: 5508
      0.339863 in 1 packets: 6936
MIU packet time from 2022-05-28T12:04:45.940.066.000 to 2022-05-28T12:05:04.823.389.000
   Differences in neigboring packets: 
      0.000000 in 9104 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      18.543460 in 1 packets: 5508
      0.339863 in 1 packets: 6936
MIU orbit number: 
      60000 in 9107 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 9107 packets (all packets)
MIU orbit information / number of events: 
      0 in 9107 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 9107 packets (all packets)
Instrument anomalies: 
      0 (OK) in 9107 packets (all packets)
FPGA configuration version: 
      211 in 9107 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.6642947 to 30.6642309
   Differences in neigboring packets: 
      0.0000000 in 9105 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      18.9999362 in 1 packets: 5508
Temperature FPGA: 
      01 -> 53.7 degC in 9107 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 9107 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 9107 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 9107 packets (all packets)
Number of external events: 
      1 in 5508 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1428 packets: 5509, 5510, 5511, 5512, 5513, 5514, 5515, 5516, 5517, 5518, 5519, 5520, 5521, 5522, 5523, 5524, 5525, 5526, ...
      3 in 2171 packets: 6937, 6938, 6939, 6940, 6941, 6942, 6943, 6944, 6945, 6946, 6947, 6948, 6949, 6950, 6951, 6952, 6953, 6954, ...
Number of generated alerts: 
      2 in 5508 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 1428 packets: 5509, 5510, 5511, 5512, 5513, 5514, 5515, 5516, 5517, 5518, 5519, 5520, 5521, 5522, 5523, 5524, 5525, 5526, ...
      4 in 2171 packets: 6937, 6938, 6939, 6940, 6941, 6942, 6943, 6944, 6945, 6946, 6947, 6948, 6949, 6950, 6951, 6952, 6953, 6954, ...
Survey memory error: 
      0 in 9107 packets (all packets)
Event memory error: 
      0 in 9107 packets (all packets)
Configuration error: 
      0 in 9107 packets (all packets)
SDRAM CRC error: 
      0 in 9107 packets (all packets)
Radio gain: 
      0 in 9107 packets (all packets)
Radio bandwidth: 
      0 in 9107 packets (all packets)
Digital gain: 
      0 in 9107 packets (all packets)
Analog gain: 
      0 in 9107 packets (all packets)

EVENT DATA READER
-- File 20220528_120432_1229.dat, 44 EW pages found. Extra packets: 131.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      1 in 5508 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1428 packets: 5509, 5510, 5511, 5512, 5513, 5514, 5515, 5516, 5517, 5518, 5519, 5520, 5521, 5522, 5523, 5524, 5525, 5526, ...
      3 in 2040 packets: 6937, 6938, 6939, 6940, 6941, 6942, 6943, 6944, 6945, 6946, 6947, 6948, 6949, 6950, 6951, 6952, 6953, 6954, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 12.6633181 to 31.4667971
      Difference between the neighboring packets: 
      0.0000170 in 8973 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      18.5191370 in 1 packets: 5508
      0.1318010 in 1 packets: 6936
   Memory page number from 0 to 81
      Difference between the neighboring packets: 
      0 in 8932 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 42 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      39 in 1 packets: 6936
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 8932 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 41 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -6 in 1 packets: 5508
      -59 in 1 packets: 6936
   MIU event flag 
      1 in 8976 packets (all packets)
   Time triggered event flag 
      0 in 8976 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-05-28T12:04:45.884.577.637
   Min EW header abs. time found among all events            : 2022-05-28T12:04:45.940.066.000
   Max EW event (1st sample) abs. time found among all events: 2022-05-28T12:05:04.767.900.642
   Max EW header abs. time found among all events            : 2022-05-28T12:05:04.823.389.000
   TM values from -272 to 237 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 5508
    Page no. 34:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 27  ordered packets: 5508
   TM values from -272 to 237 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=10 C1MHz=0.940066
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 5507 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 12.6633181
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-28T12:04:45.940.066.000
   Event 1st sample abs. time : 2022-05-28T12:04:45.884.577.637)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.663318100 corrected by 40bit 10MHz counter of 11.664294700 obtained at 1Hz counter of 10. Result: 10.999026797
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              10.999026622
   Event data descriptor XO time (reconstructed seconds from XP start)    :              10.999026797
   Sampling frequency correction factor for this event:     0.999996600

 Event no. 2 total number of packets: 1428
    Page no. 54:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 7  ordered packets: 1428
   TM values from -254 to 224 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=29 C1MHz=0.483526
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1427 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 31.2760741
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-28T12:05:04.483.526.000
   Event 1st sample abs. time : 2022-05-28T12:05:04.428.037.642)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 29, 32bit 10MHz counter = 31.276074100 corrected by 40bit 10MHz counter of 29.664234200 obtained at 1Hz counter of 28. Result: 29.611845219
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              29.611845044
   Event data descriptor XO time (reconstructed seconds from XP start)    :              29.611845219
   Sampling frequency correction factor for this event:     0.999996700

 Event no. 3 total number of packets: 2040
    Page no. 1:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 10  ordered packets: 2040
   TM values from -266 to 218 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=29 C1MHz=0.823389
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 2039 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 31.4321341
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-28T12:05:04.823.389.000
   Event 1st sample abs. time : 2022-05-28T12:05:04.767.900.642)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 29, 32bit 10MHz counter = 31.432134100 corrected by 40bit 10MHz counter of 29.664234200 obtained at 1Hz counter of 28. Result: 29.767905734
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              29.767905559
   Event data descriptor XO time (reconstructed seconds from XP start)    :              29.767905734
   Sampling frequency correction factor for this event:     0.999996700
--
Extremes in separate events:

 - Event number: 1 from 2022-05-28T12:04:45.884.577.637 to 2022-05-28T12:04:45.978.213.955 (XO: 10.999026622 - 11.092662940)
   Min TM value:-272 at an offset of 91519.524 usec -> 2022-05-28T12:04:45.976.097.160 (XO: 11.090546145)
   Max TM value:237 at an offset of 91519.261 usec -> 2022-05-28T12:04:45.976.096.898 (XO: 11.090545883)
   Max dTM/dt [1/us]:2.50e+03 at an offset of 43513.810 usec -> 2022-05-28T12:04:45.928.091.447 (XO: 11.042540432)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 91519.361 usec -> 2022-05-28T12:04:45.976.096.998 (XO: 11.090545983)
   Med: -4.00000   Ave: -4.4639949   Std: 8.0489888

 - Event number: 2 from 2022-05-28T12:05:04.428.037.642 to 2022-05-28T12:05:04.452.313.722 (XO: 29.611845044 - 29.636121124)
   Min TM value:-254 at an offset of 17029.419 usec -> 2022-05-28T12:05:04.445.067.060 (XO: 29.628874463)
   Max TM value:224 at an offset of 7104.373 usec -> 2022-05-28T12:05:04.435.142.015 (XO: 29.618949418)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 17031.619 usec -> 2022-05-28T12:05:04.445.069.260 (XO: 29.628876663)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 7104.461 usec -> 2022-05-28T12:05:04.435.142.102 (XO: 29.618949505)
   Med: -4.00000   Ave: -4.4770159   Std: 7.6953158

 - Event number: 3 from 2022-05-28T12:05:04.767.900.642 to 2022-05-28T12:05:04.802.580.756 (XO: 29.767905559 - 29.802585674)
   Min TM value:-266 at an offset of 908.128 usec -> 2022-05-28T12:05:04.768.808.769 (XO: 29.768813687)
   Max TM value:218 at an offset of 1011.641 usec -> 2022-05-28T12:05:04.768.912.282 (XO: 29.768917200)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 1011.516 usec -> 2022-05-28T12:05:04.768.912.157 (XO: 29.768917075)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 985.228 usec -> 2022-05-28T12:05:04.768.885.870 (XO: 29.768890787)
   Med: -4.00000   Ave: -4.4928810   Std: 8.0096660
